12/15/23 12:46:41.869	Information	Image Studio Version 5.2.5	GMT -5:00
12/15/23 12:46:41.869	Information	OS: Windows 8 6.2	GMT -5:00
12/15/23 12:46:41.869	Information	Java Home: c:\program files\licor\image studio\jre	GMT -5:00
12/15/23 12:46:41.885	Information	JRE Version: 1.7.0_72 32-bit	GMT -5:00
12/15/23 12:46:41.885	Information	Java library: C:\Program Files\Licor\Image Studio;C:\WINDOWS\Sun\Java\bin;C:\WINDOWS\system32;C:\WINDOWS;C:\Program Files (x86)\Common Files\Oracle\Java\javapath;C:\WINDOWS\system32;C:\WINDOWS;C:\WINDOWS\System32\Wbem;C:\WINDOWS\System32\WindowsPowerShell\v1.0\;C:\WINDOWS\System32\OpenSSH\;C:\Program Files\PolicyPak\;C:\Users\wmao2\AppData\Local\Microsoft\WindowsApps;;c:\program files\licor\image studio\jre\bin;. 32-bit	GMT -5:00
12/15/23 12:46:41.885	Information	Database migrations: 5.0_Initial, 5.0.66_New log entries fks, 5.0.67_Change lane wrappers, 5.2.68_Update database version, 5.2.69_Add Image Id Constraint	GMT -5:00
12/15/23 12:46:41.885	Information	Installed Analysis Key: Grid_Analysis	GMT -5:00
12/15/23 12:46:41.900	Information	Installed Analysis Key: Image_Studio_Base	GMT -5:00
12/15/23 12:46:41.900	Information	Installed Analysis Key: Shapes	GMT -5:00
12/15/23 12:46:41.900	Information	Installed Analysis Key: Western_Analysis	GMT -5:00
12/15/23 12:46:41.900	Information	Installed Instrument Key: CDG_Acquire	GMT -5:00
12/15/23 12:46:41.900	Information	Installed Instrument Key: Odyssey_Acquire	GMT -5:00
12/15/23 12:46:41.900	Information	Installed Instrument Key: OdysseyCLx_Acquire	GMT -5:00
12/15/23 12:46:41.916	Information	Installed Instrument Key: OdysseyFc_Acquire	GMT -5:00
12/15/23 12:46:41.916	Information	Installed Instrument Key: OdysseySa_Acquire	GMT -5:00
12/15/23 12:46:41.916	Information	Installed Instrument Key: Pearl_Acquire	GMT -5:00
12/15/23 12:46:41.916	Information	Selected Instrument Key: OdysseyCLx_Acquire	GMT -5:00
12/15/23 12:46:41.916	Information	Loading project C:\Users\wmao2\2023-12-15\Images\...	GMT -5:00
12/15/23 12:46:42.385	Information	Loading more acquisitions...	GMT -5:00
12/15/23 12:46:42.416	Information	USBCat Client Version 1.0.2	GMT -5:00
12/15/23 12:46:43.228	Information	Finished adding Acquisitions to Work Area.	GMT -5:00
12/15/23 12:46:43.244	Information	Image Studio Version 5.2.5	GMT -5:00
12/15/23 12:46:49.868	Information	Have discovered 1 instruments.	GMT -5:00
12/15/23 12:46:49.868	Information	Discovered Name: CLX-2217 IP Address: fe80:0:0:0:201:29ff:fe8d:a2af%14	GMT -5:00
12/15/23 12:46:49.868	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/15/23 12:46:49.883	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/15/23 12:46:49.930	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/15/23 12:46:49.977	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/15/23 12:46:49.977	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/15/23 12:46:49.977	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/15/23 12:46:49.977	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/15/23 12:46:49.977	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/15/23 12:46:49.977	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/15/23 12:46:49.977	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/15/23 12:46:49.977	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/15/23 12:46:49.977	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/15/23 12:46:50.664	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    640 mV 	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               25 C  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1710 mA 	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          26 C  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                31 C  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 4.93 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.87 V  	GMT -5:00
12/15/23 12:46:52.961	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.0 V  	GMT -5:00
12/15/23 12:46:53.007	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/15/23 12:46:57.975	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/15/23 12:47:00.990	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1669 mA 	GMT -5:00
12/15/23 12:47:01.990	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1638 mA 	GMT -5:00
12/15/23 12:47:02.053	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          24 C  	GMT -5:00
12/15/23 12:47:02.990	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1598 mA 	GMT -5:00
12/15/23 12:47:04.005	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1547 mA 	GMT -5:00
12/15/23 12:47:05.005	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1486 mA 	GMT -5:00
12/15/23 12:47:06.005	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1415 mA 	GMT -5:00
12/15/23 12:47:07.005	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1323 mA 	GMT -5:00
12/15/23 12:47:08.020	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1221 mA 	GMT -5:00
12/15/23 12:47:08.067	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          23 C  	GMT -5:00
12/15/23 12:47:09.020	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1119 mA 	GMT -5:00
12/15/23 12:47:10.020	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1018 mA 	GMT -5:00
12/15/23 12:47:11.019	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      916 mA 	GMT -5:00
12/15/23 12:47:12.035	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      814 mA 	GMT -5:00
12/15/23 12:47:13.035	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      712 mA 	GMT -5:00
12/15/23 12:47:14.035	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      621 mA 	GMT -5:00
12/15/23 12:47:14.097	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/15/23 12:47:15.034	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      539 mA 	GMT -5:00
12/15/23 12:47:15.097	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          22 C  	GMT -5:00
12/15/23 12:47:16.050	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      468 mA 	GMT -5:00
12/15/23 12:47:17.051	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      397 mA 	GMT -5:00
12/15/23 12:47:18.050	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      336 mA 	GMT -5:00
12/15/23 12:47:19.053	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/15/23 12:47:20.059	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      234 mA 	GMT -5:00
12/15/23 12:47:20.121	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/15/23 12:47:21.069	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      183 mA 	GMT -5:00
12/15/23 12:47:22.060	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      153 mA 	GMT -5:00
12/15/23 12:47:23.078	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      122 mA 	GMT -5:00
12/15/23 12:47:24.078	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                       92 mA 	GMT -5:00
12/15/23 12:47:26.078	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                       51 mA 	GMT -5:00
12/15/23 12:47:28.546	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 12:47:29.093	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                       20 mA 	GMT -5:00
12/15/23 12:47:41.658	Information	[Instrument Request]  1 ScanAutoEstimate 169um medium 0.0 0.0 8.1 0.0 4.9 700 800 	GMT -5:00
12/15/23 12:47:41.752	Information	[Instrument Response] 1 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0 8.1 0 4.9 700 800	GMT -5:00
12/15/23 12:47:41.783	Information	[Instrument Response] 1 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 12:47:41.783	Information	[Instrument Response] 1 Status 0 X mapping: [x0=0,x1=8.1] => [left=-60,right=3780] w=479	GMT -5:00
12/15/23 12:47:41.799	Information	[Instrument Response] 1 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 12:47:41.799	Information	[Instrument Response] 1 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/15/23 12:47:41.814	Information	[Instrument Response] 1 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 12:47:41.814	Information	[Instrument Response] 1 Status 0 Computed scan parameters	GMT -5:00
12/15/23 12:47:41.830	Information	[Instrument Response] 1 Status 0 Seconds/line=0.495775	GMT -5:00
12/15/23 12:47:41.830	Information	[Instrument Response] 1 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/15/23 12:47:41.830	Information	[Instrument Response] 1 Status 0 linesPerBlock=20 flushLinesPerBlock=10	GMT -5:00
12/15/23 12:47:41.845	Information	[Instrument Response] 1 Status 0 Blocks=30	GMT -5:00
12/15/23 12:47:41.845	Information	[Instrument Response] 1 Status 0 Estimates (sec): setup=2.02 line=0.525521 => total=307	GMT -5:00
12/15/23 12:47:41.861	Information	[Instrument Response] 1 Progress 0% 7 minutes	GMT -5:00
12/15/23 12:47:41.861	Information	[Instrument Response] 1 Result 0 OK	GMT -5:00
12/15/23 12:47:43.142	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          23 C  	GMT -5:00
12/15/23 12:47:45.095	Information	[Instrument Request]  2 ScanAutoEstimate 169um medium 0.0 0.0 8.1 0.0 5.1 700 800 	GMT -5:00
12/15/23 12:47:45.110	Information	[Instrument Response] 2 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0 8.1 0 5.1 700 800	GMT -5:00
12/15/23 12:47:45.157	Information	[Instrument Response] 2 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 12:47:45.157	Information	[Instrument Response] 2 Status 0 X mapping: [x0=0,x1=8.1] => [left=-60,right=3780] w=479	GMT -5:00
12/15/23 12:47:45.173	Information	[Instrument Response] 2 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 12:47:45.173	Information	[Instrument Response] 2 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 12:47:45.188	Information	[Instrument Response] 2 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 12:47:45.188	Information	[Instrument Response] 2 Status 0 Computed scan parameters	GMT -5:00
12/15/23 12:47:45.204	Information	[Instrument Response] 2 Status 0 Seconds/line=0.495775	GMT -5:00
12/15/23 12:47:45.204	Information	[Instrument Response] 2 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 12:47:45.204	Information	[Instrument Response] 2 Status 0 linesPerBlock=20 flushLinesPerBlock=10	GMT -5:00
12/15/23 12:47:45.204	Information	[Instrument Response] 2 Status 0 Blocks=31	GMT -5:00
12/15/23 12:47:45.204	Information	[Instrument Response] 2 Status 0 Estimates (sec): setup=2.02 line=0.525521 => total=320	GMT -5:00
12/15/23 12:47:45.204	Information	[Instrument Response] 2 Progress 0% 7 minutes	GMT -5:00
12/15/23 12:47:45.220	Information	[Instrument Response] 2 Result 0 OK	GMT -5:00
12/15/23 12:47:46.392	Information	Starting acquisition...	GMT -5:00
12/15/23 12:47:47.907	Information	[Instrument Request]  3 ScanAuto 169um medium 0.0 0.0 8.1 0.0 5.1 700 800 	GMT -5:00
12/15/23 12:47:48.000	Information	[Instrument Response] 3 Status 1 Executing command: ScanAuto 169um medium 0 0 8.1 0 5.1 700 800	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Status 0 X mapping: [x0=0,x1=8.1] => [left=-60,right=3780] w=479	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Status 0 Computed scan parameters	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Status 0 Seconds/line=0.495775	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Status 0 linesPerBlock=20 flushLinesPerBlock=10	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Status 0 Blocks=31	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Status 0 Estimates (sec): setup=2.02 line=0.525521 => total=320	GMT -5:00
12/15/23 12:47:48.016	Information	[Instrument Response] 3 Progress 0% 7 minutes	GMT -5:00
12/15/23 12:47:48.032	Information	[Instrument Response] 3 Progress 0% Initializing	GMT -5:00
12/15/23 12:47:48.047	Information	[Instrument Response] 3 Status 0 Preparation complete	GMT -5:00
12/15/23 12:47:50.937	Information	[Instrument Response] 3 Status 0 Motors initialized	GMT -5:00
12/15/23 12:47:50.984	Information	[Instrument Response] 3 Status 0 X-Motor configured	GMT -5:00
12/15/23 12:47:50.984	Information	[Instrument Response] 3 Status 0 DSP configured	GMT -5:00
12/15/23 12:47:51.031	Information	[Instrument Response] 3 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 12:47:51.031	Information	[Instrument Response] 3 Status 0 Cooling	GMT -5:00
12/15/23 12:47:51.031	Information	[Instrument Response] 3 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 12:47:51.187	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 12:47:59.205	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                       51 mA 	GMT -5:00
12/15/23 12:47:59.267	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          24 C  	GMT -5:00
12/15/23 12:48:03.230	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                       81 mA 	GMT -5:00
12/15/23 12:48:08.241	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      112 mA 	GMT -5:00
12/15/23 12:48:12.265	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      142 mA 	GMT -5:00
12/15/23 12:48:18.281	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      173 mA 	GMT -5:00
12/15/23 12:48:23.311	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 12:48:27.327	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      204 mA 	GMT -5:00
12/15/23 12:48:28.327	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          24 C  	GMT -5:00
12/15/23 12:48:28.983	Information	[Instrument Response] 3 Status 0 Create MergeImageChannel: 700 20 0 0 479 10 479 302 xofs=1 yofs=0	GMT -5:00
12/15/23 12:48:29.046	Information	[Instrument Response] 3 Status 0 Create MergeImageChannel: 800 20 0 0 479 10 479 302 xofs=0 yofs=0	GMT -5:00
12/15/23 12:48:29.046	Information	[Instrument Response] 3 Status 0 start=34 scan_lines=10 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 12:48:29.046	Information	[Instrument Response] 3 Status 0 start=34 scan_lines=20 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 12:48:29.046	Information	[Instrument Response] 3 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 12:48:29.327	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.87 V  	GMT -5:00
12/15/23 12:48:29.765	Information	[Instrument Response] 3 Progress 0% 7 minutes	GMT -5:00
12/15/23 12:48:30.327	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:48:30.389	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 12:48:30.389	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 12:48:30.389	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 12:48:30.389	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 12:48:32.342	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 12:48:34.342	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    653 mV 	GMT -5:00
12/15/23 12:48:34.404	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      234 mA 	GMT -5:00
12/15/23 12:48:37.357	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:48:37.435	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         131 V  	GMT -5:00
12/15/23 12:48:37.435	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:48:37.435	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:48:38.356	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          92 V  	GMT -5:00
12/15/23 12:48:39.360	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:48:41.375	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      265 mA 	GMT -5:00
12/15/23 12:48:44.453	Information	[Instrument Response] 3 ImageBlock 700 0 0 479 10 479 302 9580	GMT -5:00
12/15/23 12:48:44.453	Information	[Instrument Response] 3 ImageBlock 800 0 0 479 10 479 302 9580	GMT -5:00
12/15/23 12:48:44.468	Information	Transferring image block at (0,0) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:48:44.593	Information	Transferring image block at (0,0) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:48:49.405	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/15/23 12:48:50.405	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/15/23 12:48:50.468	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 12:48:50.468	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 12:48:50.468	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 12:48:51.420	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:48:57.188	Information	[Instrument Response] 3 ImageBlock 700 0 10 479 10 479 302 9580	GMT -5:00
12/15/23 12:48:57.188	Information	[Instrument Response] 3 ImageBlock 800 0 10 479 10 479 302 9580	GMT -5:00
12/15/23 12:48:57.203	Information	Transferring image block at (0,10) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:48:57.234	Information	Transferring image block at (0,10) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:48:58.294	Information	[Instrument Response] 3 Progress 7% 6 minutes	GMT -5:00
12/15/23 12:49:03.460	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:49:03.522	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 12:49:03.522	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:49:03.522	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:49:04.475	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:49:09.916	Information	[Instrument Response] 3 ImageBlock 700 0 20 479 10 479 302 9580	GMT -5:00
12/15/23 12:49:09.916	Information	[Instrument Response] 3 ImageBlock 800 0 20 479 10 479 302 9580	GMT -5:00
12/15/23 12:49:09.931	Information	Transferring image block at (0,20) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:49:09.978	Information	Transferring image block at (0,20) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:49:10.494	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 12:49:11.494	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:49:15.508	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         134 V  	GMT -5:00
12/15/23 12:49:15.571	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:49:15.571	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:49:15.571	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:49:15.571	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 12:49:16.517	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          91 V  	GMT -5:00
12/15/23 12:49:16.580	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:49:17.517	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:49:20.530	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 12:49:20.592	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 12:49:21.530	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/15/23 12:49:21.592	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:49:22.655	Information	[Instrument Response] 3 ImageBlock 700 0 30 479 10 479 302 9580	GMT -5:00
12/15/23 12:49:22.655	Information	[Instrument Response] 3 ImageBlock 800 0 30 479 10 479 302 9580	GMT -5:00
12/15/23 12:49:22.655	Information	Transferring image block at (0,30) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:49:22.701	Information	Transferring image block at (0,30) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:49:28.560	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:49:28.622	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         113 V  	GMT -5:00
12/15/23 12:49:28.622	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:49:28.622	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:49:29.575	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:49:31.583	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 12:49:35.379	Information	[Instrument Response] 3 ImageBlock 700 0 40 479 10 479 302 9580	GMT -5:00
12/15/23 12:49:35.379	Information	[Instrument Response] 3 ImageBlock 800 0 40 479 10 479 302 9580	GMT -5:00
12/15/23 12:49:35.395	Information	Transferring image block at (0,40) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:49:35.426	Information	Transferring image block at (0,40) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:49:41.623	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:49:41.670	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:49:41.670	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:49:41.670	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:49:48.124	Information	[Instrument Response] 3 ImageBlock 700 0 50 479 10 479 302 9580	GMT -5:00
12/15/23 12:49:48.124	Information	[Instrument Response] 3 ImageBlock 800 0 50 479 10 479 302 9580	GMT -5:00
12/15/23 12:49:48.124	Information	Transferring image block at (0,50) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:49:48.155	Information	Transferring image block at (0,50) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:49:53.670	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 12:49:53.717	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         139 V  	GMT -5:00
12/15/23 12:49:53.717	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:49:53.717	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:49:53.717	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 12:49:54.670	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          87 V  	GMT -5:00
12/15/23 12:49:54.733	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:49:55.670	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:49:58.218	Information	[Instrument Response] 3 Progress 22% 5 minutes	GMT -5:00
12/15/23 12:50:00.859	Information	[Instrument Response] 3 ImageBlock 700 0 60 479 10 479 302 9580	GMT -5:00
12/15/23 12:50:00.859	Information	[Instrument Response] 3 ImageBlock 800 0 60 479 10 479 302 9580	GMT -5:00
12/15/23 12:50:00.859	Information	Transferring image block at (0,60) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:50:00.890	Information	Transferring image block at (0,60) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:50:05.718	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 12:50:06.718	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         115 V  	GMT -5:00
12/15/23 12:50:06.781	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:50:06.781	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:50:06.781	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 12:50:06.781	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 12:50:07.734	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:50:13.586	Information	[Instrument Response] 3 ImageBlock 700 0 70 479 10 479 302 9580	GMT -5:00
12/15/23 12:50:13.586	Information	[Instrument Response] 3 ImageBlock 800 0 70 479 10 479 302 9580	GMT -5:00
12/15/23 12:50:13.586	Information	Transferring image block at (0,70) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:50:13.617	Information	Transferring image block at (0,70) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:50:15.757	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 12:50:15.820	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 12:50:16.755	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:50:16.818	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 12:50:19.782	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:50:19.829	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          98 V  	GMT -5:00
12/15/23 12:50:19.829	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:50:19.829	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:50:20.782	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:50:24.797	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/15/23 12:50:25.806	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 12:50:25.853	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 12:50:26.321	Information	[Instrument Response] 3 ImageBlock 700 0 80 479 10 479 302 9580	GMT -5:00
12/15/23 12:50:26.321	Information	[Instrument Response] 3 ImageBlock 800 0 80 479 10 479 302 9580	GMT -5:00
12/15/23 12:50:26.321	Information	Transferring image block at (0,80) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:50:26.352	Information	Transferring image block at (0,80) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:50:26.806	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 12:50:27.962	Information	[Instrument Response] 3 Progress 30% 4 minutes, 30 seconds	GMT -5:00
12/15/23 12:50:31.821	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         142 V  	GMT -5:00
12/15/23 12:50:31.884	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/15/23 12:50:31.884	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 12:50:31.884	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 12:50:31.884	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 12:50:32.821	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          86 V  	GMT -5:00
12/15/23 12:50:32.883	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 12:50:32.883	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:50:33.829	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:50:35.831	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 12:50:36.846	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:50:39.043	Information	[Instrument Response] 3 ImageBlock 700 0 90 479 10 479 302 9580	GMT -5:00
12/15/23 12:50:39.043	Information	[Instrument Response] 3 ImageBlock 800 0 90 479 10 479 302 9580	GMT -5:00
12/15/23 12:50:39.058	Information	Transferring image block at (0,90) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:50:39.089	Information	Transferring image block at (0,90) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:50:44.872	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:50:44.935	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         118 V  	GMT -5:00
12/15/23 12:50:44.935	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:50:44.935	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:50:45.881	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:50:51.781	Information	[Instrument Response] 3 ImageBlock 700 0 100 479 10 479 302 9580	GMT -5:00
12/15/23 12:50:51.781	Information	[Instrument Response] 3 ImageBlock 800 0 100 479 10 479 302 9580	GMT -5:00
12/15/23 12:50:51.781	Information	Transferring image block at (0,100) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:50:51.812	Information	Transferring image block at (0,100) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:50:57.175	Information	[Instrument Response] 3 Progress 38% 4 minutes	GMT -5:00
12/15/23 12:50:57.925	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         100 V  	GMT -5:00
12/15/23 12:50:57.972	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:50:57.972	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:50:57.972	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:50:58.934	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:51:02.949	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 12:51:03.950	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:51:04.513	Information	[Instrument Response] 3 ImageBlock 700 0 110 479 10 479 302 9580	GMT -5:00
12/15/23 12:51:04.513	Information	[Instrument Response] 3 ImageBlock 800 0 110 479 10 479 302 9580	GMT -5:00
12/15/23 12:51:04.513	Information	Transferring image block at (0,110) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:51:04.544	Information	Transferring image block at (0,110) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:51:09.965	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         151 V  	GMT -5:00
12/15/23 12:51:10.028	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         147 V  	GMT -5:00
12/15/23 12:51:10.028	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:51:10.028	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:51:10.028	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 12:51:10.981	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:51:11.028	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          86 V  	GMT -5:00
12/15/23 12:51:11.028	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:51:11.981	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:51:17.245	Information	[Instrument Response] 3 ImageBlock 700 0 120 479 10 479 302 9580	GMT -5:00
12/15/23 12:51:17.245	Information	[Instrument Response] 3 ImageBlock 800 0 120 479 10 479 302 9580	GMT -5:00
12/15/23 12:51:17.245	Information	Transferring image block at (0,120) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:51:17.276	Information	Transferring image block at (0,120) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:51:23.030	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         119 V  	GMT -5:00
12/15/23 12:51:23.073	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 12:51:23.073	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:51:23.074	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:51:24.034	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:51:27.853	Information	[Instrument Response] 3 Progress 46% 3 minutes, 30 seconds	GMT -5:00
12/15/23 12:51:29.962	Information	[Instrument Response] 3 ImageBlock 700 0 130 479 10 479 302 9580	GMT -5:00
12/15/23 12:51:29.962	Information	[Instrument Response] 3 ImageBlock 800 0 130 479 10 479 302 9580	GMT -5:00
12/15/23 12:51:29.978	Information	Transferring image block at (0,130) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:51:30.009	Information	Transferring image block at (0,130) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:51:31.056	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 12:51:32.056	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:51:36.068	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:51:36.131	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         103 V  	GMT -5:00
12/15/23 12:51:36.131	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:51:36.131	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:51:37.084	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:51:42.700	Information	[Instrument Response] 3 ImageBlock 700 0 140 479 10 479 302 9580	GMT -5:00
12/15/23 12:51:42.700	Information	[Instrument Response] 3 ImageBlock 800 0 140 479 10 479 302 9580	GMT -5:00
12/15/23 12:51:42.700	Information	Transferring image block at (0,140) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:51:42.731	Information	Transferring image block at (0,140) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:51:48.122	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         150 V  	GMT -5:00
12/15/23 12:51:48.169	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:51:48.169	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:51:48.169	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:51:48.169	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 12:51:49.122	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          87 V  	GMT -5:00
12/15/23 12:51:49.184	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:51:50.122	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:51:55.434	Information	[Instrument Response] 3 ImageBlock 700 0 150 479 10 479 302 9580	GMT -5:00
12/15/23 12:51:55.434	Information	[Instrument Response] 3 ImageBlock 800 0 150 479 10 479 302 9580	GMT -5:00
12/15/23 12:51:55.434	Information	Transferring image block at (0,150) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:51:55.465	Information	Transferring image block at (0,150) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:51:58.152	Information	[Instrument Response] 3 Progress 54% 3 minutes	GMT -5:00
12/15/23 12:52:01.183	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:52:01.245	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/15/23 12:52:01.245	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         123 V  	GMT -5:00
12/15/23 12:52:01.245	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:52:01.245	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:52:02.182	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 12:52:02.245	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:52:08.169	Information	[Instrument Response] 3 ImageBlock 700 0 160 479 10 479 302 9580	GMT -5:00
12/15/23 12:52:08.169	Information	[Instrument Response] 3 ImageBlock 800 0 160 479 10 479 302 9580	GMT -5:00
12/15/23 12:52:08.169	Information	Transferring image block at (0,160) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:52:08.200	Information	Transferring image block at (0,160) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:52:14.230	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         104 V  	GMT -5:00
12/15/23 12:52:14.293	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:52:14.293	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:52:14.293	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:52:15.246	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:52:18.245	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 12:52:19.261	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:52:20.901	Information	[Instrument Response] 3 ImageBlock 700 0 170 479 10 479 302 9580	GMT -5:00
12/15/23 12:52:20.901	Information	[Instrument Response] 3 ImageBlock 800 0 170 479 10 479 302 9580	GMT -5:00
12/15/23 12:52:20.901	Information	Transferring image block at (0,170) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:52:20.947	Information	Transferring image block at (0,170) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:52:26.305	Information	[Instrument Response] 3 Progress 61% 2 minutes, 30 seconds	GMT -5:00
12/15/23 12:52:27.290	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:52:27.337	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          89 V  	GMT -5:00
12/15/23 12:52:27.337	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:52:27.337	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:52:28.289	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:52:33.635	Information	[Instrument Response] 3 ImageBlock 700 0 180 479 10 479 302 9580	GMT -5:00
12/15/23 12:52:33.635	Information	[Instrument Response] 3 ImageBlock 800 0 180 479 10 479 302 9580	GMT -5:00
12/15/23 12:52:33.635	Information	Transferring image block at (0,180) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:52:33.666	Information	Transferring image block at (0,180) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:52:36.319	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 12:52:37.334	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:52:39.328	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         125 V  	GMT -5:00
12/15/23 12:52:39.391	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:52:39.391	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:52:39.391	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:52:40.347	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 12:52:46.359	Information	[Instrument Response] 3 ImageBlock 700 0 190 479 10 479 302 9580	GMT -5:00
12/15/23 12:52:46.359	Information	[Instrument Response] 3 ImageBlock 800 0 190 479 10 479 302 9580	GMT -5:00
12/15/23 12:52:46.374	Information	Transferring image block at (0,190) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:52:46.406	Information	Transferring image block at (0,190) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:52:52.400	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:52:52.447	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         107 V  	GMT -5:00
12/15/23 12:52:52.447	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:52:52.447	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:52:53.400	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:52:58.066	Information	[Instrument Response] 3 Progress 69% 2 minutes	GMT -5:00
12/15/23 12:52:59.101	Information	[Instrument Response] 3 ImageBlock 700 0 200 479 10 479 302 9580	GMT -5:00
12/15/23 12:52:59.101	Information	[Instrument Response] 3 ImageBlock 800 0 200 479 10 479 302 9580	GMT -5:00
12/15/23 12:52:59.101	Information	Transferring image block at (0,200) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:52:59.132	Information	Transferring image block at (0,200) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:53:05.454	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          91 V  	GMT -5:00
12/15/23 12:53:05.516	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 12:53:05.516	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:53:05.516	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:53:06.454	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:53:08.122	Information	[Instrument Response] 3 Progress 72% 1 minute, 50 seconds	GMT -5:00
12/15/23 12:53:11.835	Information	[Instrument Response] 3 ImageBlock 700 0 210 479 10 479 302 9580	GMT -5:00
12/15/23 12:53:11.835	Information	[Instrument Response] 3 ImageBlock 800 0 210 479 10 479 302 9580	GMT -5:00
12/15/23 12:53:11.851	Information	Transferring image block at (0,210) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:53:11.866	Information	Transferring image block at (0,210) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:53:17.236	Information	[Instrument Response] 3 Progress 75% 1 minute, 40 seconds	GMT -5:00
12/15/23 12:53:17.502	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:53:17.564	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         128 V  	GMT -5:00
12/15/23 12:53:17.564	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:53:17.564	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:53:18.505	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/15/23 12:53:19.505	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:53:23.521	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 12:53:24.536	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:53:24.568	Information	[Instrument Response] 3 ImageBlock 700 0 220 479 10 479 302 9580	GMT -5:00
12/15/23 12:53:24.568	Information	[Instrument Response] 3 ImageBlock 800 0 220 479 10 479 302 9580	GMT -5:00
12/15/23 12:53:24.568	Information	Transferring image block at (0,220) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:53:24.599	Information	Transferring image block at (0,220) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:53:28.354	Information	[Instrument Response] 3 Progress 77% 1 minute, 30 seconds	GMT -5:00
12/15/23 12:53:30.550	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         108 V  	GMT -5:00
12/15/23 12:53:30.613	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 12:53:30.613	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:53:30.613	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:53:31.550	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:53:37.295	Information	[Instrument Response] 3 ImageBlock 700 0 230 479 10 479 302 9580	GMT -5:00
12/15/23 12:53:37.295	Information	[Instrument Response] 3 ImageBlock 800 0 230 479 10 479 302 9580	GMT -5:00
12/15/23 12:53:37.310	Information	Transferring image block at (0,230) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:53:37.342	Information	Transferring image block at (0,230) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:53:38.404	Information	[Instrument Response] 3 Progress 80% 1 minute, 20 seconds	GMT -5:00
12/15/23 12:53:43.604	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:53:43.667	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/15/23 12:53:43.667	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:53:43.667	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:53:44.605	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:53:47.932	Information	[Instrument Response] 3 Progress 82% 1 minute, 10 seconds	GMT -5:00
12/15/23 12:53:50.026	Information	[Instrument Response] 3 ImageBlock 700 0 240 479 10 479 302 9580	GMT -5:00
12/15/23 12:53:50.026	Information	[Instrument Response] 3 ImageBlock 800 0 240 479 10 479 302 9580	GMT -5:00
12/15/23 12:53:50.042	Information	Transferring image block at (0,240) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:53:50.073	Information	Transferring image block at (0,240) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:53:55.658	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         131 V  	GMT -5:00
12/15/23 12:53:55.721	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:53:55.721	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 12:53:55.721	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:53:55.721	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 12:53:56.658	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          92 V  	GMT -5:00
12/15/23 12:53:56.721	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:53:57.674	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:53:57.970	Information	[Instrument Response] 3 Progress 85% 1 minute	GMT -5:00
12/15/23 12:54:02.766	Information	[Instrument Response] 3 ImageBlock 700 0 250 479 10 479 302 9580	GMT -5:00
12/15/23 12:54:02.766	Information	[Instrument Response] 3 ImageBlock 800 0 250 479 10 479 302 9580	GMT -5:00
12/15/23 12:54:02.766	Information	Transferring image block at (0,250) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:54:02.797	Information	Transferring image block at (0,250) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:54:08.172	Information	[Instrument Response] 3 Progress 88% 50 seconds	GMT -5:00
12/15/23 12:54:08.703	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 12:54:08.765	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         111 V  	GMT -5:00
12/15/23 12:54:08.765	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:54:08.765	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:54:09.718	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:54:15.502	Information	[Instrument Response] 3 ImageBlock 700 0 260 479 10 479 302 9580	GMT -5:00
12/15/23 12:54:15.502	Information	[Instrument Response] 3 ImageBlock 800 0 260 479 10 479 302 9580	GMT -5:00
12/15/23 12:54:15.502	Information	Transferring image block at (0,260) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:54:15.549	Information	Transferring image block at (0,260) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:54:18.230	Information	[Instrument Response] 3 Progress 90% 40 seconds	GMT -5:00
12/15/23 12:54:18.755	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 12:54:19.755	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:54:21.754	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:54:21.817	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 12:54:21.817	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:54:21.817	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:54:28.236	Information	[Instrument Response] 3 ImageBlock 700 0 270 479 10 479 302 9580	GMT -5:00
12/15/23 12:54:28.236	Information	[Instrument Response] 3 ImageBlock 800 0 270 479 10 479 302 9580	GMT -5:00
12/15/23 12:54:28.236	Information	Transferring image block at (0,270) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:54:28.267	Information	Transferring image block at (0,270) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:54:28.330	Information	[Instrument Response] 3 Progress 93% 30 seconds	GMT -5:00
12/15/23 12:54:28.783	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 12:54:29.787	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:54:33.817	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 12:54:33.864	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 12:54:33.864	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:54:33.864	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:54:34.816	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          89 V  	GMT -5:00
12/15/23 12:54:35.816	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:54:38.331	Information	[Instrument Response] 3 Progress 95% 20 seconds	GMT -5:00
12/15/23 12:54:38.831	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 12:54:39.835	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:54:40.963	Information	[Instrument Response] 3 ImageBlock 700 0 280 479 10 479 302 9580	GMT -5:00
12/15/23 12:54:40.963	Information	[Instrument Response] 3 ImageBlock 800 0 280 479 10 479 302 9580	GMT -5:00
12/15/23 12:54:40.978	Information	Transferring image block at (0,280) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:54:41.009	Information	Transferring image block at (0,280) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:54:46.366	Information	[Instrument Response] 3 Progress 98% 10 seconds	GMT -5:00
12/15/23 12:54:46.866	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         114 V  	GMT -5:00
12/15/23 12:54:46.913	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 12:54:46.913	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 12:54:46.913	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 12:54:47.866	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:54:53.706	Information	[Instrument Response] 3 ImageBlock 700 0 290 479 10 479 302 9580	GMT -5:00
12/15/23 12:54:53.706	Information	[Instrument Response] 3 ImageBlock 800 0 290 479 10 479 302 9580	GMT -5:00
12/15/23 12:54:53.706	Information	Transferring image block at (0,290) with size 479x10 to the 700 channel.	GMT -5:00
12/15/23 12:54:53.737	Information	Transferring image block at (0,290) with size 479x10 to the 800 channel.	GMT -5:00
12/15/23 12:54:54.894	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 12:54:54.956	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         145 V  	GMT -5:00
12/15/23 12:54:54.956	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 12:54:54.956	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 12:54:54.956	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 12:54:55.909	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          85 V  	GMT -5:00
12/15/23 12:54:55.956	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:54:56.909	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 12:54:57.888	Information	[Instrument Response] 3 Progress 100% 0 seconds	GMT -5:00
12/15/23 12:54:57.903	Information	[Instrument Response] 3 ImageBlock 700 0 300 479 2 479 302 1916	GMT -5:00
12/15/23 12:54:57.903	Information	[Instrument Response] 3 ImageBlock 800 0 300 479 2 479 302 1916	GMT -5:00
12/15/23 12:54:57.903	Information	[Instrument Response] 3 Status 0 COMPLETE SCAN 388.115 seconds	GMT -5:00
12/15/23 12:54:57.903	Information	Transferring image block at (0,300) with size 479x2 to the 700 channel.	GMT -5:00
12/15/23 12:54:57.903	Information	[Instrument Response] 3 Status 0 Averages (sec): setup=2.00005 line=0.536611 => total=327	GMT -5:00
12/15/23 12:54:57.934	Information	Transferring image block at (0,300) with size 479x2 to the 800 channel.	GMT -5:00
12/15/23 12:54:57.950	Information	[Instrument Response] 3 Status 0 Comparisons: setup=1.00997 line=0.979334	GMT -5:00
12/15/23 12:54:57.997	Information	[Instrument Response] 3 Status 0 Finalize motors	GMT -5:00
12/15/23 12:54:57.997	Information	[Instrument Response] 3 Progress 100% Scan complete	GMT -5:00
12/15/23 12:54:58.013	Information	[Instrument Response] 3 Result 0 OK	GMT -5:00
12/15/23 12:54:58.013	Information	Acquisition 0000005_01 finished with channels: 700 800.	GMT -5:00
12/15/23 12:54:58.420	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/15/23 12:54:58.420	Information	Scan completed successfully	GMT -5:00
12/15/23 12:54:58.982	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          88 V  	GMT -5:00
12/15/23 12:54:58.998	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          55 V  	GMT -5:00
12/15/23 12:54:58.998	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 12:54:58.998	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 12:54:58.998	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 12:54:59.921	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          54 V  	GMT -5:00
12/15/23 12:54:59.968	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          34 V  	GMT -5:00
12/15/23 12:55:00.922	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          34 V  	GMT -5:00
12/15/23 12:55:00.985	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          21 V  	GMT -5:00
12/15/23 12:55:01.932	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          23 V  	GMT -5:00
12/15/23 12:55:01.977	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          14 V  	GMT -5:00
12/15/23 12:55:02.937	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          15 V  	GMT -5:00
12/15/23 12:55:02.982	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          10 V  	GMT -5:00
12/15/23 12:55:03.935	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          10 V  	GMT -5:00
12/15/23 12:55:03.988	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           7 V  	GMT -5:00
12/15/23 12:55:04.939	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           7 V  	GMT -5:00
12/15/23 12:55:04.989	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           4 V  	GMT -5:00
12/15/23 12:55:05.938	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/15/23 12:55:06.939	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/15/23 12:55:07.958	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/15/23 12:55:10.969	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 12:55:11.017	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      234 mA 	GMT -5:00
12/15/23 12:55:11.971	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/15/23 12:55:55.548	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/15/23 12:56:50.912	Information	Starting acquisition...	GMT -5:00
12/15/23 12:56:51.773	Information	[Instrument Request]  4 ScanAuto 169um medium 0.0 0.0 8.1 0.0 5.1 700 800 	GMT -5:00
12/15/23 12:57:00.693	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 1 Executing command: ScanAuto 169um medium 0 0 8.1 0 5.1 700 800	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 0 X mapping: [x0=0,x1=8.1] => [left=-60,right=3780] w=479	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 0 Computed scan parameters	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 0 Seconds/line=0.495775	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 0 linesPerBlock=20 flushLinesPerBlock=10	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 0 Blocks=31	GMT -5:00
12/15/23 12:57:00.756	Information	[Instrument Response] 4 Status 0 Estimates (sec): setup=2.02 line=0.525521 => total=320	GMT -5:00
12/15/23 12:57:00.771	Information	[Instrument Response] 4 Progress 0% 7 minutes	GMT -5:00
12/15/23 12:57:00.771	Information	[Instrument Response] 4 Progress 0% Initializing	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 Preparation complete	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 Motors initialized	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 X-Motor configured	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 DSP configured	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 Cooling	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 Create MergeImageChannel: 700 20 0 0 479 10 479 302 xofs=1 yofs=0	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 Create MergeImageChannel: 800 20 0 0 479 10 479 302 xofs=0 yofs=0	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 start=34 scan_lines=10 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 start=34 scan_lines=20 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 12:57:00.787	Information	[Instrument Response] 4 Progress 0% 7 minutes	GMT -5:00
12/15/23 12:57:00.803	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 12:57:00.803	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 12:57:00.803	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 12:57:00.803	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 12:57:00.803	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 12:57:03.403	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      265 mA 	GMT -5:00
12/15/23 12:57:08.433	Information	Canceling acquisition...	GMT -5:00
12/15/23 12:57:08.433	Information	[Instrument Request]  5 Stop 	GMT -5:00
12/15/23 12:59:08.447	Error      	The 'Stop' command timed out. The elapsed time exceeded 120 seconds. Check the instrument connection.	GMT -5:00
12/15/23 13:01:00.794	Error      	The 'ScanAuto 169um medium 0.0 0.0 8.1 0.0 5.1 700 800' command timed out. The elapsed time exceeded 240 seconds. Check the instrument connection.	GMT -5:00
12/15/23 13:01:13.654	Error      	The instrument connection was closed. If a connection is expected, check the cable connection and confirm the instrument is powered on.  To reconnect, click the Image Studio Application Button, hover over Instrument, and click Connect.	GMT -5:00
12/15/23 13:01:36.382	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/15/23 13:01:44.371	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/15/23 13:01:44.417	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/15/23 13:01:44.464	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/15/23 13:01:44.464	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/15/23 13:01:44.464	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/15/23 13:01:44.464	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/15/23 13:01:44.464	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/15/23 13:01:44.464	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/15/23 13:01:44.464	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/15/23 13:01:44.464	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/15/23 13:01:44.464	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/15/23 13:01:44.480	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/15/23 13:01:44.480	Information	[Instrument Request]  6 ScanAutoEstimate 169um medium 0.0 0.0 8.1 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0 8.1 0 5.1 700 800	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 X mapping: [x0=0,x1=8.1] => [left=-60,right=3780] w=479	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 Seconds/line=0.495775	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 linesPerBlock=20 flushLinesPerBlock=10	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 Blocks=31	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Status 0 Estimates (sec): setup=2.02 line=0.525521 => total=320	GMT -5:00
12/15/23 13:01:47.448	Information	[Instrument Response] 6 Progress 0% 7 minutes	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 6 Result 0 OK	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    659 mV 	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               26 C  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      244 mA 	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 4.98 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/15/23 13:01:47.463	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.1 V  	GMT -5:00
12/15/23 13:01:47.495	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 13:01:48.807	Information	[Instrument Request]  7 ScanAutoEstimate 169um medium 0.0 0.0 4.8 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:01:48.822	Information	[Instrument Response] 7 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0 4.8 0 5.1 700 800	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Status 0 X mapping: [x0=0,x1=4.8] => [left=-60,right=2220] w=284	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Status 0 Seconds/line=0.330775	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Status 0 Blocks=21	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Status 0 Estimates (sec): setup=2.02 line=0.350621 => total=214	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/15/23 13:01:48.870	Information	[Instrument Response] 7 Result 0 OK	GMT -5:00
12/15/23 13:01:51.671	Information	Starting acquisition...	GMT -5:00
12/15/23 13:01:52.452	Information	[Instrument Request]  8 ScanAuto 169um medium 0.0 0.0 4.8 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:01:52.530	Information	[Instrument Response] 8 Status 1 Executing command: ScanAuto 169um medium 0 0 4.8 0 5.1 700 800	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Status 0 X mapping: [x0=0,x1=4.8] => [left=-60,right=2220] w=284	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Status 0 Seconds/line=0.330775	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Status 0 Blocks=21	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Status 0 Estimates (sec): setup=2.02 line=0.350621 => total=214	GMT -5:00
12/15/23 13:01:52.546	Information	[Instrument Response] 8 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/15/23 13:01:52.561	Information	[Instrument Response] 8 Progress 0% Initializing	GMT -5:00
12/15/23 13:01:52.577	Information	[Instrument Response] 8 Status 0 Preparation complete	GMT -5:00
12/15/23 13:01:55.483	Information	[Instrument Response] 8 Status 0 Motors initialized	GMT -5:00
12/15/23 13:01:55.530	Information	[Instrument Response] 8 Status 0 X-Motor configured	GMT -5:00
12/15/23 13:01:55.530	Information	[Instrument Response] 8 Status 0 DSP configured	GMT -5:00
12/15/23 13:01:55.530	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 13:01:55.577	Information	[Instrument Response] 8 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 13:01:55.577	Information	[Instrument Response] 8 Status 0 Cooling	GMT -5:00
12/15/23 13:01:55.577	Information	[Instrument Response] 8 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 13:01:55.577	Information	[Instrument Response] 8 Status 0 Create MergeImageChannel: 700 20 0 0 284 15 284 302 xofs=1 yofs=0	GMT -5:00
12/15/23 13:01:55.577	Information	[Instrument Response] 8 Status 0 Create MergeImageChannel: 800 20 0 0 284 15 284 302 xofs=0 yofs=0	GMT -5:00
12/15/23 13:01:55.577	Information	[Instrument Response] 8 Status 0 start=34 scan_lines=15 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 13:01:55.577	Information	[Instrument Response] 8 Status 0 start=34 scan_lines=30 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 13:01:55.577	Information	[Instrument Response] 8 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 13:01:56.296	Information	[Instrument Response] 8 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/15/23 13:01:56.483	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         100 V  	GMT -5:00
12/15/23 13:01:56.546	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         159 V  	GMT -5:00
12/15/23 13:01:56.546	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:01:56.546	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:01:57.483	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:01:57.546	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:02:00.505	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:02:01.505	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/15/23 13:02:01.567	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:02:03.520	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:02:04.520	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:02:04.582	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         106 V  	GMT -5:00
12/15/23 13:02:04.582	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:02:04.582	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:02:04.582	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:02:05.520	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:02:11.434	Information	[Instrument Response] 8 ImageBlock 700 0 0 284 15 284 302 8520	GMT -5:00
12/15/23 13:02:11.434	Information	[Instrument Response] 8 ImageBlock 800 0 0 284 15 284 302 8520	GMT -5:00
12/15/23 13:02:11.449	Information	Transferring image block at (0,0) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:02:11.481	Information	Transferring image block at (0,0) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:02:13.574	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      305 mA 	GMT -5:00
12/15/23 13:02:17.577	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         111 V  	GMT -5:00
12/15/23 13:02:17.640	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:02:17.640	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:02:17.640	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:02:18.592	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:02:24.614	Information	[Instrument Response] 8 ImageBlock 700 0 15 284 15 284 302 8520	GMT -5:00
12/15/23 13:02:24.614	Information	[Instrument Response] 8 ImageBlock 800 0 15 284 15 284 302 8520	GMT -5:00
12/15/23 13:02:24.614	Information	Transferring image block at (0,15) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:02:24.645	Information	Transferring image block at (0,15) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:02:25.395	Information	[Instrument Response] 8 Progress 10% 4 minutes	GMT -5:00
12/15/23 13:02:27.621	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:02:27.684	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:02:28.621	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:02:28.684	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:02:30.636	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:02:30.699	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         119 V  	GMT -5:00
12/15/23 13:02:30.699	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:02:30.699	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:02:31.645	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:02:37.786	Information	[Instrument Response] 8 ImageBlock 700 0 30 284 15 284 302 8520	GMT -5:00
12/15/23 13:02:37.786	Information	[Instrument Response] 8 ImageBlock 800 0 30 284 15 284 302 8520	GMT -5:00
12/15/23 13:02:37.786	Information	Transferring image block at (0,30) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:02:37.817	Information	Transferring image block at (0,30) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:02:42.686	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:02:42.733	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:02:43.686	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         128 V  	GMT -5:00
12/15/23 13:02:43.748	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:02:43.748	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:02:43.748	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:02:43.748	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:02:43.748	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:02:44.693	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/15/23 13:02:44.740	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:02:45.693	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:02:50.983	Information	[Instrument Response] 8 ImageBlock 700 0 45 284 15 284 302 8520	GMT -5:00
12/15/23 13:02:50.983	Information	[Instrument Response] 8 ImageBlock 800 0 45 284 15 284 302 8520	GMT -5:00
12/15/23 13:02:50.999	Information	Transferring image block at (0,45) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:02:51.030	Information	Transferring image block at (0,45) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:02:54.727	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:02:54.790	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:02:55.508	Information	[Instrument Response] 8 Progress 22% 3 minutes, 30 seconds	GMT -5:00
12/15/23 13:02:55.742	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:02:55.789	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:02:56.743	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 13:02:56.790	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         141 V  	GMT -5:00
12/15/23 13:02:56.790	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:02:56.790	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:02:56.790	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:02:57.743	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          86 V  	GMT -5:00
12/15/23 13:02:57.805	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:02:58.742	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:03:04.155	Information	[Instrument Response] 8 ImageBlock 700 0 60 284 15 284 302 8520	GMT -5:00
12/15/23 13:03:04.155	Information	[Instrument Response] 8 ImageBlock 800 0 60 284 15 284 302 8520	GMT -5:00
12/15/23 13:03:04.170	Information	Transferring image block at (0,60) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:03:04.186	Information	Transferring image block at (0,60) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:03:09.785	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:03:10.801	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          88 V  	GMT -5:00
12/15/23 13:03:10.848	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:03:10.848	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:03:10.848	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:03:10.848	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:03:11.803	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:03:12.803	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/15/23 13:03:17.351	Information	[Instrument Response] 8 ImageBlock 700 0 75 284 15 284 302 8520	GMT -5:00
12/15/23 13:03:17.351	Information	[Instrument Response] 8 ImageBlock 800 0 75 284 15 284 302 8520	GMT -5:00
12/15/23 13:03:17.351	Information	Transferring image block at (0,75) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:03:17.382	Information	Transferring image block at (0,75) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:03:23.853	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:03:23.900	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:03:23.900	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:03:23.900	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:03:25.719	Information	[Instrument Response] 8 Progress 33% 3 minutes	GMT -5:00
12/15/23 13:03:30.525	Information	[Instrument Response] 8 ImageBlock 700 0 90 284 15 284 302 8520	GMT -5:00
12/15/23 13:03:30.525	Information	[Instrument Response] 8 ImageBlock 800 0 90 284 15 284 302 8520	GMT -5:00
12/15/23 13:03:30.525	Information	Transferring image block at (0,90) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:03:30.556	Information	Transferring image block at (0,90) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:03:33.884	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:03:34.885	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:03:36.900	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:03:36.947	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:03:36.947	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:03:36.947	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:03:37.910	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:03:43.690	Information	[Instrument Response] 8 ImageBlock 700 0 105 284 15 284 302 8520	GMT -5:00
12/15/23 13:03:43.690	Information	[Instrument Response] 8 ImageBlock 800 0 105 284 15 284 302 8520	GMT -5:00
12/15/23 13:03:43.690	Information	Transferring image block at (0,105) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:03:43.721	Information	Transferring image block at (0,105) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:03:45.933	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:03:46.949	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:03:49.959	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:03:50.006	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         106 V  	GMT -5:00
12/15/23 13:03:50.006	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:03:50.006	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:03:50.959	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:03:55.426	Information	[Instrument Response] 8 Progress 44% 2 minutes, 30 seconds	GMT -5:00
12/15/23 13:03:56.876	Information	[Instrument Response] 8 ImageBlock 700 0 120 284 15 284 302 8520	GMT -5:00
12/15/23 13:03:56.876	Information	[Instrument Response] 8 ImageBlock 800 0 120 284 15 284 302 8520	GMT -5:00
12/15/23 13:03:56.876	Information	Transferring image block at (0,120) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:03:56.907	Information	Transferring image block at (0,120) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:04:01.005	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:04:02.013	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:04:03.013	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         113 V  	GMT -5:00
12/15/23 13:04:03.060	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:04:03.060	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:04:03.060	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:04:04.013	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:04:10.048	Information	[Instrument Response] 8 ImageBlock 700 0 135 284 15 284 302 8520	GMT -5:00
12/15/23 13:04:10.048	Information	[Instrument Response] 8 ImageBlock 800 0 135 284 15 284 302 8520	GMT -5:00
12/15/23 13:04:10.048	Information	Transferring image block at (0,135) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:04:10.079	Information	Transferring image block at (0,135) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:04:13.064	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:04:14.070	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:04:16.073	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:04:16.120	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         119 V  	GMT -5:00
12/15/23 13:04:16.120	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:04:16.120	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:04:17.073	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:04:23.229	Information	[Instrument Response] 8 ImageBlock 700 0 150 284 15 284 302 8520	GMT -5:00
12/15/23 13:04:23.229	Information	[Instrument Response] 8 ImageBlock 800 0 150 284 15 284 302 8520	GMT -5:00
12/15/23 13:04:23.229	Information	Transferring image block at (0,150) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:04:23.260	Information	Transferring image block at (0,150) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:04:25.104	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:04:25.166	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:04:25.513	Information	[Instrument Response] 8 Progress 56% 2 minutes	GMT -5:00
12/15/23 13:04:26.123	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:04:26.170	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:04:28.117	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:04:28.180	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:04:29.132	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         128 V  	GMT -5:00
12/15/23 13:04:29.179	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:04:29.179	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:04:29.179	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:04:29.179	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:04:29.179	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:04:30.132	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/15/23 13:04:30.179	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:04:31.139	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:04:35.710	Information	[Instrument Response] 8 Progress 59% 1 minute, 50 seconds	GMT -5:00
12/15/23 13:04:36.413	Information	[Instrument Response] 8 ImageBlock 700 0 165 284 15 284 302 8520	GMT -5:00
12/15/23 13:04:36.413	Information	[Instrument Response] 8 ImageBlock 800 0 165 284 15 284 302 8520	GMT -5:00
12/15/23 13:04:36.413	Information	Transferring image block at (0,165) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:04:36.444	Information	Transferring image block at (0,165) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:04:40.172	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:04:41.187	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:04:42.181	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 13:04:42.243	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         140 V  	GMT -5:00
12/15/23 13:04:42.243	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:04:42.243	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:04:42.243	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:04:43.185	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          86 V  	GMT -5:00
12/15/23 13:04:43.247	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:04:44.200	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:04:45.528	Information	[Instrument Response] 8 Progress 63% 1 minute, 40 seconds	GMT -5:00
12/15/23 13:04:49.596	Information	[Instrument Response] 8 ImageBlock 700 0 180 284 15 284 302 8520	GMT -5:00
12/15/23 13:04:49.596	Information	[Instrument Response] 8 ImageBlock 800 0 180 284 15 284 302 8520	GMT -5:00
12/15/23 13:04:49.596	Information	Transferring image block at (0,180) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:04:49.627	Information	Transferring image block at (0,180) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:04:52.221	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:04:52.283	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:04:53.221	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:04:53.283	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:04:55.236	Information	[Instrument Response] 8 Progress 67% 1 minute, 30 seconds	GMT -5:00
12/15/23 13:04:55.283	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         113 V  	GMT -5:00
12/15/23 13:04:55.283	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:04:55.283	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          36 mA 	GMT -5:00
12/15/23 13:04:56.236	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          87 V  	GMT -5:00
12/15/23 13:04:56.298	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:04:56.298	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:04:57.251	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:04:57.298	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:05:02.788	Information	[Instrument Response] 8 ImageBlock 700 0 195 284 15 284 302 8520	GMT -5:00
12/15/23 13:05:02.788	Information	[Instrument Response] 8 ImageBlock 800 0 195 284 15 284 302 8520	GMT -5:00
12/15/23 13:05:02.788	Information	Transferring image block at (0,195) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:05:02.835	Information	Transferring image block at (0,195) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:05:05.446	Information	[Instrument Response] 8 Progress 71% 1 minute, 20 seconds	GMT -5:00
12/15/23 13:05:07.278	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:05:08.289	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:05:09.289	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:05:09.351	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:05:09.351	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:05:09.351	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:05:15.663	Information	[Instrument Response] 8 Progress 74% 1 minute, 10 seconds	GMT -5:00
12/15/23 13:05:15.991	Information	[Instrument Response] 8 ImageBlock 700 0 210 284 15 284 302 8520	GMT -5:00
12/15/23 13:05:15.991	Information	[Instrument Response] 8 ImageBlock 800 0 210 284 15 284 302 8520	GMT -5:00
12/15/23 13:05:15.991	Information	Transferring image block at (0,210) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:05:16.037	Information	Transferring image block at (0,210) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:05:22.341	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         102 V  	GMT -5:00
12/15/23 13:05:22.403	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:05:22.403	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:05:22.403	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:05:23.356	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:05:24.356	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:05:25.363	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:05:25.503	Information	[Instrument Response] 8 Progress 78% 1 minute	GMT -5:00
12/15/23 13:05:27.372	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:05:28.372	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:05:29.200	Information	[Instrument Response] 8 ImageBlock 700 0 225 284 15 284 302 8520	GMT -5:00
12/15/23 13:05:29.200	Information	[Instrument Response] 8 ImageBlock 800 0 225 284 15 284 302 8520	GMT -5:00
12/15/23 13:05:29.200	Information	Transferring image block at (0,225) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:05:29.231	Information	Transferring image block at (0,225) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:05:34.847	Information	[Instrument Response] 8 Progress 82% 50 seconds	GMT -5:00
12/15/23 13:05:35.394	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:05:35.456	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         110 V  	GMT -5:00
12/15/23 13:05:35.456	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:05:35.456	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:05:36.403	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:05:39.419	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:05:40.418	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:05:42.387	Information	[Instrument Response] 8 ImageBlock 700 0 240 284 15 284 302 8520	GMT -5:00
12/15/23 13:05:42.387	Information	[Instrument Response] 8 ImageBlock 800 0 240 284 15 284 302 8520	GMT -5:00
12/15/23 13:05:42.402	Information	Transferring image block at (0,240) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:05:42.433	Information	Transferring image block at (0,240) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:05:42.496	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:05:43.433	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:05:45.793	Information	[Instrument Response] 8 Progress 86% 40 seconds	GMT -5:00
12/15/23 13:05:48.458	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         117 V  	GMT -5:00
12/15/23 13:05:48.505	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:05:48.505	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:05:48.505	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:05:49.457	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:05:54.478	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:05:55.478	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:05:55.588	Information	[Instrument Response] 8 ImageBlock 700 0 255 284 15 284 302 8520	GMT -5:00
12/15/23 13:05:55.588	Information	[Instrument Response] 8 ImageBlock 800 0 255 284 15 284 302 8520	GMT -5:00
12/15/23 13:05:55.588	Information	Transferring image block at (0,255) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:05:55.619	Information	Transferring image block at (0,255) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:05:55.665	Information	[Instrument Response] 8 Progress 89% 30 seconds	GMT -5:00
12/15/23 13:05:57.493	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:05:58.493	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:06:01.508	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:06:01.555	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         127 V  	GMT -5:00
12/15/23 13:06:01.555	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:06:01.555	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:06:01.555	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:06:02.510	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:06:02.572	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:06:03.509	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:06:05.821	Information	[Instrument Response] 8 Progress 93% 20 seconds	GMT -5:00
12/15/23 13:06:08.774	Information	[Instrument Response] 8 ImageBlock 700 0 270 284 15 284 302 8520	GMT -5:00
12/15/23 13:06:08.774	Information	[Instrument Response] 8 ImageBlock 800 0 270 284 15 284 302 8520	GMT -5:00
12/15/23 13:06:08.774	Information	Transferring image block at (0,270) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:06:08.805	Information	Transferring image block at (0,270) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:06:09.539	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:06:09.586	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:06:10.539	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:06:10.602	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:06:14.401	Information	[Instrument Response] 8 Progress 97% 10 seconds	GMT -5:00
12/15/23 13:06:14.557	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:06:14.620	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:06:14.620	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:06:14.620	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:06:14.620	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:06:14.620	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:06:15.557	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          88 V  	GMT -5:00
12/15/23 13:06:15.620	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:06:15.620	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:06:16.572	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:06:21.962	Information	[Instrument Response] 8 ImageBlock 700 0 285 284 15 284 302 8520	GMT -5:00
12/15/23 13:06:21.962	Information	[Instrument Response] 8 ImageBlock 800 0 285 284 15 284 302 8520	GMT -5:00
12/15/23 13:06:21.962	Information	Transferring image block at (0,285) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:06:21.993	Information	Transferring image block at (0,285) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:06:23.602	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:06:23.649	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:06:23.649	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:06:23.649	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:06:25.493	Information	[Instrument Response] 8 Progress 100% 0 seconds	GMT -5:00
12/15/23 13:06:25.508	Information	[Instrument Response] 8 ImageBlock 700 0 300 284 2 284 302 1136	GMT -5:00
12/15/23 13:06:25.508	Information	[Instrument Response] 8 ImageBlock 800 0 300 284 2 284 302 1136	GMT -5:00
12/15/23 13:06:25.508	Information	[Instrument Response] 8 Status 0 COMPLETE SCAN 269.184 seconds	GMT -5:00
12/15/23 13:06:25.508	Information	Transferring image block at (0,300) with size 284x2 to the 700 channel.	GMT -5:00
12/15/23 13:06:25.508	Information	[Instrument Response] 8 Status 0 Averages (sec): setup=2.00005 line=0.37282 => total=228	GMT -5:00
12/15/23 13:06:25.539	Information	Transferring image block at (0,300) with size 284x2 to the 800 channel.	GMT -5:00
12/15/23 13:06:25.539	Information	[Instrument Response] 8 Status 0 Comparisons: setup=1.00997 line=0.940458	GMT -5:00
12/15/23 13:06:25.586	Information	[Instrument Response] 8 Status 0 Finalize motors	GMT -5:00
12/15/23 13:06:25.586	Information	[Instrument Response] 8 Progress 100% Scan complete	GMT -5:00
12/15/23 13:06:25.618	Information	[Instrument Response] 8 Result 0 OK	GMT -5:00
12/15/23 13:06:25.618	Information	Acquisition 0000006_01 finished with channels: 700 800.	GMT -5:00
12/15/23 13:06:25.899	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/15/23 13:06:25.914	Information	Scan completed successfully	GMT -5:00
12/15/23 13:06:25.945	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         147 V  	GMT -5:00
12/15/23 13:06:26.024	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          92 V  	GMT -5:00
12/15/23 13:06:26.039	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 13:06:26.039	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 13:06:26.055	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 13:06:26.606	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/15/23 13:06:26.668	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          53 V  	GMT -5:00
12/15/23 13:06:27.622	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          52 V  	GMT -5:00
12/15/23 13:06:27.672	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          32 V  	GMT -5:00
12/15/23 13:06:28.627	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          33 V  	GMT -5:00
12/15/23 13:06:28.682	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          20 V  	GMT -5:00
12/15/23 13:06:29.618	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          22 V  	GMT -5:00
12/15/23 13:06:29.681	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          13 V  	GMT -5:00
12/15/23 13:06:30.666	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          14 V  	GMT -5:00
12/15/23 13:06:30.720	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           9 V  	GMT -5:00
12/15/23 13:06:31.638	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          10 V  	GMT -5:00
12/15/23 13:06:31.685	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           7 V  	GMT -5:00
12/15/23 13:06:32.639	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           7 V  	GMT -5:00
12/15/23 13:06:32.686	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           4 V  	GMT -5:00
12/15/23 13:06:33.710	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/15/23 13:06:34.643	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/15/23 13:06:34.690	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      244 mA 	GMT -5:00
12/15/23 13:06:35.647	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/15/23 13:06:38.655	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 13:06:40.664	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/15/23 13:06:59.743	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:07:33.706	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/15/23 13:08:10.787	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 13:08:18.777	Information	Starting acquisition...	GMT -5:00
12/15/23 13:08:20.386	Information	[Instrument Request]  9 ScanAuto 169um medium 0.0 0.0 4.8 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:08:20.433	Information	[Instrument Response] 9 Status 1 Executing command: ScanAuto 169um medium 0 0 4.8 0 5.1 700 800	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Status 0 X mapping: [x0=0,x1=4.8] => [left=-60,right=2220] w=284	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Status 0 Seconds/line=0.330775	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Status 0 Blocks=21	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Status 0 Estimates (sec): setup=2.02 line=0.350621 => total=214	GMT -5:00
12/15/23 13:08:20.448	Information	[Instrument Response] 9 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/15/23 13:08:20.464	Information	[Instrument Response] 9 Progress 0% Initializing	GMT -5:00
12/15/23 13:08:20.464	Information	[Instrument Response] 9 Status 0 Preparation complete	GMT -5:00
12/15/23 13:08:23.402	Information	[Instrument Response] 9 Status 0 Motors initialized	GMT -5:00
12/15/23 13:08:23.449	Information	[Instrument Response] 9 Status 0 X-Motor configured	GMT -5:00
12/15/23 13:08:23.449	Information	[Instrument Response] 9 Status 0 DSP configured	GMT -5:00
12/15/23 13:08:23.496	Information	[Instrument Response] 9 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 13:08:23.496	Information	[Instrument Response] 9 Status 0 Cooling	GMT -5:00
12/15/23 13:08:23.496	Information	[Instrument Response] 9 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 13:08:23.496	Information	[Instrument Response] 9 Status 0 Create MergeImageChannel: 700 20 0 0 284 15 284 302 xofs=1 yofs=0	GMT -5:00
12/15/23 13:08:23.496	Information	[Instrument Response] 9 Status 0 Create MergeImageChannel: 800 20 0 0 284 15 284 302 xofs=0 yofs=0	GMT -5:00
12/15/23 13:08:23.496	Information	[Instrument Response] 9 Status 0 start=34 scan_lines=15 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 13:08:23.496	Information	[Instrument Response] 9 Status 0 start=34 scan_lines=30 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 13:08:23.496	Information	[Instrument Response] 9 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 13:08:24.058	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 13:08:24.121	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:08:26.067	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:08:27.083	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:08:27.442	Information	[Instrument Response] 9 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/15/23 13:08:28.082	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:08:28.129	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:08:28.129	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:08:28.129	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:08:28.129	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:08:32.098	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/15/23 13:08:35.101	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         151 V  	GMT -5:00
12/15/23 13:08:35.164	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:08:35.164	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         146 V  	GMT -5:00
12/15/23 13:08:35.164	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:08:35.164	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:08:35.164	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:08:36.117	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:08:36.164	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          86 V  	GMT -5:00
12/15/23 13:08:36.164	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:08:37.127	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:08:40.126	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:08:41.141	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      305 mA 	GMT -5:00
12/15/23 13:08:42.583	Information	[Instrument Response] 9 ImageBlock 700 0 0 284 15 284 302 8520	GMT -5:00
12/15/23 13:08:42.583	Information	[Instrument Response] 9 ImageBlock 800 0 0 284 15 284 302 8520	GMT -5:00
12/15/23 13:08:42.583	Information	Transferring image block at (0,0) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:08:42.614	Information	Transferring image block at (0,0) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:08:49.173	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:08:49.235	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:08:49.235	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:08:49.235	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:08:50.172	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:08:55.756	Information	[Instrument Response] 9 ImageBlock 700 0 15 284 15 284 302 8520	GMT -5:00
12/15/23 13:08:55.756	Information	[Instrument Response] 9 ImageBlock 800 0 15 284 15 284 302 8520	GMT -5:00
12/15/23 13:08:55.756	Information	Transferring image block at (0,15) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:08:55.787	Information	Transferring image block at (0,15) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:08:56.537	Information	[Instrument Response] 9 Progress 10% 4 minutes	GMT -5:00
12/15/23 13:09:02.233	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:09:02.280	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          97 V  	GMT -5:00
12/15/23 13:09:02.280	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:09:02.280	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:09:05.240	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:09:08.949	Information	[Instrument Response] 9 ImageBlock 700 0 30 284 15 284 302 8520	GMT -5:00
12/15/23 13:09:08.949	Information	[Instrument Response] 9 ImageBlock 800 0 30 284 15 284 302 8520	GMT -5:00
12/15/23 13:09:08.949	Information	Transferring image block at (0,30) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:09:08.981	Information	Transferring image block at (0,30) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:09:15.284	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         103 V  	GMT -5:00
12/15/23 13:09:15.331	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:09:15.331	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:09:15.331	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:09:16.283	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:09:22.129	Information	[Instrument Response] 9 ImageBlock 700 0 45 284 15 284 302 8520	GMT -5:00
12/15/23 13:09:22.129	Information	[Instrument Response] 9 ImageBlock 800 0 45 284 15 284 302 8520	GMT -5:00
12/15/23 13:09:22.145	Information	Transferring image block at (0,45) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:09:22.160	Information	Transferring image block at (0,45) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:09:26.655	Information	[Instrument Response] 9 Progress 22% 3 minutes, 30 seconds	GMT -5:00
12/15/23 13:09:28.333	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:09:28.396	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         110 V  	GMT -5:00
12/15/23 13:09:28.396	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:09:28.396	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:09:29.333	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:09:32.347	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:09:33.362	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:09:35.331	Information	[Instrument Response] 9 ImageBlock 700 0 60 284 15 284 302 8520	GMT -5:00
12/15/23 13:09:35.331	Information	[Instrument Response] 9 ImageBlock 800 0 60 284 15 284 302 8520	GMT -5:00
12/15/23 13:09:35.331	Information	Transferring image block at (0,60) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:09:35.347	Information	Transferring image block at (0,60) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:09:35.409	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:09:36.362	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:09:41.381	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         117 V  	GMT -5:00
12/15/23 13:09:41.444	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:09:41.444	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:09:41.444	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:09:41.444	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:09:42.391	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:09:45.408	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:09:47.408	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:09:48.408	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:09:48.517	Information	[Instrument Response] 9 ImageBlock 700 0 75 284 15 284 302 8520	GMT -5:00
12/15/23 13:09:48.517	Information	[Instrument Response] 9 ImageBlock 800 0 75 284 15 284 302 8520	GMT -5:00
12/15/23 13:09:48.517	Information	Transferring image block at (0,75) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:09:48.548	Information	Transferring image block at (0,75) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:09:53.442	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:09:54.441	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:09:54.488	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         126 V  	GMT -5:00
12/15/23 13:09:54.488	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:09:54.488	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:09:55.441	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:09:56.441	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:09:56.878	Information	[Instrument Response] 9 Progress 33% 3 minutes	GMT -5:00
12/15/23 13:10:01.690	Information	[Instrument Response] 9 ImageBlock 700 0 90 284 15 284 302 8520	GMT -5:00
12/15/23 13:10:01.690	Information	[Instrument Response] 9 ImageBlock 800 0 90 284 15 284 302 8520	GMT -5:00
12/15/23 13:10:01.690	Information	Transferring image block at (0,90) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:10:01.721	Information	Transferring image block at (0,90) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:10:02.480	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:10:02.527	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:10:02.527	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:10:03.480	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:10:03.527	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:10:03.527	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:10:07.495	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         136 V  	GMT -5:00
12/15/23 13:10:07.542	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:10:07.542	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:10:07.542	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:10:07.542	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:10:07.542	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:10:08.504	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:10:08.550	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:10:09.503	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:10:12.518	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:10:14.528	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:10:14.871	Information	[Instrument Response] 9 ImageBlock 700 0 105 284 15 284 302 8520	GMT -5:00
12/15/23 13:10:14.871	Information	[Instrument Response] 9 ImageBlock 800 0 105 284 15 284 302 8520	GMT -5:00
12/15/23 13:10:14.871	Information	Transferring image block at (0,105) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:10:14.902	Information	Transferring image block at (0,105) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:10:15.528	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:10:20.551	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         151 V  	GMT -5:00
12/15/23 13:10:20.598	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:10:20.598	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         149 V  	GMT -5:00
12/15/23 13:10:20.598	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:10:20.598	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:10:20.598	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:10:21.551	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:10:21.598	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          87 V  	GMT -5:00
12/15/23 13:10:21.598	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:10:22.552	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:10:26.976	Information	[Instrument Response] 9 Progress 44% 2 minutes, 30 seconds	GMT -5:00
12/15/23 13:10:28.057	Information	[Instrument Response] 9 ImageBlock 700 0 120 284 15 284 302 8520	GMT -5:00
12/15/23 13:10:28.057	Information	[Instrument Response] 9 ImageBlock 800 0 120 284 15 284 302 8520	GMT -5:00
12/15/23 13:10:28.057	Information	Transferring image block at (0,120) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:10:28.088	Information	Transferring image block at (0,120) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:10:34.602	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          92 V  	GMT -5:00
12/15/23 13:10:34.664	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:10:34.664	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:10:34.664	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:10:35.617	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:10:41.245	Information	[Instrument Response] 9 ImageBlock 700 0 135 284 15 284 302 8520	GMT -5:00
12/15/23 13:10:41.245	Information	[Instrument Response] 9 ImageBlock 800 0 135 284 15 284 302 8520	GMT -5:00
12/15/23 13:10:41.260	Information	Transferring image block at (0,135) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:10:41.276	Information	Transferring image block at (0,135) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:10:41.635	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:10:42.639	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:10:47.653	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:10:47.716	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:10:47.716	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:10:47.716	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:10:48.668	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:10:54.433	Information	[Instrument Response] 9 ImageBlock 700 0 150 284 15 284 302 8520	GMT -5:00
12/15/23 13:10:54.433	Information	[Instrument Response] 9 ImageBlock 800 0 150 284 15 284 302 8520	GMT -5:00
12/15/23 13:10:54.433	Information	Transferring image block at (0,150) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:10:54.464	Information	Transferring image block at (0,150) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:10:56.698	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:10:56.761	Information	[Instrument Response] 9 Progress 56% 2 minutes	GMT -5:00
12/15/23 13:10:57.698	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:11:00.713	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         105 V  	GMT -5:00
12/15/23 13:11:00.775	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:11:00.775	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:11:00.775	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:11:01.728	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:11:06.900	Information	[Instrument Response] 9 Progress 59% 1 minute, 50 seconds	GMT -5:00
12/15/23 13:11:07.603	Information	[Instrument Response] 9 ImageBlock 700 0 165 284 15 284 302 8520	GMT -5:00
12/15/23 13:11:07.603	Information	[Instrument Response] 9 ImageBlock 800 0 165 284 15 284 302 8520	GMT -5:00
12/15/23 13:11:07.618	Information	Transferring image block at (0,165) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:11:07.634	Information	Transferring image block at (0,165) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:11:13.762	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:11:13.824	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         113 V  	GMT -5:00
12/15/23 13:11:13.824	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:11:13.824	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:11:14.777	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:11:16.731	Information	[Instrument Response] 9 Progress 63% 1 minute, 40 seconds	GMT -5:00
12/15/23 13:11:20.784	Information	[Instrument Response] 9 ImageBlock 700 0 180 284 15 284 302 8520	GMT -5:00
12/15/23 13:11:20.784	Information	[Instrument Response] 9 ImageBlock 800 0 180 284 15 284 302 8520	GMT -5:00
12/15/23 13:11:20.800	Information	Transferring image block at (0,180) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:11:20.831	Information	Transferring image block at (0,180) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:11:20.831	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:11:21.803	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:11:23.803	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:11:24.818	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:11:26.421	Information	[Instrument Response] 9 Progress 67% 1 minute, 30 seconds	GMT -5:00
12/15/23 13:11:26.827	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         118 V  	GMT -5:00
12/15/23 13:11:26.874	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:11:26.874	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:11:26.874	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:11:26.874	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:11:27.827	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:11:31.843	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:11:33.858	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/15/23 13:11:33.967	Information	[Instrument Response] 9 ImageBlock 700 0 195 284 15 284 302 8520	GMT -5:00
12/15/23 13:11:33.983	Information	[Instrument Response] 9 ImageBlock 800 0 195 284 15 284 302 8520	GMT -5:00
12/15/23 13:11:33.983	Information	Transferring image block at (0,195) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:11:33.998	Information	Transferring image block at (0,195) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:11:35.873	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:11:35.920	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:11:36.623	Information	[Instrument Response] 9 Progress 71% 1 minute, 20 seconds	GMT -5:00
12/15/23 13:11:36.873	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:11:39.885	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:11:39.932	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         128 V  	GMT -5:00
12/15/23 13:11:39.932	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:11:39.932	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:11:39.932	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:11:40.885	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/15/23 13:11:40.947	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:11:41.885	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:11:46.818	Information	[Instrument Response] 9 Progress 74% 1 minute, 10 seconds	GMT -5:00
12/15/23 13:11:47.162	Information	[Instrument Response] 9 ImageBlock 700 0 210 284 15 284 302 8520	GMT -5:00
12/15/23 13:11:47.162	Information	[Instrument Response] 9 ImageBlock 800 0 210 284 15 284 302 8520	GMT -5:00
12/15/23 13:11:47.162	Information	Transferring image block at (0,210) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:11:47.177	Information	Transferring image block at (0,210) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:11:47.912	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:11:48.912	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:11:52.934	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         138 V  	GMT -5:00
12/15/23 13:11:52.996	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:11:52.996	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:11:52.996	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:11:52.996	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:11:53.934	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          88 V  	GMT -5:00
12/15/23 13:11:53.996	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:11:54.950	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:11:56.646	Information	[Instrument Response] 9 Progress 78% 1 minute	GMT -5:00
12/15/23 13:12:00.338	Information	[Instrument Response] 9 ImageBlock 700 0 225 284 15 284 302 8520	GMT -5:00
12/15/23 13:12:00.338	Information	[Instrument Response] 9 ImageBlock 800 0 225 284 15 284 302 8520	GMT -5:00
12/15/23 13:12:00.338	Information	Transferring image block at (0,225) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:12:00.369	Information	Transferring image block at (0,225) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:12:02.978	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:12:03.978	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:12:05.978	Information	[Instrument Response] 9 Progress 82% 50 seconds	GMT -5:00
12/15/23 13:12:06.025	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:12:06.025	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         151 V  	GMT -5:00
12/15/23 13:12:06.025	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:12:06.025	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:12:06.025	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:12:06.993	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 13:12:07.056	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:12:07.993	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:12:13.512	Information	[Instrument Response] 9 ImageBlock 700 0 240 284 15 284 302 8520	GMT -5:00
12/15/23 13:12:13.512	Information	[Instrument Response] 9 ImageBlock 800 0 240 284 15 284 302 8520	GMT -5:00
12/15/23 13:12:13.512	Information	Transferring image block at (0,240) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:12:13.544	Information	Transferring image block at (0,240) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:12:15.028	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:12:16.044	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:12:16.903	Information	[Instrument Response] 9 Progress 86% 40 seconds	GMT -5:00
12/15/23 13:12:20.056	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/15/23 13:12:20.103	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:12:20.103	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:12:20.103	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:12:21.056	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:12:26.696	Information	[Instrument Response] 9 ImageBlock 700 0 255 284 15 284 302 8520	GMT -5:00
12/15/23 13:12:26.696	Information	[Instrument Response] 9 ImageBlock 800 0 255 284 15 284 302 8520	GMT -5:00
12/15/23 13:12:26.696	Information	Transferring image block at (0,255) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:12:26.727	Information	Transferring image block at (0,255) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:12:26.789	Information	[Instrument Response] 9 Progress 89% 30 seconds	GMT -5:00
12/15/23 13:12:27.086	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:12:28.086	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:12:33.109	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:12:33.156	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          98 V  	GMT -5:00
12/15/23 13:12:33.156	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:12:33.156	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:12:34.109	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:12:36.922	Information	[Instrument Response] 9 Progress 93% 20 seconds	GMT -5:00
12/15/23 13:12:39.127	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:12:39.189	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:12:39.861	Information	[Instrument Response] 9 ImageBlock 700 0 270 284 15 284 302 8520	GMT -5:00
12/15/23 13:12:39.861	Information	[Instrument Response] 9 ImageBlock 800 0 270 284 15 284 302 8520	GMT -5:00
12/15/23 13:12:39.861	Information	Transferring image block at (0,270) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:12:39.892	Information	Transferring image block at (0,270) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:12:40.143	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:12:40.189	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:12:42.148	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:12:43.148	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:12:45.491	Information	[Instrument Response] 9 Progress 97% 10 seconds	GMT -5:00
12/15/23 13:12:46.163	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         104 V  	GMT -5:00
12/15/23 13:12:46.210	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:12:46.210	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:12:46.210	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:12:47.169	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:12:51.188	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:12:52.188	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:12:53.031	Information	[Instrument Response] 9 ImageBlock 700 0 285 284 15 284 302 8520	GMT -5:00
12/15/23 13:12:53.031	Information	[Instrument Response] 9 ImageBlock 800 0 285 284 15 284 302 8520	GMT -5:00
12/15/23 13:12:53.047	Information	Transferring image block at (0,285) with size 284x15 to the 700 channel.	GMT -5:00
12/15/23 13:12:53.062	Information	Transferring image block at (0,285) with size 284x15 to the 800 channel.	GMT -5:00
12/15/23 13:12:54.203	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:12:54.250	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         120 V  	GMT -5:00
12/15/23 13:12:54.250	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:12:54.250	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:12:55.203	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:12:56.562	Information	[Instrument Response] 9 Progress 100% 0 seconds	GMT -5:00
12/15/23 13:12:56.577	Information	[Instrument Response] 9 ImageBlock 700 0 300 284 2 284 302 1136	GMT -5:00
12/15/23 13:12:56.577	Information	[Instrument Response] 9 ImageBlock 800 0 300 284 2 284 302 1136	GMT -5:00
12/15/23 13:12:56.577	Information	[Instrument Response] 9 Status 0 COMPLETE SCAN 269.115 seconds	GMT -5:00
12/15/23 13:12:56.577	Information	Transferring image block at (0,300) with size 284x2 to the 700 channel.	GMT -5:00
12/15/23 13:12:56.577	Information	[Instrument Response] 9 Status 0 Averages (sec): setup=2.00005 line=0.372705 => total=228	GMT -5:00
12/15/23 13:12:56.609	Information	Transferring image block at (0,300) with size 284x2 to the 800 channel.	GMT -5:00
12/15/23 13:12:56.624	Information	[Instrument Response] 9 Status 0 Comparisons: setup=1.00997 line=0.940748	GMT -5:00
12/15/23 13:12:56.655	Information	[Instrument Response] 9 Status 0 Finalize motors	GMT -5:00
12/15/23 13:12:56.655	Information	[Instrument Response] 9 Progress 100% Scan complete	GMT -5:00
12/15/23 13:12:56.687	Information	[Instrument Response] 9 Result 0 OK	GMT -5:00
12/15/23 13:12:56.687	Information	Acquisition 0000007_01 finished with channels: 700 800.	GMT -5:00
12/15/23 13:12:56.890	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/15/23 13:12:56.890	Information	Scan completed successfully	GMT -5:00
12/15/23 13:12:57.223	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         107 V  	GMT -5:00
12/15/23 13:12:57.270	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          67 V  	GMT -5:00
12/15/23 13:12:57.270	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 13:12:57.270	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 13:12:57.270	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 13:12:58.213	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          65 V  	GMT -5:00
12/15/23 13:12:58.273	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          41 V  	GMT -5:00
12/15/23 13:12:59.233	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          40 V  	GMT -5:00
12/15/23 13:12:59.293	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          25 V  	GMT -5:00
12/15/23 13:13:00.223	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          26 V  	GMT -5:00
12/15/23 13:13:00.273	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          17 V  	GMT -5:00
12/15/23 13:13:00.273	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/15/23 13:13:01.217	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          18 V  	GMT -5:00
12/15/23 13:13:01.280	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          11 V  	GMT -5:00
12/15/23 13:13:02.349	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          12 V  	GMT -5:00
12/15/23 13:13:02.396	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           8 V  	GMT -5:00
12/15/23 13:13:03.236	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           8 V  	GMT -5:00
12/15/23 13:13:03.283	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/15/23 13:13:04.238	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           6 V  	GMT -5:00
12/15/23 13:13:04.285	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           3 V  	GMT -5:00
12/15/23 13:13:05.240	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           3 V  	GMT -5:00
12/15/23 13:13:07.243	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           1 V  	GMT -5:00
12/15/23 13:13:08.245	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           1 V  	GMT -5:00
12/15/23 13:13:10.266	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      244 mA 	GMT -5:00
12/15/23 13:13:24.903	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/15/23 13:14:12.578	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 13:14:16.828	Information	Starting acquisition...	GMT -5:00
12/15/23 13:14:17.718	Information	[Instrument Request]  10 ScanAuto 169um medium 0.0 0.0 4.8 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:14:17.781	Information	[Instrument Response] 10 Status 1 Executing command: ScanAuto 169um medium 0 0 4.8 0 5.1 700 800	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Status 0 X mapping: [x0=0,x1=4.8] => [left=-60,right=2220] w=284	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Status 0 Seconds/line=0.330775	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Status 0 Blocks=21	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Status 0 Estimates (sec): setup=2.02 line=0.350621 => total=214	GMT -5:00
12/15/23 13:14:17.796	Information	[Instrument Response] 10 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/15/23 13:14:17.812	Information	[Instrument Response] 10 Progress 0% Initializing	GMT -5:00
12/15/23 13:14:17.812	Information	[Instrument Response] 10 Status 0 Preparation complete	GMT -5:00
12/15/23 13:15:58.179	Information	Stopping acquisition...	GMT -5:00
12/15/23 13:15:58.179	Information	[Instrument Request]  11 Stop 	GMT -5:00
12/15/23 13:16:02.886	Error      	The instrument connection was closed. If a connection is expected, check the cable connection and confirm the instrument is powered on.  To reconnect, click the Image Studio Application Button, hover over Instrument, and click Connect.	GMT -5:00
12/15/23 13:16:24.142	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/15/23 13:16:31.877	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/15/23 13:16:31.924	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/15/23 13:16:31.971	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/15/23 13:16:31.971	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/15/23 13:16:31.971	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/15/23 13:16:31.971	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/15/23 13:16:31.971	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/15/23 13:16:31.971	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/15/23 13:16:31.971	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/15/23 13:16:31.971	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/15/23 13:16:31.971	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/15/23 13:16:31.986	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/15/23 13:16:31.986	Information	[Instrument Request]  12 ScanAutoEstimate 169um medium 0.0 0.0 4.8 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:16:34.954	Information	[Instrument Response] 12 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0 4.8 0 5.1 700 800	GMT -5:00
12/15/23 13:16:34.954	Information	[Instrument Response] 12 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:16:34.970	Information	[Instrument Response] 12 Status 0 X mapping: [x0=0,x1=4.8] => [left=-60,right=2220] w=284	GMT -5:00
12/15/23 13:16:34.970	Information	[Instrument Response] 12 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:16:34.970	Information	[Instrument Response] 12 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:16:34.970	Information	[Instrument Response] 12 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:16:34.985	Information	[Instrument Response] 12 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:16:34.985	Information	[Instrument Response] 12 Status 0 Seconds/line=0.330775	GMT -5:00
12/15/23 13:16:35.001	Information	[Instrument Response] 12 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:16:35.001	Information	[Instrument Response] 12 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/15/23 13:16:35.001	Information	[Instrument Response] 12 Status 0 Blocks=21	GMT -5:00
12/15/23 13:16:35.017	Information	[Instrument Response] 12 Status 0 Estimates (sec): setup=2.02 line=0.350621 => total=214	GMT -5:00
12/15/23 13:16:35.017	Information	[Instrument Response] 12 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 12 Result 0 OK	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    666 mV 	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               26 C  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.0 V  	GMT -5:00
12/15/23 13:16:35.032	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 13:16:37.940	Information	[Instrument Request]  13 ScanAutoEstimate 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:16:37.956	Information	[Instrument Response] 13 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Status 0 Blocks=23	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Progress 0% 5 minutes	GMT -5:00
12/15/23 13:16:38.003	Information	[Instrument Response] 13 Result 0 OK	GMT -5:00
12/15/23 13:16:39.990	Information	Starting acquisition...	GMT -5:00
12/15/23 13:16:40.798	Information	[Instrument Request]  14 ScanAuto 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:16:40.845	Information	[Instrument Response] 14 Status 1 Executing command: ScanAuto 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Status 0 Blocks=23	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 13:16:40.861	Information	[Instrument Response] 14 Progress 0% 5 minutes	GMT -5:00
12/15/23 13:16:40.876	Information	[Instrument Response] 14 Progress 0% Initializing	GMT -5:00
12/15/23 13:16:40.892	Information	[Instrument Response] 14 Status 0 Preparation complete	GMT -5:00
12/15/23 13:16:43.813	Information	[Instrument Response] 14 Status 0 Motors initialized	GMT -5:00
12/15/23 13:16:43.860	Information	[Instrument Response] 14 Status 0 X-Motor configured	GMT -5:00
12/15/23 13:16:43.860	Information	[Instrument Response] 14 Status 0 DSP configured	GMT -5:00
12/15/23 13:16:43.907	Information	[Instrument Response] 14 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 13:16:43.907	Information	[Instrument Response] 14 Status 0 Cooling	GMT -5:00
12/15/23 13:16:43.907	Information	[Instrument Response] 14 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 13:16:43.907	Information	[Instrument Response] 14 Status 0 Create MergeImageChannel: 700 20 0 0 319 14 319 302 xofs=1 yofs=0	GMT -5:00
12/15/23 13:16:43.907	Information	[Instrument Response] 14 Status 0 Create MergeImageChannel: 800 20 0 0 319 14 319 302 xofs=0 yofs=0	GMT -5:00
12/15/23 13:16:43.907	Information	[Instrument Response] 14 Status 0 start=34 scan_lines=14 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 13:16:43.907	Information	[Instrument Response] 14 Status 0 start=34 scan_lines=28 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 13:16:43.907	Information	[Instrument Response] 14 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 13:16:43.985	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 13:16:44.048	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:16:44.751	Information	[Instrument Response] 14 Progress 0% 5 minutes	GMT -5:00
12/15/23 13:16:44.985	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          98 V  	GMT -5:00
12/15/23 13:16:45.048	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         157 V  	GMT -5:00
12/15/23 13:16:45.048	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:16:45.048	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:16:45.048	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:16:46.000	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:16:46.047	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:16:50.007	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/15/23 13:16:53.019	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:16:53.082	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         106 V  	GMT -5:00
12/15/23 13:16:53.082	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:16:53.082	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:16:54.019	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:16:59.051	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      315 mA 	GMT -5:00
12/15/23 13:16:59.973	Information	[Instrument Response] 14 ImageBlock 700 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 13:16:59.973	Information	[Instrument Response] 14 ImageBlock 800 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 13:16:59.973	Information	Transferring image block at (0,0) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:17:00.004	Information	Transferring image block at (0,0) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:17:05.649	Information	[Instrument Response] 14 Progress 7% 4 minutes, 30 seconds	GMT -5:00
12/15/23 13:17:06.070	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         117 V  	GMT -5:00
12/15/23 13:17:06.133	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:17:06.133	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:17:06.133	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:17:07.071	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:17:13.240	Information	[Instrument Response] 14 ImageBlock 700 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 13:17:13.240	Information	[Instrument Response] 14 ImageBlock 800 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 13:17:13.240	Information	Transferring image block at (0,14) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:17:13.302	Information	Transferring image block at (0,14) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:17:19.124	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 13:17:19.186	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         134 V  	GMT -5:00
12/15/23 13:17:19.186	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:17:19.186	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:17:19.186	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:17:20.123	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/15/23 13:17:20.186	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:17:21.139	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:17:26.503	Information	[Instrument Response] 14 ImageBlock 700 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 13:17:26.503	Information	[Instrument Response] 14 ImageBlock 800 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 13:17:26.519	Information	Transferring image block at (0,28) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:17:26.534	Information	Transferring image block at (0,28) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:17:28.161	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:17:29.161	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:17:30.161	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:17:31.177	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:17:32.176	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:17:33.180	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          88 V  	GMT -5:00
12/15/23 13:17:33.243	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:17:33.243	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:17:33.243	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:17:33.243	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:17:34.180	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:17:36.195	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:17:36.242	Information	[Instrument Response] 14 Progress 17% 4 minutes	GMT -5:00
12/15/23 13:17:37.195	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:17:38.196	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:17:39.196	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:17:39.772	Information	[Instrument Response] 14 ImageBlock 700 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 13:17:39.772	Information	[Instrument Response] 14 ImageBlock 800 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 13:17:39.787	Information	Transferring image block at (0,42) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:17:39.818	Information	Transferring image block at (0,42) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:17:40.209	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:17:41.209	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:17:42.209	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:17:42.271	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:17:43.225	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:17:43.272	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:17:46.226	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:17:46.289	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:17:46.289	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:17:46.289	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:17:47.242	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:17:50.250	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:17:51.250	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:17:52.250	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:17:53.047	Information	[Instrument Response] 14 ImageBlock 700 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 13:17:53.047	Information	[Instrument Response] 14 ImageBlock 800 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 13:17:53.047	Information	Transferring image block at (0,56) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:17:53.078	Information	Transferring image block at (0,56) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:17:53.265	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:17:56.266	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:17:57.281	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:17:58.187	Error      	The 'Stop' command timed out. The elapsed time exceeded 120 seconds. Check the instrument connection.	GMT -5:00
12/15/23 13:17:59.281	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:17:59.343	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:17:59.343	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:17:59.343	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:18:00.296	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:18:02.296	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:18:04.311	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:18:05.953	Information	[Instrument Response] 14 Progress 28% 3 minutes, 30 seconds	GMT -5:00
12/15/23 13:18:06.312	Information	[Instrument Response] 14 ImageBlock 700 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 13:18:06.312	Information	[Instrument Response] 14 ImageBlock 800 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 13:18:06.328	Information	Transferring image block at (0,70) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:18:06.343	Information	Transferring image block at (0,70) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:18:12.358	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:18:12.405	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/15/23 13:18:12.405	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:18:12.405	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:18:13.358	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:18:14.357	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:18:15.357	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:18:19.591	Information	[Instrument Response] 14 ImageBlock 700 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 13:18:19.591	Information	[Instrument Response] 14 ImageBlock 800 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 13:18:19.591	Information	Transferring image block at (0,84) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:18:19.622	Information	Transferring image block at (0,84) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:18:25.418	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:18:25.480	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:18:25.480	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:18:25.480	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:18:25.480	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:18:26.417	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:18:26.480	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:18:27.433	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:18:28.433	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:18:29.432	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:18:32.853	Information	[Instrument Response] 14 ImageBlock 700 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 13:18:32.853	Information	[Instrument Response] 14 ImageBlock 800 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 13:18:32.853	Information	Transferring image block at (0,98) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:18:32.885	Information	Transferring image block at (0,98) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:18:36.103	Information	[Instrument Response] 14 Progress 38% 3 minutes	GMT -5:00
12/15/23 13:18:38.484	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          55 mA 	GMT -5:00
12/15/23 13:18:39.500	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:18:39.547	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/15/23 13:18:39.547	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:18:39.547	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:18:40.499	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:18:42.499	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:18:43.514	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:18:46.118	Information	[Instrument Response] 14 ImageBlock 700 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 13:18:46.118	Information	[Instrument Response] 14 ImageBlock 800 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 13:18:46.133	Information	Transferring image block at (0,112) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:18:46.165	Information	Transferring image block at (0,112) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:18:52.562	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          98 V  	GMT -5:00
12/15/23 13:18:52.625	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:18:52.625	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:18:52.625	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:18:53.570	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:18:54.570	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:18:56.570	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:18:59.382	Information	[Instrument Response] 14 ImageBlock 700 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 13:18:59.397	Information	[Instrument Response] 14 ImageBlock 800 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 13:18:59.397	Information	Transferring image block at (0,126) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:18:59.429	Information	Transferring image block at (0,126) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:19:05.057	Information	[Instrument Response] 14 Progress 49% 2 minutes, 30 seconds	GMT -5:00
12/15/23 13:19:05.638	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:19:05.685	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:19:05.685	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:19:05.685	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:19:06.639	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:19:08.646	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:19:10.647	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:19:12.657	Information	[Instrument Response] 14 ImageBlock 700 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 13:19:12.657	Information	[Instrument Response] 14 ImageBlock 800 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 13:19:12.672	Information	Transferring image block at (0,140) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:19:12.688	Information	Transferring image block at (0,140) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:19:18.701	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/15/23 13:19:18.748	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:19:18.748	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:19:18.748	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:19:19.701	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:19:20.704	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:19:21.708	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:19:25.932	Information	[Instrument Response] 14 ImageBlock 700 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 13:19:25.932	Information	[Instrument Response] 14 ImageBlock 800 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 13:19:25.932	Information	Transferring image block at (0,154) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:19:25.947	Information	Transferring image block at (0,154) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:19:31.765	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 13:19:31.812	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:19:31.812	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:19:31.812	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:19:31.812	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:19:32.774	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 13:19:32.821	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:19:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:19:34.770	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:19:35.781	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:19:36.013	Information	[Instrument Response] 14 Progress 59% 2 minutes	GMT -5:00
12/15/23 13:19:39.200	Information	[Instrument Response] 14 ImageBlock 700 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 13:19:39.200	Information	[Instrument Response] 14 ImageBlock 800 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 13:19:39.200	Information	Transferring image block at (0,168) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:19:39.231	Information	Transferring image block at (0,168) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:19:44.863	Information	[Instrument Response] 14 Progress 63% 1 minute, 50 seconds	GMT -5:00
12/15/23 13:19:45.832	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:19:45.894	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:19:45.894	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:19:45.894	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:19:46.847	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:19:48.847	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:19:49.856	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:19:52.465	Information	[Instrument Response] 14 ImageBlock 700 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 13:19:52.465	Information	[Instrument Response] 14 ImageBlock 800 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 13:19:52.465	Information	Transferring image block at (0,182) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:19:52.496	Information	Transferring image block at (0,182) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:19:56.130	Information	[Instrument Response] 14 Progress 66% 1 minute, 40 seconds	GMT -5:00
12/15/23 13:19:58.911	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:19:58.958	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:19:58.958	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:19:58.958	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:19:59.911	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:20:00.911	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:20:02.920	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:20:05.733	Information	[Instrument Response] 14 ImageBlock 700 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 13:20:05.733	Information	[Instrument Response] 14 ImageBlock 800 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 13:20:05.733	Information	Transferring image block at (0,196) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:20:05.764	Information	Transferring image block at (0,196) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:20:06.170	Information	[Instrument Response] 14 Progress 70% 1 minute, 30 seconds	GMT -5:00
12/15/23 13:20:11.968	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:20:12.030	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:20:12.030	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:20:12.030	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:20:12.983	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:20:14.983	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:20:15.826	Information	[Instrument Response] 14 Progress 73% 1 minute, 20 seconds	GMT -5:00
12/15/23 13:20:16.986	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:20:19.002	Information	[Instrument Response] 14 ImageBlock 700 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 13:20:19.002	Information	[Instrument Response] 14 ImageBlock 800 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 13:20:19.002	Information	Transferring image block at (0,210) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:20:19.033	Information	Transferring image block at (0,210) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:20:24.672	Information	[Instrument Response] 14 Progress 76% 1 minute, 10 seconds	GMT -5:00
12/15/23 13:20:25.032	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:20:25.094	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/15/23 13:20:25.094	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:20:25.094	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:20:26.047	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:20:27.047	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:20:28.047	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:20:32.265	Information	[Instrument Response] 14 ImageBlock 700 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 13:20:32.265	Information	[Instrument Response] 14 ImageBlock 800 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 13:20:32.280	Information	Transferring image block at (0,224) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:20:32.296	Information	Transferring image block at (0,224) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:20:35.934	Information	[Instrument Response] 14 Progress 80% 1 minute	GMT -5:00
12/15/23 13:20:38.107	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:20:38.153	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:20:38.153	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:20:38.153	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:20:38.153	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:20:39.106	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:20:39.169	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:20:40.112	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:20:41.115	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:20:42.119	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:20:45.540	Information	[Instrument Response] 14 ImageBlock 700 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 13:20:45.540	Information	[Instrument Response] 14 ImageBlock 800 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 13:20:45.540	Information	Transferring image block at (0,238) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:20:45.571	Information	Transferring image block at (0,238) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:20:45.977	Information	[Instrument Response] 14 Progress 84% 50 seconds	GMT -5:00
12/15/23 13:20:52.172	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:20:52.234	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/15/23 13:20:52.234	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:20:52.234	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:20:53.187	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:20:55.190	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:20:56.034	Information	[Instrument Response] 14 Progress 87% 40 seconds	GMT -5:00
12/15/23 13:20:56.190	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:20:58.799	Information	[Instrument Response] 14 ImageBlock 700 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 13:20:58.799	Information	[Instrument Response] 14 ImageBlock 800 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 13:20:58.814	Information	Transferring image block at (0,252) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:20:58.830	Information	Transferring image block at (0,252) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:21:04.472	Information	[Instrument Response] 14 Progress 90% 30 seconds	GMT -5:00
12/15/23 13:21:05.253	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:21:05.300	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:21:05.300	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:21:05.300	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:21:06.253	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:21:07.252	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:21:09.268	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:21:12.069	Information	[Instrument Response] 14 ImageBlock 700 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 13:21:12.069	Information	[Instrument Response] 14 ImageBlock 800 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 13:21:12.085	Information	Transferring image block at (0,266) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:21:12.101	Information	Transferring image block at (0,266) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:21:16.132	Information	[Instrument Response] 14 Progress 94% 20 seconds	GMT -5:00
12/15/23 13:21:18.317	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:21:18.379	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:21:18.379	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:21:18.379	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:21:19.316	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:21:21.332	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:21:23.331	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:21:25.347	Information	[Instrument Response] 14 ImageBlock 700 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 13:21:25.347	Information	[Instrument Response] 14 ImageBlock 800 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 13:21:25.347	Information	Transferring image block at (0,280) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:21:25.378	Information	Transferring image block at (0,280) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:21:26.192	Information	[Instrument Response] 14 Progress 98% 10 seconds	GMT -5:00
12/15/23 13:21:29.366	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:21:29.429	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:21:29.429	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:21:29.429	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:21:30.366	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:21:31.382	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:21:32.382	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:21:33.819	Information	[Instrument Response] 14 Progress 100% 0 seconds	GMT -5:00
12/15/23 13:21:34.398	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:21:34.445	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:21:34.445	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:21:34.445	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:21:35.397	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:21:35.819	Information	[Instrument Response] 14 ImageBlock 700 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 13:21:35.819	Information	[Instrument Response] 14 ImageBlock 800 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 13:21:35.835	Information	Transferring image block at (0,294) with size 319x8 to the 700 channel.	GMT -5:00
12/15/23 13:21:35.835	Information	[Instrument Response] 14 Status 0 COMPLETE SCAN 291.071 seconds	GMT -5:00
12/15/23 13:21:35.850	Information	Transferring image block at (0,294) with size 319x8 to the 800 channel.	GMT -5:00
12/15/23 13:21:35.850	Information	[Instrument Response] 14 Status 0 Averages (sec): setup=2.00005 line=0.402433 => total=246	GMT -5:00
12/15/23 13:21:35.913	Information	[Instrument Response] 14 Status 0 Comparisons: setup=1.00997 line=0.950274	GMT -5:00
12/15/23 13:21:35.913	Information	[Instrument Response] 14 Status 0 Finalize motors	GMT -5:00
12/15/23 13:21:35.913	Information	[Instrument Response] 14 Progress 100% Scan complete	GMT -5:00
12/15/23 13:21:35.944	Information	[Instrument Response] 14 Result 0 OK	GMT -5:00
12/15/23 13:21:35.944	Information	Acquisition 0000008_01 finished with channels: 700 800.	GMT -5:00
12/15/23 13:21:36.194	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/15/23 13:21:36.194	Information	Scan completed successfully	GMT -5:00
12/15/23 13:21:36.397	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         110 V  	GMT -5:00
12/15/23 13:21:36.460	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          70 V  	GMT -5:00
12/15/23 13:21:36.460	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 13:21:36.460	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 13:21:36.460	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 13:21:37.406	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          67 V  	GMT -5:00
12/15/23 13:21:37.466	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          42 V  	GMT -5:00
12/15/23 13:21:37.466	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/15/23 13:21:38.405	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          41 V  	GMT -5:00
12/15/23 13:21:38.466	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          26 V  	GMT -5:00
12/15/23 13:21:39.416	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          28 V  	GMT -5:00
12/15/23 13:21:39.476	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          17 V  	GMT -5:00
12/15/23 13:21:40.416	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          19 V  	GMT -5:00
12/15/23 13:21:40.463	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          11 V  	GMT -5:00
12/15/23 13:21:41.419	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          12 V  	GMT -5:00
12/15/23 13:21:41.466	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           8 V  	GMT -5:00
12/15/23 13:21:42.422	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           9 V  	GMT -5:00
12/15/23 13:21:42.484	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/15/23 13:21:43.431	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           6 V  	GMT -5:00
12/15/23 13:21:43.480	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           3 V  	GMT -5:00
12/15/23 13:21:44.464	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           3 V  	GMT -5:00
12/15/23 13:21:46.438	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           1 V  	GMT -5:00
12/15/23 13:21:47.456	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           1 V  	GMT -5:00
12/15/23 13:21:48.458	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/15/23 13:22:07.501	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/15/23 13:23:13.259	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 13:23:18.227	Information	Starting acquisition...	GMT -5:00
12/15/23 13:23:19.157	Information	[Instrument Request]  15 ScanAuto 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:23:19.220	Information	[Instrument Response] 15 Status 1 Executing command: ScanAuto 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Status 0 Blocks=23	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 13:23:19.251	Information	[Instrument Response] 15 Progress 0% 5 minutes	GMT -5:00
12/15/23 13:23:19.266	Information	[Instrument Response] 15 Progress 0% Initializing	GMT -5:00
12/15/23 13:23:19.282	Information	[Instrument Response] 15 Status 0 Preparation complete	GMT -5:00
12/15/23 13:25:58.421	Information	Canceling acquisition...	GMT -5:00
12/15/23 13:25:58.421	Information	[Instrument Request]  16 Stop 	GMT -5:00
12/15/23 13:26:03.405	Error      	The instrument connection was closed. If a connection is expected, check the cable connection and confirm the instrument is powered on.  To reconnect, click the Image Studio Application Button, hover over Instrument, and click Connect.	GMT -5:00
12/15/23 13:26:13.828	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/15/23 13:26:21.403	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/15/23 13:26:21.450	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/15/23 13:26:21.497	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/15/23 13:26:21.497	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/15/23 13:26:21.497	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/15/23 13:26:21.497	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/15/23 13:26:21.497	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/15/23 13:26:21.497	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/15/23 13:26:21.497	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/15/23 13:26:21.497	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/15/23 13:26:21.497	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/15/23 13:26:21.513	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/15/23 13:26:21.513	Information	[Instrument Request]  17 ScanAutoEstimate 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:26:24.467	Information	[Instrument Response] 17 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Status 0 Blocks=23	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 13:26:24.483	Information	[Instrument Response] 17 Progress 0% 5 minutes	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 17 Result 0 OK	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    672 mV 	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               26 C  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      265 mA 	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/15/23 13:26:24.498	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.0 V  	GMT -5:00
12/15/23 13:26:24.530	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 13:26:26.314	Information	Starting acquisition...	GMT -5:00
12/15/23 13:26:27.220	Information	[Instrument Request]  18 ScanAuto 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:26:27.298	Information	[Instrument Response] 18 Status 1 Executing command: ScanAuto 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 13:26:27.298	Information	[Instrument Response] 18 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:26:27.298	Information	[Instrument Response] 18 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 13:26:27.314	Information	[Instrument Response] 18 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:26:27.314	Information	[Instrument Response] 18 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:26:27.314	Information	[Instrument Response] 18 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:26:27.314	Information	[Instrument Response] 18 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:26:27.314	Information	[Instrument Response] 18 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 13:26:27.314	Information	[Instrument Response] 18 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:26:27.314	Information	[Instrument Response] 18 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 13:26:27.314	Information	[Instrument Response] 18 Status 0 Blocks=23	GMT -5:00
12/15/23 13:26:27.314	Information	[Instrument Response] 18 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 13:26:27.314	Information	[Instrument Response] 18 Progress 0% 5 minutes	GMT -5:00
12/15/23 13:26:27.314	Information	[Instrument Response] 18 Progress 0% Initializing	GMT -5:00
12/15/23 13:26:27.329	Information	[Instrument Response] 18 Status 0 Preparation complete	GMT -5:00
12/15/23 13:26:30.251	Information	[Instrument Response] 18 Status 0 Motors initialized	GMT -5:00
12/15/23 13:26:30.298	Information	[Instrument Response] 18 Status 0 X-Motor configured	GMT -5:00
12/15/23 13:26:30.298	Information	[Instrument Response] 18 Status 0 DSP configured	GMT -5:00
12/15/23 13:26:30.345	Information	[Instrument Response] 18 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 13:26:30.345	Information	[Instrument Response] 18 Status 0 Cooling	GMT -5:00
12/15/23 13:26:30.345	Information	[Instrument Response] 18 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 13:26:30.345	Information	[Instrument Response] 18 Status 0 Create MergeImageChannel: 700 20 0 0 319 14 319 302 xofs=1 yofs=0	GMT -5:00
12/15/23 13:26:30.345	Information	[Instrument Response] 18 Status 0 Create MergeImageChannel: 800 20 0 0 319 14 319 302 xofs=0 yofs=0	GMT -5:00
12/15/23 13:26:30.345	Information	[Instrument Response] 18 Status 0 start=34 scan_lines=14 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 13:26:30.345	Information	[Instrument Response] 18 Status 0 start=34 scan_lines=28 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 13:26:30.345	Information	[Instrument Response] 18 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 13:26:30.501	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 13:26:30.548	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:26:31.292	Information	[Instrument Response] 18 Progress 0% 5 minutes	GMT -5:00
12/15/23 13:26:31.509	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:26:31.561	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         158 V  	GMT -5:00
12/15/23 13:26:31.561	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:26:31.561	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:26:31.561	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:26:32.514	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:26:32.558	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:26:36.521	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      295 mA 	GMT -5:00
12/15/23 13:26:38.526	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:26:39.541	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:26:39.588	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         108 V  	GMT -5:00
12/15/23 13:26:39.588	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:26:39.588	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:26:39.588	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:26:40.545	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:26:44.560	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:26:45.560	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:26:46.516	Information	[Instrument Response] 18 ImageBlock 700 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 13:26:46.516	Information	[Instrument Response] 18 ImageBlock 800 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 13:26:46.532	Information	Transferring image block at (0,0) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:26:46.563	Information	Transferring image block at (0,0) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:26:46.626	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      326 mA 	GMT -5:00
12/15/23 13:26:50.580	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:26:51.580	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:26:52.198	Information	[Instrument Response] 18 Progress 7% 4 minutes, 30 seconds	GMT -5:00
12/15/23 13:26:52.589	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         119 V  	GMT -5:00
12/15/23 13:26:52.636	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:26:52.636	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:26:52.636	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:26:53.589	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:26:54.592	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:26:55.596	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:26:56.596	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:26:56.658	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:26:57.605	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:26:58.604	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:26:58.667	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:26:59.620	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:26:59.667	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:26:59.792	Information	[Instrument Response] 18 ImageBlock 700 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 13:26:59.792	Information	[Instrument Response] 18 ImageBlock 800 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 13:26:59.792	Information	Transferring image block at (0,14) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:26:59.823	Information	Transferring image block at (0,14) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:27:00.620	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:27:01.627	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:27:02.627	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:27:03.636	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:27:04.639	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:27:05.639	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 13:27:05.686	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:27:05.686	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:27:05.686	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:27:06.648	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:27:06.695	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:27:07.648	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:27:08.648	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:27:09.648	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:27:10.651	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:27:11.667	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:27:13.067	Information	[Instrument Response] 18 ImageBlock 700 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 13:27:13.067	Information	[Instrument Response] 18 ImageBlock 800 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 13:27:13.067	Information	Transferring image block at (0,28) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:27:13.082	Information	Transferring image block at (0,28) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:27:19.707	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:27:19.753	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:27:19.753	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:27:19.753	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:27:20.706	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:27:22.753	Information	[Instrument Response] 18 Progress 17% 4 minutes	GMT -5:00
12/15/23 13:27:26.324	Information	[Instrument Response] 18 ImageBlock 700 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 13:27:26.324	Information	[Instrument Response] 18 ImageBlock 800 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 13:27:26.339	Information	Transferring image block at (0,42) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:27:26.355	Information	Transferring image block at (0,42) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:27:32.767	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:27:32.830	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:27:32.830	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:27:32.830	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:27:33.767	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:27:34.783	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:27:36.788	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:27:39.600	Information	[Instrument Response] 18 ImageBlock 700 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 13:27:39.600	Information	[Instrument Response] 18 ImageBlock 800 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 13:27:39.600	Information	Transferring image block at (0,56) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:27:39.631	Information	Transferring image block at (0,56) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:27:45.840	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:27:45.887	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:27:45.887	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:27:45.887	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:27:46.840	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:27:48.855	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:27:50.854	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:27:52.504	Information	[Instrument Response] 18 Progress 28% 3 minutes, 30 seconds	GMT -5:00
12/15/23 13:27:52.863	Information	[Instrument Response] 18 ImageBlock 700 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 13:27:52.863	Information	[Instrument Response] 18 ImageBlock 800 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 13:27:52.879	Information	Transferring image block at (0,70) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:27:52.895	Information	Transferring image block at (0,70) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:27:58.425	Error      	The 'Stop' command timed out. The elapsed time exceeded 120 seconds. Check the instrument connection.	GMT -5:00
12/15/23 13:27:58.894	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:27:58.956	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/15/23 13:27:58.956	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:27:58.956	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:27:59.909	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:28:06.127	Information	[Instrument Response] 18 ImageBlock 700 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 13:28:06.142	Information	[Instrument Response] 18 ImageBlock 800 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 13:28:06.142	Information	Transferring image block at (0,84) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:28:06.158	Information	Transferring image block at (0,84) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:28:11.976	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:28:12.023	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:28:12.023	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:28:12.023	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:28:12.023	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:28:12.976	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:28:13.023	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:28:13.976	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:28:14.982	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      295 mA 	GMT -5:00
12/15/23 13:28:19.397	Information	[Instrument Response] 18 ImageBlock 700 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 13:28:19.397	Information	[Instrument Response] 18 ImageBlock 800 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 13:28:19.413	Information	Transferring image block at (0,98) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:28:19.428	Information	Transferring image block at (0,98) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:28:22.657	Information	[Instrument Response] 18 Progress 38% 3 minutes	GMT -5:00
12/15/23 13:28:25.047	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          55 mA 	GMT -5:00
12/15/23 13:28:26.047	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:28:26.094	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/15/23 13:28:26.094	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:28:26.094	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:28:27.044	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:28:29.063	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:28:30.063	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:28:32.675	Information	[Instrument Response] 18 ImageBlock 700 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 13:28:32.675	Information	[Instrument Response] 18 ImageBlock 800 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 13:28:32.675	Information	Transferring image block at (0,112) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:28:32.706	Information	Transferring image block at (0,112) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:28:39.117	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          98 V  	GMT -5:00
12/15/23 13:28:39.164	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:28:39.164	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:28:39.164	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:28:40.117	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:28:41.126	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:28:43.126	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:28:45.938	Information	[Instrument Response] 18 ImageBlock 700 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 13:28:45.938	Information	[Instrument Response] 18 ImageBlock 800 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 13:28:45.938	Information	Transferring image block at (0,126) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:28:45.969	Information	Transferring image block at (0,126) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:28:51.610	Information	[Instrument Response] 18 Progress 49% 2 minutes, 30 seconds	GMT -5:00
12/15/23 13:28:52.189	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:28:52.236	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:28:52.236	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:28:52.236	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:28:53.183	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:28:55.198	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:28:57.201	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:28:59.204	Information	[Instrument Response] 18 ImageBlock 700 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 13:28:59.204	Information	[Instrument Response] 18 ImageBlock 800 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 13:28:59.219	Information	Transferring image block at (0,140) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:28:59.250	Information	Transferring image block at (0,140) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:29:05.242	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/15/23 13:29:05.304	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:29:05.304	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:29:05.304	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:29:06.242	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:29:07.251	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:29:08.251	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:29:12.473	Information	[Instrument Response] 18 ImageBlock 700 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 13:29:12.473	Information	[Instrument Response] 18 ImageBlock 800 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 13:29:12.488	Information	Transferring image block at (0,154) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:29:12.520	Information	Transferring image block at (0,154) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:29:18.310	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 13:29:18.372	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:29:18.372	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:29:18.372	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:29:18.372	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:29:19.319	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 13:29:19.366	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:29:20.319	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:29:21.328	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:29:22.333	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:29:22.567	Information	[Instrument Response] 18 Progress 59% 2 minutes	GMT -5:00
12/15/23 13:29:25.738	Information	[Instrument Response] 18 ImageBlock 700 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 13:29:25.738	Information	[Instrument Response] 18 ImageBlock 800 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 13:29:25.754	Information	Transferring image block at (0,168) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:29:25.770	Information	Transferring image block at (0,168) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:29:31.411	Information	[Instrument Response] 18 Progress 63% 1 minute, 50 seconds	GMT -5:00
12/15/23 13:29:32.380	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:29:32.442	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:29:32.442	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:29:32.442	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:29:33.395	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:29:35.395	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:29:36.410	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:29:39.007	Information	[Instrument Response] 18 ImageBlock 700 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 13:29:39.007	Information	[Instrument Response] 18 ImageBlock 800 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 13:29:39.023	Information	Transferring image block at (0,182) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:29:39.038	Information	Transferring image block at (0,182) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:29:42.683	Information	[Instrument Response] 18 Progress 66% 1 minute, 40 seconds	GMT -5:00
12/15/23 13:29:45.448	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:29:45.511	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:29:45.511	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:29:45.511	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:29:46.464	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:29:47.464	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:29:49.464	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:29:52.276	Information	[Instrument Response] 18 ImageBlock 700 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 13:29:52.276	Information	[Instrument Response] 18 ImageBlock 800 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 13:29:52.291	Information	Transferring image block at (0,196) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:29:52.323	Information	Transferring image block at (0,196) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:29:52.721	Information	[Instrument Response] 18 Progress 70% 1 minute, 30 seconds	GMT -5:00
12/15/23 13:29:58.518	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:29:58.580	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:29:58.580	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:29:58.580	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:29:59.534	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:30:01.533	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:30:02.377	Information	[Instrument Response] 18 Progress 73% 1 minute, 20 seconds	GMT -5:00
12/15/23 13:30:03.540	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:30:05.555	Information	[Instrument Response] 18 ImageBlock 700 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 13:30:05.555	Information	[Instrument Response] 18 ImageBlock 800 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 13:30:05.555	Information	Transferring image block at (0,210) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:30:05.586	Information	Transferring image block at (0,210) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:30:11.222	Information	[Instrument Response] 18 Progress 76% 1 minute, 10 seconds	GMT -5:00
12/15/23 13:30:11.581	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:30:11.643	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/15/23 13:30:11.643	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:30:11.643	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:30:12.596	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:30:13.596	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:30:14.596	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:30:18.818	Information	[Instrument Response] 18 ImageBlock 700 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 13:30:18.818	Information	[Instrument Response] 18 ImageBlock 800 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 13:30:18.818	Information	Transferring image block at (0,224) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:30:18.850	Information	Transferring image block at (0,224) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:30:22.478	Information	[Instrument Response] 18 Progress 80% 1 minute	GMT -5:00
12/15/23 13:30:24.656	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:30:24.718	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:30:24.718	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:30:24.718	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:30:24.718	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:30:25.656	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:30:25.718	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:30:26.671	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:30:32.092	Information	[Instrument Response] 18 ImageBlock 700 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 13:30:32.092	Information	[Instrument Response] 18 ImageBlock 800 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 13:30:32.092	Information	Transferring image block at (0,238) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:30:32.123	Information	Transferring image block at (0,238) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:30:32.530	Information	[Instrument Response] 18 Progress 84% 50 seconds	GMT -5:00
12/15/23 13:30:38.723	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:30:38.785	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/15/23 13:30:38.785	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:30:38.785	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:30:39.726	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:30:42.573	Information	[Instrument Response] 18 Progress 87% 40 seconds	GMT -5:00
12/15/23 13:30:45.354	Information	[Instrument Response] 18 ImageBlock 700 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 13:30:45.354	Information	[Instrument Response] 18 ImageBlock 800 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 13:30:45.354	Information	Transferring image block at (0,252) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:30:45.385	Information	Transferring image block at (0,252) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:30:51.034	Information	[Instrument Response] 18 Progress 90% 30 seconds	GMT -5:00
12/15/23 13:30:51.799	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:30:51.846	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:30:51.846	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:30:51.846	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:30:52.800	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:30:53.800	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:30:55.807	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:30:58.619	Information	[Instrument Response] 18 ImageBlock 700 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 13:30:58.619	Information	[Instrument Response] 18 ImageBlock 800 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 13:30:58.634	Information	Transferring image block at (0,266) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:30:58.650	Information	Transferring image block at (0,266) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:31:02.693	Information	[Instrument Response] 18 Progress 94% 20 seconds	GMT -5:00
12/15/23 13:31:04.872	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:31:04.919	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:31:04.919	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:31:04.919	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:31:05.865	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:31:07.881	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:31:09.884	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:31:11.899	Information	[Instrument Response] 18 ImageBlock 700 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 13:31:11.899	Information	[Instrument Response] 18 ImageBlock 800 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 13:31:11.899	Information	Transferring image block at (0,280) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:31:11.946	Information	Transferring image block at (0,280) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:31:12.743	Information	[Instrument Response] 18 Progress 98% 10 seconds	GMT -5:00
12/15/23 13:31:15.914	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:31:15.976	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:31:15.976	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:31:15.976	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:31:16.928	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:31:17.932	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:31:19.933	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:31:20.371	Information	[Instrument Response] 18 Progress 100% 0 seconds	GMT -5:00
12/15/23 13:31:20.949	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:31:20.996	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:31:20.996	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:31:20.996	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:31:21.942	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:31:22.366	Information	[Instrument Response] 18 ImageBlock 700 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 13:31:22.366	Information	[Instrument Response] 18 ImageBlock 800 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 13:31:22.382	Information	[Instrument Response] 18 Status 0 COMPLETE SCAN 291.071 seconds	GMT -5:00
12/15/23 13:31:22.382	Information	Transferring image block at (0,294) with size 319x8 to the 700 channel.	GMT -5:00
12/15/23 13:31:22.382	Information	[Instrument Response] 18 Status 0 Averages (sec): setup=2.00005 line=0.402433 => total=246	GMT -5:00
12/15/23 13:31:22.397	Information	Transferring image block at (0,294) with size 319x8 to the 800 channel.	GMT -5:00
12/15/23 13:31:22.413	Information	[Instrument Response] 18 Status 0 Comparisons: setup=1.00997 line=0.950274	GMT -5:00
12/15/23 13:31:22.460	Information	[Instrument Response] 18 Status 0 Finalize motors	GMT -5:00
12/15/23 13:31:22.460	Information	[Instrument Response] 18 Progress 100% Scan complete	GMT -5:00
12/15/23 13:31:22.491	Information	[Instrument Response] 18 Result 0 OK	GMT -5:00
12/15/23 13:31:22.491	Information	Acquisition 0000009_01 finished with channels: 700 800.	GMT -5:00
12/15/23 13:31:22.757	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/15/23 13:31:22.757	Information	Scan completed successfully	GMT -5:00
12/15/23 13:31:22.944	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         110 V  	GMT -5:00
12/15/23 13:31:22.998	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          70 V  	GMT -5:00
12/15/23 13:31:22.998	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 13:31:22.998	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 13:31:22.998	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 13:31:23.960	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          67 V  	GMT -5:00
12/15/23 13:31:24.010	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          42 V  	GMT -5:00
12/15/23 13:31:24.960	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          41 V  	GMT -5:00
12/15/23 13:31:25.005	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          25 V  	GMT -5:00
12/15/23 13:31:25.965	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          28 V  	GMT -5:00
12/15/23 13:31:26.030	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          18 V  	GMT -5:00
12/15/23 13:31:26.959	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          19 V  	GMT -5:00
12/15/23 13:31:27.012	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          11 V  	GMT -5:00
12/15/23 13:31:27.977	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          12 V  	GMT -5:00
12/15/23 13:31:28.061	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           8 V  	GMT -5:00
12/15/23 13:31:28.979	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           9 V  	GMT -5:00
12/15/23 13:31:29.033	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/15/23 13:31:29.983	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           6 V  	GMT -5:00
12/15/23 13:31:30.036	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           3 V  	GMT -5:00
12/15/23 13:31:30.985	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           3 V  	GMT -5:00
12/15/23 13:31:31.986	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      265 mA 	GMT -5:00
12/15/23 13:31:32.989	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           1 V  	GMT -5:00
12/15/23 13:31:33.989	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           1 V  	GMT -5:00
12/15/23 13:32:33.409	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/15/23 13:33:36.774	Information	Starting acquisition...	GMT -5:00
12/15/23 13:33:37.618	Information	[Instrument Request]  19 ScanAuto 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:33:42.414	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 13:33:42.460	Information	[Instrument Response] 19 Status 1 Executing command: ScanAuto 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 13:33:42.476	Information	[Instrument Response] 19 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:33:42.491	Information	[Instrument Response] 19 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 13:33:42.491	Information	[Instrument Response] 19 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:33:42.491	Information	[Instrument Response] 19 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:33:42.491	Information	[Instrument Response] 19 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:33:42.491	Information	[Instrument Response] 19 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:33:42.491	Information	[Instrument Response] 19 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 13:33:42.491	Information	[Instrument Response] 19 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:33:42.491	Information	[Instrument Response] 19 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 13:33:42.491	Information	[Instrument Response] 19 Status 0 Blocks=23	GMT -5:00
12/15/23 13:33:42.491	Information	[Instrument Response] 19 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 13:33:42.491	Information	[Instrument Response] 19 Progress 0% 5 minutes	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Progress 0% Initializing	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 Preparation complete	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 Motors initialized	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 X-Motor configured	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 DSP configured	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 Cooling	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 Create MergeImageChannel: 700 20 0 0 319 14 319 302 xofs=1 yofs=0	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 Create MergeImageChannel: 800 20 0 0 319 14 319 302 xofs=0 yofs=0	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 start=34 scan_lines=14 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 start=34 scan_lines=28 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 19 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 13:33:42.507	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:33:43.491	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:33:44.679	Information	[Instrument Response] 19 Progress 0% 5 minutes	GMT -5:00
12/15/23 13:33:45.492	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:33:45.554	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:33:45.554	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:33:45.554	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:33:45.554	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:33:49.511	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      295 mA 	GMT -5:00
12/15/23 13:33:52.526	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:33:52.589	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         132 V  	GMT -5:00
12/15/23 13:33:52.589	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:33:52.589	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:33:52.589	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:33:52.589	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:33:53.533	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          92 V  	GMT -5:00
12/15/23 13:33:53.580	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:33:53.580	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:33:54.533	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:33:57.548	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      326 mA 	GMT -5:00
12/15/23 13:33:59.907	Information	[Instrument Response] 19 ImageBlock 700 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 13:33:59.907	Information	[Instrument Response] 19 ImageBlock 800 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 13:33:59.907	Information	Transferring image block at (0,0) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:33:59.938	Information	Transferring image block at (0,0) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:34:05.579	Information	[Instrument Response] 19 Progress 7% 4 minutes, 30 seconds	GMT -5:00
12/15/23 13:34:05.626	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:34:06.582	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          87 V  	GMT -5:00
12/15/23 13:34:06.629	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:34:06.629	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:34:06.629	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:34:06.629	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:34:07.582	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:34:09.598	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:34:10.598	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:34:11.605	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:34:12.604	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:34:13.167	Information	[Instrument Response] 19 ImageBlock 700 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 13:34:13.167	Information	[Instrument Response] 19 ImageBlock 800 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 13:34:13.182	Information	Transferring image block at (0,14) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:34:13.198	Information	Transferring image block at (0,14) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:34:13.604	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    749 mV 	GMT -5:00
12/15/23 13:34:13.667	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:34:14.620	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/15/23 13:34:14.666	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:34:15.619	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:34:16.614	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:34:17.630	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:34:18.630	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:34:19.630	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:34:19.692	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          98 V  	GMT -5:00
12/15/23 13:34:19.692	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:34:19.692	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:34:20.645	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:34:21.645	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:34:22.649	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:34:23.649	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:34:23.711	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:34:24.649	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:34:24.711	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 13:34:26.445	Information	[Instrument Response] 19 ImageBlock 700 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 13:34:26.445	Information	[Instrument Response] 19 ImageBlock 800 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 13:34:26.445	Information	Transferring image block at (0,28) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:34:26.476	Information	Transferring image block at (0,28) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:34:27.661	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:34:28.665	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:34:29.668	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:34:30.683	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:34:32.683	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:34:32.745	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:34:32.745	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:34:32.745	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:34:33.686	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:34:35.690	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:34:35.753	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:34:36.128	Information	[Instrument Response] 19 Progress 17% 4 minutes	GMT -5:00
12/15/23 13:34:36.706	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:34:36.753	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:34:39.706	Information	[Instrument Response] 19 ImageBlock 700 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 13:34:39.706	Information	[Instrument Response] 19 ImageBlock 800 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 13:34:39.706	Information	Transferring image block at (0,42) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:34:39.737	Information	Transferring image block at (0,42) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:34:45.739	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:34:45.802	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/15/23 13:34:45.802	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:34:45.802	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:34:46.755	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:34:52.974	Information	[Instrument Response] 19 ImageBlock 700 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 13:34:52.974	Information	[Instrument Response] 19 ImageBlock 800 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 13:34:52.990	Information	Transferring image block at (0,56) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:34:53.021	Information	Transferring image block at (0,56) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:34:58.812	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:34:58.874	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:34:58.874	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:34:58.874	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:34:58.874	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/15/23 13:34:59.827	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:34:59.874	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:35:00.828	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:35:05.885	Information	[Instrument Response] 19 Progress 28% 3 minutes, 30 seconds	GMT -5:00
12/15/23 13:35:06.244	Information	[Instrument Response] 19 ImageBlock 700 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 13:35:06.244	Information	[Instrument Response] 19 ImageBlock 800 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 13:35:06.260	Information	Transferring image block at (0,70) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:35:06.275	Information	Transferring image block at (0,70) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:35:12.887	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:35:12.949	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/15/23 13:35:12.949	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:35:12.949	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:35:13.902	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:35:15.903	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:35:16.903	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:35:19.512	Information	[Instrument Response] 19 ImageBlock 700 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 13:35:19.512	Information	[Instrument Response] 19 ImageBlock 800 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 13:35:19.527	Information	Transferring image block at (0,84) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:35:19.543	Information	Transferring image block at (0,84) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:35:25.953	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          98 V  	GMT -5:00
12/15/23 13:35:26.015	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:35:26.015	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:35:26.015	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:35:26.968	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:35:27.968	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:35:29.970	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:35:32.782	Information	[Instrument Response] 19 ImageBlock 700 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 13:35:32.782	Information	[Instrument Response] 19 ImageBlock 800 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 13:35:32.797	Information	Transferring image block at (0,98) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:35:32.813	Information	Transferring image block at (0,98) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:35:36.048	Information	[Instrument Response] 19 Progress 38% 3 minutes	GMT -5:00
12/15/23 13:35:39.023	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:35:39.085	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:35:39.085	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:35:39.085	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:35:40.038	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:35:42.040	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    749 mV 	GMT -5:00
12/15/23 13:35:42.087	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:35:43.049	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/15/23 13:35:44.055	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:35:46.054	Information	[Instrument Response] 19 ImageBlock 700 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 13:35:46.054	Information	[Instrument Response] 19 ImageBlock 800 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 13:35:46.054	Information	Transferring image block at (0,112) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:35:46.101	Information	Transferring image block at (0,112) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:35:52.092	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/15/23 13:35:52.139	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:35:52.139	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:35:52.139	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:35:53.092	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:35:54.106	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:35:55.106	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:35:59.330	Information	[Instrument Response] 19 ImageBlock 700 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 13:35:59.330	Information	[Instrument Response] 19 ImageBlock 800 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 13:35:59.330	Information	Transferring image block at (0,126) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:35:59.361	Information	Transferring image block at (0,126) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:36:05.004	Information	[Instrument Response] 19 Progress 49% 2 minutes, 30 seconds	GMT -5:00
12/15/23 13:36:05.160	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 13:36:05.222	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:36:05.222	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:36:05.222	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:36:05.222	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:36:05.222	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/15/23 13:36:06.158	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 13:36:06.221	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:36:06.221	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:36:07.173	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:36:12.589	Information	[Instrument Response] 19 ImageBlock 700 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 13:36:12.589	Information	[Instrument Response] 19 ImageBlock 800 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 13:36:12.604	Information	Transferring image block at (0,140) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:36:12.635	Information	Transferring image block at (0,140) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:36:19.232	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:36:19.295	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:36:19.295	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:36:19.295	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:36:20.232	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:36:22.248	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:36:23.248	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:36:25.857	Information	[Instrument Response] 19 ImageBlock 700 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 13:36:25.857	Information	[Instrument Response] 19 ImageBlock 800 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 13:36:25.872	Information	Transferring image block at (0,154) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:36:25.888	Information	Transferring image block at (0,154) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:36:32.308	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:36:32.355	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:36:32.355	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:36:32.355	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:36:33.311	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:36:34.311	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:36:35.955	Information	[Instrument Response] 19 Progress 59% 2 minutes	GMT -5:00
12/15/23 13:36:36.314	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:36:39.127	Information	[Instrument Response] 19 ImageBlock 700 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 13:36:39.127	Information	[Instrument Response] 19 ImageBlock 800 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 13:36:39.142	Information	Transferring image block at (0,168) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:36:39.158	Information	Transferring image block at (0,168) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:36:44.798	Information	[Instrument Response] 19 Progress 63% 1 minute, 50 seconds	GMT -5:00
12/15/23 13:36:45.376	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:36:45.423	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:36:45.423	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:36:45.423	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:36:46.376	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:36:48.377	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:36:50.392	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:36:52.391	Information	[Instrument Response] 19 ImageBlock 700 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 13:36:52.391	Information	[Instrument Response] 19 ImageBlock 800 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 13:36:52.407	Information	Transferring image block at (0,182) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:36:52.422	Information	Transferring image block at (0,182) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:36:56.066	Information	[Instrument Response] 19 Progress 66% 1 minute, 40 seconds	GMT -5:00
12/15/23 13:36:58.434	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:36:58.496	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/15/23 13:36:58.496	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:36:58.496	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:36:59.441	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:37:05.667	Information	[Instrument Response] 19 ImageBlock 700 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 13:37:05.667	Information	[Instrument Response] 19 ImageBlock 800 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 13:37:05.667	Information	Transferring image block at (0,196) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:37:05.698	Information	Transferring image block at (0,196) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:37:06.105	Information	[Instrument Response] 19 Progress 70% 1 minute, 30 seconds	GMT -5:00
12/15/23 13:37:11.504	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:37:11.551	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:37:11.551	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:37:11.551	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:37:11.551	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:37:11.551	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/15/23 13:37:12.504	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 13:37:12.566	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:37:12.566	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:37:13.503	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:37:14.519	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:37:15.519	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:37:16.160	Information	[Instrument Response] 19 Progress 73% 1 minute, 20 seconds	GMT -5:00
12/15/23 13:37:18.940	Information	[Instrument Response] 19 ImageBlock 700 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 13:37:18.940	Information	[Instrument Response] 19 ImageBlock 800 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 13:37:18.940	Information	Transferring image block at (0,210) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:37:18.955	Information	Transferring image block at (0,210) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:37:24.602	Information	[Instrument Response] 19 Progress 76% 1 minute, 10 seconds	GMT -5:00
12/15/23 13:37:25.571	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:37:25.633	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/15/23 13:37:25.633	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:37:25.633	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:37:26.586	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:37:28.586	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:37:29.586	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:37:32.198	Information	[Instrument Response] 19 ImageBlock 700 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 13:37:32.198	Information	[Instrument Response] 19 ImageBlock 800 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 13:37:32.214	Information	Transferring image block at (0,224) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:37:32.245	Information	Transferring image block at (0,224) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:37:35.872	Information	[Instrument Response] 19 Progress 80% 1 minute	GMT -5:00
12/15/23 13:37:38.640	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          99 V  	GMT -5:00
12/15/23 13:37:38.703	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:37:38.703	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:37:38.703	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:37:39.640	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:37:40.655	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:37:42.655	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:37:45.467	Information	[Instrument Response] 19 ImageBlock 700 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 13:37:45.467	Information	[Instrument Response] 19 ImageBlock 800 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 13:37:45.483	Information	Transferring image block at (0,238) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:37:45.498	Information	Transferring image block at (0,238) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:37:45.920	Information	[Instrument Response] 19 Progress 84% 50 seconds	GMT -5:00
12/15/23 13:37:51.720	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:37:51.767	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 13:37:51.767	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:37:51.767	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:37:52.714	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:37:54.729	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:37:55.965	Information	[Instrument Response] 19 Progress 87% 40 seconds	GMT -5:00
12/15/23 13:37:56.730	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:37:58.740	Information	[Instrument Response] 19 ImageBlock 700 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 13:37:58.740	Information	[Instrument Response] 19 ImageBlock 800 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 13:37:58.740	Information	Transferring image block at (0,252) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:37:58.771	Information	Transferring image block at (0,252) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:38:04.418	Information	[Instrument Response] 19 Progress 90% 30 seconds	GMT -5:00
12/15/23 13:38:04.777	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/15/23 13:38:04.824	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:38:04.824	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:38:04.824	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:38:05.777	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:38:06.791	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 13:38:07.784	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:38:12.003	Information	[Instrument Response] 19 ImageBlock 700 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 13:38:12.003	Information	[Instrument Response] 19 ImageBlock 800 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 13:38:12.018	Information	Transferring image block at (0,266) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:38:12.050	Information	Transferring image block at (0,266) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:38:16.074	Information	[Instrument Response] 19 Progress 94% 20 seconds	GMT -5:00
12/15/23 13:38:17.844	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 13:38:17.907	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:38:17.907	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:38:17.907	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:38:17.907	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:38:17.907	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/15/23 13:38:18.844	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 13:38:18.907	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:38:19.860	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:38:20.863	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:38:21.863	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:38:23.863	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:38:25.284	Information	[Instrument Response] 19 ImageBlock 700 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 13:38:25.284	Information	[Instrument Response] 19 ImageBlock 800 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 13:38:25.284	Information	Transferring image block at (0,280) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:38:25.315	Information	Transferring image block at (0,280) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:38:26.128	Information	[Instrument Response] 19 Progress 98% 10 seconds	GMT -5:00
12/15/23 13:38:28.893	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/15/23 13:38:28.956	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:38:28.956	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:38:28.956	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:38:29.902	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:38:33.761	Information	[Instrument Response] 19 Progress 100% 0 seconds	GMT -5:00
12/15/23 13:38:33.918	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 13:38:33.980	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 13:38:33.980	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:38:33.980	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:38:34.917	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 13:38:35.761	Information	[Instrument Response] 19 ImageBlock 700 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 13:38:35.761	Information	[Instrument Response] 19 ImageBlock 800 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 13:38:35.761	Information	[Instrument Response] 19 Status 0 COMPLETE SCAN 291.074 seconds	GMT -5:00
12/15/23 13:38:35.761	Information	Transferring image block at (0,294) with size 319x8 to the 700 channel.	GMT -5:00
12/15/23 13:38:35.761	Information	[Instrument Response] 19 Status 0 Averages (sec): setup=2.00005 line=0.402439 => total=246	GMT -5:00
12/15/23 13:38:35.792	Information	Transferring image block at (0,294) with size 319x8 to the 800 channel.	GMT -5:00
12/15/23 13:38:35.808	Information	[Instrument Response] 19 Status 0 Comparisons: setup=1.00997 line=0.950258	GMT -5:00
12/15/23 13:38:35.839	Information	[Instrument Response] 19 Status 0 Finalize motors	GMT -5:00
12/15/23 13:38:35.854	Information	[Instrument Response] 19 Progress 100% Scan complete	GMT -5:00
12/15/23 13:38:35.886	Information	[Instrument Response] 19 Result 0 OK	GMT -5:00
12/15/23 13:38:35.886	Information	Acquisition 0000010_01 finished with channels: 700 800.	GMT -5:00
12/15/23 13:38:36.136	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/15/23 13:38:36.136	Information	Scan completed successfully	GMT -5:00
12/15/23 13:38:36.245	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         142 V  	GMT -5:00
12/15/23 13:38:36.261	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 13:38:36.261	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 13:38:36.276	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 13:38:36.939	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          83 V  	GMT -5:00
12/15/23 13:38:36.986	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          52 V  	GMT -5:00
12/15/23 13:38:37.941	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          51 V  	GMT -5:00
12/15/23 13:38:38.002	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          32 V  	GMT -5:00
12/15/23 13:38:38.002	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      295 mA 	GMT -5:00
12/15/23 13:38:38.940	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          32 V  	GMT -5:00
12/15/23 13:38:39.090	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          20 V  	GMT -5:00
12/15/23 13:38:39.951	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          21 V  	GMT -5:00
12/15/23 13:38:39.998	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          13 V  	GMT -5:00
12/15/23 13:38:40.952	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          14 V  	GMT -5:00
12/15/23 13:38:40.999	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           9 V  	GMT -5:00
12/15/23 13:38:41.953	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          10 V  	GMT -5:00
12/15/23 13:38:42.000	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           7 V  	GMT -5:00
12/15/23 13:38:42.954	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           7 V  	GMT -5:00
12/15/23 13:38:43.023	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           4 V  	GMT -5:00
12/15/23 13:38:43.956	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/15/23 13:38:44.025	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/15/23 13:38:45.961	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/15/23 13:38:48.983	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 13:38:49.990	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      265 mA 	GMT -5:00
12/15/23 13:38:50.990	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/15/23 13:39:23.533	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/15/23 13:40:07.430	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 13:40:15.022	Information	Starting acquisition...	GMT -5:00
12/15/23 13:40:15.945	Information	[Instrument Request]  20 ScanAuto 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 13:40:48.423	Information	[Instrument Response] 20 Status 1 Executing command: ScanAuto 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Status 0 Computed scan parameters	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Status 0 Blocks=23	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 13:40:48.454	Information	[Instrument Response] 20 Progress 0% 5 minutes	GMT -5:00
12/15/23 13:40:48.485	Information	[Instrument Response] 20 Progress 0% Initializing	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 Preparation complete	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 Motors initialized	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 X-Motor configured	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 DSP configured	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 Cooling	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 Create MergeImageChannel: 700 20 0 0 319 14 319 302 xofs=1 yofs=0	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 Create MergeImageChannel: 800 20 0 0 319 14 319 302 xofs=0 yofs=0	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 start=34 scan_lines=14 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 start=34 scan_lines=28 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/15/23 13:40:48.500	Information	[Instrument Response] 20 Progress 0% 5 minutes	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      295 mA 	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         102 V  	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      326 mA 	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 20 ImageBlock 700 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 20 ImageBlock 800 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 13:40:48.516	Information	[Instrument Response] 20 Progress 7% 4 minutes, 30 seconds	GMT -5:00
12/15/23 13:40:48.516	Information	Transferring image block at (0,0) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:40:48.532	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         113 V  	GMT -5:00
12/15/23 13:40:48.547	Information	Transferring image block at (0,0) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:40:48.563	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:40:48.610	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:40:48.610	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:40:48.610	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:40:51.517	Information	[Instrument Response] 20 ImageBlock 700 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 13:40:51.517	Information	[Instrument Response] 20 ImageBlock 800 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 13:40:51.517	Information	Transferring image block at (0,14) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:40:51.548	Information	Transferring image block at (0,14) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:40:57.473	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:40:57.536	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         127 V  	GMT -5:00
12/15/23 13:40:57.536	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:40:57.536	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:40:58.489	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:40:59.489	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:41:04.785	Information	[Instrument Response] 20 ImageBlock 700 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 13:41:04.785	Information	[Instrument Response] 20 ImageBlock 800 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 13:41:04.785	Information	Transferring image block at (0,28) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:41:04.816	Information	Transferring image block at (0,28) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:41:10.529	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         145 V  	GMT -5:00
12/15/23 13:41:10.591	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    749 mV 	GMT -5:00
12/15/23 13:41:10.591	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/15/23 13:41:10.591	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:41:10.591	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 13:41:10.591	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:41:10.591	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:41:11.532	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/15/23 13:41:11.595	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:41:11.595	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:41:11.595	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:41:12.548	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:41:13.547	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/15/23 13:41:14.473	Information	[Instrument Response] 20 Progress 17% 4 minutes	GMT -5:00
12/15/23 13:41:18.051	Information	[Instrument Response] 20 ImageBlock 700 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 13:41:18.051	Information	[Instrument Response] 20 ImageBlock 800 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 13:41:18.051	Information	Transferring image block at (0,42) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:41:18.082	Information	Transferring image block at (0,42) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:41:24.602	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:41:24.664	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:41:24.664	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:41:24.664	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:41:25.617	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:41:31.314	Information	[Instrument Response] 20 ImageBlock 700 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 13:41:31.314	Information	[Instrument Response] 20 ImageBlock 800 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 13:41:31.329	Information	Transferring image block at (0,56) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:41:31.345	Information	Transferring image block at (0,56) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:41:37.683	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         103 V  	GMT -5:00
12/15/23 13:41:37.730	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:41:37.730	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:41:37.730	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:41:38.683	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:41:44.228	Information	[Instrument Response] 20 Progress 28% 3 minutes, 30 seconds	GMT -5:00
12/15/23 13:41:44.588	Information	[Instrument Response] 20 ImageBlock 700 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 13:41:44.588	Information	[Instrument Response] 20 ImageBlock 800 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 13:41:44.588	Information	Transferring image block at (0,70) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:41:44.619	Information	Transferring image block at (0,70) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:41:50.743	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:41:50.806	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         115 V  	GMT -5:00
12/15/23 13:41:50.806	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:41:50.806	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:41:51.747	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:41:57.849	Information	[Instrument Response] 20 ImageBlock 700 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 13:41:57.849	Information	[Instrument Response] 20 ImageBlock 800 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 13:41:57.864	Information	Transferring image block at (0,84) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:41:57.880	Information	Transferring image block at (0,84) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:42:03.820	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         126 V  	GMT -5:00
12/15/23 13:42:03.867	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:42:03.867	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:42:03.867	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:42:03.867	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 13:42:04.820	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:42:04.867	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:42:11.117	Information	[Instrument Response] 20 ImageBlock 700 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 13:42:11.117	Information	[Instrument Response] 20 ImageBlock 800 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 13:42:11.133	Information	Transferring image block at (0,98) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:42:11.148	Information	Transferring image block at (0,98) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:42:14.383	Information	[Instrument Response] 20 Progress 38% 3 minutes	GMT -5:00
12/15/23 13:42:16.883	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         151 V  	GMT -5:00
12/15/23 13:42:16.930	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         145 V  	GMT -5:00
12/15/23 13:42:16.930	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:42:16.930	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:42:16.930	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:42:16.930	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/15/23 13:42:17.883	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:42:17.930	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          85 V  	GMT -5:00
12/15/23 13:42:17.930	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:42:17.930	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:42:18.883	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:42:24.388	Information	[Instrument Response] 20 ImageBlock 700 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 13:42:24.388	Information	[Instrument Response] 20 ImageBlock 800 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 13:42:24.403	Information	Transferring image block at (0,112) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:42:24.434	Information	Transferring image block at (0,112) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:42:30.947	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/15/23 13:42:31.009	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:42:31.009	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:42:31.009	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:42:31.956	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:42:37.655	Information	[Instrument Response] 20 ImageBlock 700 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 13:42:37.655	Information	[Instrument Response] 20 ImageBlock 800 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 13:42:37.671	Information	Transferring image block at (0,126) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:42:37.686	Information	Transferring image block at (0,126) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:42:43.327	Information	[Instrument Response] 20 Progress 49% 2 minutes, 30 seconds	GMT -5:00
12/15/23 13:42:44.014	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:42:44.077	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         104 V  	GMT -5:00
12/15/23 13:42:44.077	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:42:44.077	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:42:45.017	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:42:50.932	Information	[Instrument Response] 20 ImageBlock 700 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 13:42:50.932	Information	[Instrument Response] 20 ImageBlock 800 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 13:42:50.932	Information	Transferring image block at (0,140) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:42:50.963	Information	Transferring image block at (0,140) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:42:57.084	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         114 V  	GMT -5:00
12/15/23 13:42:57.146	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:42:57.146	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:42:57.146	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:42:58.083	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:43:04.193	Information	[Instrument Response] 20 ImageBlock 700 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 13:43:04.193	Information	[Instrument Response] 20 ImageBlock 800 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 13:43:04.208	Information	Transferring image block at (0,154) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:43:04.224	Information	Transferring image block at (0,154) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:43:10.161	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:43:10.207	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         127 V  	GMT -5:00
12/15/23 13:43:10.207	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:43:10.207	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:43:10.207	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:43:10.207	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:43:11.160	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:43:11.207	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:43:11.207	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:43:12.160	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:43:14.291	Information	[Instrument Response] 20 Progress 59% 2 minutes	GMT -5:00
12/15/23 13:43:17.462	Information	[Instrument Response] 20 ImageBlock 700 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 13:43:17.462	Information	[Instrument Response] 20 ImageBlock 800 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 13:43:17.478	Information	Transferring image block at (0,168) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:43:17.494	Information	Transferring image block at (0,168) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:43:23.137	Information	[Instrument Response] 20 Progress 63% 1 minute, 50 seconds	GMT -5:00
12/15/23 13:43:23.215	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         145 V  	GMT -5:00
12/15/23 13:43:23.277	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    749 mV 	GMT -5:00
12/15/23 13:43:23.277	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/15/23 13:43:23.277	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:43:23.277	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:43:23.277	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:43:23.277	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/15/23 13:43:24.230	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/15/23 13:43:24.277	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:43:24.277	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:43:25.230	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:43:25.277	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:43:27.245	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/15/23 13:43:30.736	Information	[Instrument Response] 20 ImageBlock 700 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 13:43:30.736	Information	[Instrument Response] 20 ImageBlock 800 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 13:43:30.736	Information	Transferring image block at (0,182) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:43:30.767	Information	Transferring image block at (0,182) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:43:34.392	Information	[Instrument Response] 20 Progress 66% 1 minute, 40 seconds	GMT -5:00
12/15/23 13:43:37.298	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:43:37.345	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:43:37.345	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:43:37.345	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:43:38.297	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:43:44.004	Information	[Instrument Response] 20 ImageBlock 700 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 13:43:44.004	Information	[Instrument Response] 20 ImageBlock 800 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 13:43:44.004	Information	Transferring image block at (0,196) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:43:44.035	Information	Transferring image block at (0,196) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:43:44.452	Information	[Instrument Response] 20 Progress 70% 1 minute, 30 seconds	GMT -5:00
12/15/23 13:43:50.361	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         103 V  	GMT -5:00
12/15/23 13:43:50.424	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:43:50.424	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:43:50.424	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:43:51.373	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:43:53.382	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    749 mV 	GMT -5:00
12/15/23 13:43:54.086	Information	[Instrument Response] 20 Progress 73% 1 minute, 20 seconds	GMT -5:00
12/15/23 13:43:57.273	Information	[Instrument Response] 20 ImageBlock 700 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 13:43:57.273	Information	[Instrument Response] 20 ImageBlock 800 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 13:43:57.273	Information	Transferring image block at (0,210) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:43:57.304	Information	Transferring image block at (0,210) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:43:59.407	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/15/23 13:44:02.938	Information	[Instrument Response] 20 Progress 76% 1 minute, 10 seconds	GMT -5:00
12/15/23 13:44:03.422	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:44:03.485	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         115 V  	GMT -5:00
12/15/23 13:44:03.485	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:44:03.485	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:44:04.438	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:44:10.540	Information	[Instrument Response] 20 ImageBlock 700 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 13:44:10.540	Information	[Instrument Response] 20 ImageBlock 800 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 13:44:10.540	Information	Transferring image block at (0,224) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:44:10.602	Information	Transferring image block at (0,224) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:44:14.208	Information	[Instrument Response] 20 Progress 80% 1 minute	GMT -5:00
12/15/23 13:44:16.504	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         126 V  	GMT -5:00
12/15/23 13:44:16.551	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:44:16.551	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:44:16.551	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:44:16.551	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:44:16.551	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 13:44:17.504	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 13:44:17.551	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:44:17.551	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:44:23.801	Information	[Instrument Response] 20 ImageBlock 700 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 13:44:23.817	Information	[Instrument Response] 20 ImageBlock 800 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 13:44:23.817	Information	Transferring image block at (0,238) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:44:23.833	Information	Transferring image block at (0,238) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:44:24.255	Information	[Instrument Response] 20 Progress 84% 50 seconds	GMT -5:00
12/15/23 13:44:29.567	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         151 V  	GMT -5:00
12/15/23 13:44:29.614	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         145 V  	GMT -5:00
12/15/23 13:44:29.614	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:44:29.614	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:44:29.614	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 13:44:29.614	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/15/23 13:44:30.567	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:44:30.629	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          85 V  	GMT -5:00
12/15/23 13:44:30.629	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 13:44:31.567	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:44:34.304	Information	[Instrument Response] 20 Progress 87% 40 seconds	GMT -5:00
12/15/23 13:44:34.585	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 13:44:37.085	Information	[Instrument Response] 20 ImageBlock 700 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 13:44:37.085	Information	[Instrument Response] 20 ImageBlock 800 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 13:44:37.085	Information	Transferring image block at (0,252) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:44:37.100	Information	Transferring image block at (0,252) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:44:42.750	Information	[Instrument Response] 20 Progress 90% 30 seconds	GMT -5:00
12/15/23 13:44:43.641	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/15/23 13:44:43.687	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:44:43.687	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 13:44:43.687	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:44:44.640	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:44:50.339	Information	[Instrument Response] 20 ImageBlock 700 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 13:44:50.339	Information	[Instrument Response] 20 ImageBlock 800 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 13:44:50.355	Information	Transferring image block at (0,266) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:44:50.371	Information	Transferring image block at (0,266) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:44:54.401	Information	[Instrument Response] 20 Progress 94% 20 seconds	GMT -5:00
12/15/23 13:44:56.713	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:44:56.760	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         104 V  	GMT -5:00
12/15/23 13:44:56.760	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:44:56.760	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:44:57.713	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:45:03.618	Information	[Instrument Response] 20 ImageBlock 700 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 13:45:03.618	Information	[Instrument Response] 20 ImageBlock 800 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 13:45:03.618	Information	Transferring image block at (0,280) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 13:45:03.649	Information	Transferring image block at (0,280) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 13:45:04.461	Information	[Instrument Response] 20 Progress 98% 10 seconds	GMT -5:00
12/15/23 13:45:07.757	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/15/23 13:45:07.820	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 13:45:07.820	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 13:45:07.820	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 13:45:08.757	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 13:45:08.820	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    749 mV 	GMT -5:00
12/15/23 13:45:12.089	Information	[Instrument Response] 20 Progress 100% 0 seconds	GMT -5:00
12/15/23 13:45:12.776	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 13:45:12.838	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         104 V  	GMT -5:00
12/15/23 13:45:12.838	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 13:45:12.838	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 13:45:13.792	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 13:45:14.088	Information	[Instrument Response] 20 ImageBlock 700 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 13:45:14.088	Information	[Instrument Response] 20 ImageBlock 800 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 13:45:14.088	Information	[Instrument Response] 20 Status 0 COMPLETE SCAN 291.071 seconds	GMT -5:00
12/15/23 13:45:14.104	Information	Transferring image block at (0,294) with size 319x8 to the 700 channel.	GMT -5:00
12/15/23 13:45:14.104	Information	[Instrument Response] 20 Status 0 Averages (sec): setup=2.00005 line=0.402433 => total=246	GMT -5:00
12/15/23 13:45:14.119	Information	Transferring image block at (0,294) with size 319x8 to the 800 channel.	GMT -5:00
12/15/23 13:45:14.135	Information	[Instrument Response] 20 Status 0 Comparisons: setup=1.00997 line=0.950274	GMT -5:00
12/15/23 13:45:14.182	Information	[Instrument Response] 20 Status 0 Finalize motors	GMT -5:00
12/15/23 13:45:14.182	Information	[Instrument Response] 20 Progress 100% Scan complete	GMT -5:00
12/15/23 13:45:14.213	Information	[Instrument Response] 20 Result 0 OK	GMT -5:00
12/15/23 13:45:14.213	Information	Acquisition 0000011_01 finished with channels: 700 800.	GMT -5:00
12/15/23 13:45:14.807	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/15/23 13:45:14.807	Information	Scan completed successfully	GMT -5:00
12/15/23 13:45:14.900	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         104 V  	GMT -5:00
12/15/23 13:45:14.908	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          65 V  	GMT -5:00
12/15/23 13:45:14.908	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 13:45:14.908	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 13:45:14.908	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 13:45:15.802	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          64 V  	GMT -5:00
12/15/23 13:45:15.852	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          40 V  	GMT -5:00
12/15/23 13:45:16.802	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          39 V  	GMT -5:00
12/15/23 13:45:16.862	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          24 V  	GMT -5:00
12/15/23 13:45:17.802	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          26 V  	GMT -5:00
12/15/23 13:45:17.852	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          17 V  	GMT -5:00
12/15/23 13:45:17.852	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      295 mA 	GMT -5:00
12/15/23 13:45:18.807	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          18 V  	GMT -5:00
12/15/23 13:45:18.854	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          11 V  	GMT -5:00
12/15/23 13:45:19.809	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          12 V  	GMT -5:00
12/15/23 13:45:19.872	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           8 V  	GMT -5:00
12/15/23 13:45:20.811	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           8 V  	GMT -5:00
12/15/23 13:45:20.873	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           4 V  	GMT -5:00
12/15/23 13:45:21.826	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           6 V  	GMT -5:00
12/15/23 13:45:22.828	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           3 V  	GMT -5:00
12/15/23 13:45:22.875	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/15/23 13:45:25.833	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           1 V  	GMT -5:00
12/15/23 13:45:27.836	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 13:59:21.409	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/15/23 14:00:08.395	Information	Starting acquisition...	GMT -5:00
12/15/23 14:00:09.269	Information	[Instrument Request]  21 ScanAuto 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 14:00:18.830	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 14:00:18.892	Information	[Instrument Response] 21 Status 1 Executing command: ScanAuto 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Status 0 Computed scan parameters	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Status 0 Blocks=23	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 14:00:18.908	Information	[Instrument Response] 21 Progress 0% 5 minutes	GMT -5:00
12/15/23 14:00:18.923	Information	[Instrument Response] 21 Progress 0% Initializing	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 Preparation complete	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 Motors initialized	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 X-Motor configured	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 DSP configured	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 Cooling	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 Create MergeImageChannel: 700 20 0 0 319 14 319 302 xofs=1 yofs=0	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 Create MergeImageChannel: 800 20 0 0 319 14 319 302 xofs=0 yofs=0	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 start=34 scan_lines=14 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 start=34 scan_lines=28 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 21 Progress 0% 5 minutes	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:00:18.939	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:00:18.955	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:00:18.955	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 14:00:18.955	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 14:02:55.983	Information	Canceling acquisition...	GMT -5:00
12/15/23 14:02:55.983	Information	[Instrument Request]  22 Stop 	GMT -5:00
12/15/23 14:03:06.501	Error      	The instrument connection was closed. If a connection is expected, check the cable connection and confirm the instrument is powered on.  To reconnect, click the Image Studio Application Button, hover over Instrument, and click Connect.	GMT -5:00
12/15/23 14:03:21.843	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/15/23 14:03:26.442	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/15/23 14:03:26.489	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/15/23 14:03:26.535	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/15/23 14:03:26.535	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/15/23 14:03:26.535	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/15/23 14:03:26.535	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/15/23 14:03:26.535	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/15/23 14:03:26.535	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/15/23 14:03:26.535	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/15/23 14:03:26.535	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/15/23 14:03:26.535	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/15/23 14:03:26.535	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/15/23 14:03:26.535	Information	[Instrument Request]  23 ScanAutoEstimate 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 14:03:29.513	Information	[Instrument Response] 23 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 14:03:29.513	Information	[Instrument Response] 23 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 14:03:29.513	Information	[Instrument Response] 23 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 14:03:29.513	Information	[Instrument Response] 23 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 14:03:29.513	Information	[Instrument Response] 23 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 14:03:29.513	Information	[Instrument Response] 23 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 14:03:29.513	Information	[Instrument Response] 23 Status 0 Computed scan parameters	GMT -5:00
12/15/23 14:03:29.513	Information	[Instrument Response] 23 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 14:03:29.513	Information	[Instrument Response] 23 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 14:03:29.513	Information	[Instrument Response] 23 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 14:03:29.528	Information	[Instrument Response] 23 Status 0 Blocks=23	GMT -5:00
12/15/23 14:03:29.528	Information	[Instrument Response] 23 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 14:03:29.528	Information	[Instrument Response] 23 Progress 0% 5 minutes	GMT -5:00
12/15/23 14:03:29.528	Information	[Instrument Response] 23 Result 0 OK	GMT -5:00
12/15/23 14:03:29.528	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/15/23 14:03:29.528	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/15/23 14:03:29.528	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    678 mV 	GMT -5:00
12/15/23 14:03:29.528	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/15/23 14:03:29.528	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/15/23 14:03:29.528	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    742 mV 	GMT -5:00
12/15/23 14:03:29.528	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               26 C  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/15/23 14:03:29.544	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.0 V  	GMT -5:00
12/15/23 14:03:29.575	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 14:04:08.132	Information	Starting acquisition...	GMT -5:00
12/15/23 14:04:09.007	Information	[Instrument Request]  24 ScanAuto 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 14:04:09.085	Information	[Instrument Response] 24 Status 1 Executing command: ScanAuto 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Status 0 Computed scan parameters	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Status 0 Blocks=23	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 14:04:09.101	Information	[Instrument Response] 24 Progress 0% 5 minutes	GMT -5:00
12/15/23 14:04:09.116	Information	[Instrument Response] 24 Progress 0% Initializing	GMT -5:00
12/15/23 14:04:09.132	Information	[Instrument Response] 24 Status 0 Preparation complete	GMT -5:00
12/15/23 14:04:12.038	Information	[Instrument Response] 24 Status 0 Motors initialized	GMT -5:00
12/15/23 14:04:12.085	Information	[Instrument Response] 24 Status 0 X-Motor configured	GMT -5:00
12/15/23 14:04:12.085	Information	[Instrument Response] 24 Status 0 DSP configured	GMT -5:00
12/15/23 14:04:12.131	Information	[Instrument Response] 24 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 14:04:12.131	Information	[Instrument Response] 24 Status 0 Cooling	GMT -5:00
12/15/23 14:04:12.131	Information	[Instrument Response] 24 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 14:04:12.131	Information	[Instrument Response] 24 Status 0 Create MergeImageChannel: 700 20 0 0 319 14 319 302 xofs=1 yofs=0	GMT -5:00
12/15/23 14:04:12.131	Information	[Instrument Response] 24 Status 0 Create MergeImageChannel: 800 20 0 0 319 14 319 302 xofs=0 yofs=0	GMT -5:00
12/15/23 14:04:12.131	Information	[Instrument Response] 24 Status 0 start=34 scan_lines=14 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 14:04:12.131	Information	[Instrument Response] 24 Status 0 start=34 scan_lines=28 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 14:04:12.131	Information	[Instrument Response] 24 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 14:04:12.678	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 14:04:12.741	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:04:13.194	Information	[Instrument Response] 24 Progress 0% 5 minutes	GMT -5:00
12/15/23 14:04:13.678	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:04:13.741	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:04:13.741	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:04:13.741	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:04:13.741	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:04:18.699	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      305 mA 	GMT -5:00
12/15/23 14:04:21.717	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:04:21.780	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:04:21.780	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:04:21.780	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:04:22.722	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:04:27.745	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      336 mA 	GMT -5:00
12/15/23 14:04:28.417	Information	[Instrument Response] 24 ImageBlock 700 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 14:04:28.433	Information	[Instrument Response] 24 ImageBlock 800 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 14:04:28.433	Information	Transferring image block at (0,0) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:04:28.464	Information	Transferring image block at (0,0) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:04:34.101	Information	[Instrument Response] 24 Progress 7% 4 minutes, 30 seconds	GMT -5:00
12/15/23 14:04:34.789	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         103 V  	GMT -5:00
12/15/23 14:04:34.836	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:04:34.836	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:04:34.836	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 14:04:35.789	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:04:41.688	Information	[Instrument Response] 24 ImageBlock 700 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 14:04:41.688	Information	[Instrument Response] 24 ImageBlock 800 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 14:04:41.703	Information	Transferring image block at (0,14) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:04:41.719	Information	Transferring image block at (0,14) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:04:47.843	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:04:47.906	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         115 V  	GMT -5:00
12/15/23 14:04:47.906	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:04:47.906	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:04:48.859	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:04:54.957	Information	[Instrument Response] 24 ImageBlock 700 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 14:04:54.957	Information	[Instrument Response] 24 ImageBlock 800 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 14:04:54.973	Information	Transferring image block at (0,28) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:04:54.989	Information	Transferring image block at (0,28) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:04:55.988	Error      	The 'Stop' command timed out. The elapsed time exceeded 120 seconds. Check the instrument connection.	GMT -5:00
12/15/23 14:05:00.925	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         126 V  	GMT -5:00
12/15/23 14:05:00.972	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:05:00.972	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:05:00.972	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 14:05:00.972	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:05:00.972	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 14:05:01.925	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:05:01.972	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:05:01.972	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 14:05:04.643	Information	[Instrument Response] 24 Progress 17% 4 minutes	GMT -5:00
12/15/23 14:05:08.236	Information	[Instrument Response] 24 ImageBlock 700 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 14:05:08.236	Information	[Instrument Response] 24 ImageBlock 800 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 14:05:08.236	Information	Transferring image block at (0,42) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:05:08.251	Information	Transferring image block at (0,42) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:05:13.985	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         151 V  	GMT -5:00
12/15/23 14:05:14.032	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         145 V  	GMT -5:00
12/15/23 14:05:14.032	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:05:14.032	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:05:14.032	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:05:14.984	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:05:15.047	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          85 V  	GMT -5:00
12/15/23 14:05:15.047	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:05:15.988	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:05:21.501	Information	[Instrument Response] 24 ImageBlock 700 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 14:05:21.501	Information	[Instrument Response] 24 ImageBlock 800 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 14:05:21.501	Information	Transferring image block at (0,56) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:05:21.532	Information	Transferring image block at (0,56) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:05:28.060	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/15/23 14:05:28.107	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:05:28.107	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:05:28.107	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:05:29.067	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:05:30.060	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      305 mA 	GMT -5:00
12/15/23 14:05:34.412	Information	[Instrument Response] 24 Progress 28% 3 minutes, 30 seconds	GMT -5:00
12/15/23 14:05:34.771	Information	[Instrument Response] 24 ImageBlock 700 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 14:05:34.771	Information	[Instrument Response] 24 ImageBlock 800 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 14:05:34.771	Information	Transferring image block at (0,70) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:05:34.802	Information	Transferring image block at (0,70) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:05:41.132	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:05:41.179	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         104 V  	GMT -5:00
12/15/23 14:05:41.179	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:05:41.179	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:05:42.132	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:05:48.042	Information	[Instrument Response] 24 ImageBlock 700 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 14:05:48.042	Information	[Instrument Response] 24 ImageBlock 800 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 14:05:48.042	Information	Transferring image block at (0,84) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:05:48.073	Information	Transferring image block at (0,84) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:05:54.197	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         114 V  	GMT -5:00
12/15/23 14:05:54.244	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:05:54.244	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:05:54.244	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:05:55.197	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:06:01.306	Information	[Instrument Response] 24 ImageBlock 700 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 14:06:01.306	Information	[Instrument Response] 24 ImageBlock 800 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 14:06:01.306	Information	Transferring image block at (0,98) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:06:01.337	Information	Transferring image block at (0,98) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:06:04.559	Information	[Instrument Response] 24 Progress 38% 3 minutes	GMT -5:00
12/15/23 14:06:07.267	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:06:07.314	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         127 V  	GMT -5:00
12/15/23 14:06:07.314	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:06:07.314	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:06:07.314	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:06:07.314	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 14:06:08.267	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:06:08.314	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:06:08.314	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 14:06:09.267	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:06:14.566	Information	[Instrument Response] 24 ImageBlock 700 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 14:06:14.566	Information	[Instrument Response] 24 ImageBlock 800 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 14:06:14.581	Information	Transferring image block at (0,112) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:06:14.597	Information	Transferring image block at (0,112) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:06:20.331	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         145 V  	GMT -5:00
12/15/23 14:06:20.378	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/15/23 14:06:20.378	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:06:20.378	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:06:20.378	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:06:21.331	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/15/23 14:06:21.378	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:06:21.378	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:06:22.331	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:06:27.833	Information	[Instrument Response] 24 ImageBlock 700 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 14:06:27.833	Information	[Instrument Response] 24 ImageBlock 800 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 14:06:27.849	Information	Transferring image block at (0,126) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:06:27.880	Information	Transferring image block at (0,126) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:06:33.518	Information	[Instrument Response] 24 Progress 49% 2 minutes, 30 seconds	GMT -5:00
12/15/23 14:06:34.393	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:06:34.455	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:06:34.455	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:06:34.455	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:06:35.408	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:06:41.115	Information	[Instrument Response] 24 ImageBlock 700 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 14:06:41.115	Information	[Instrument Response] 24 ImageBlock 800 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 14:06:41.115	Information	Transferring image block at (0,140) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:06:41.146	Information	Transferring image block at (0,140) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:06:47.467	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         103 V  	GMT -5:00
12/15/23 14:06:47.530	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:06:47.530	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:06:47.530	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:06:48.467	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:06:54.372	Information	[Instrument Response] 24 ImageBlock 700 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 14:06:54.372	Information	[Instrument Response] 24 ImageBlock 800 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 14:06:54.388	Information	Transferring image block at (0,154) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:06:54.403	Information	Transferring image block at (0,154) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:07:00.535	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:07:00.597	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         115 V  	GMT -5:00
12/15/23 14:07:00.597	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:07:00.597	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:07:01.534	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:07:04.472	Information	[Instrument Response] 24 Progress 59% 2 minutes	GMT -5:00
12/15/23 14:07:07.644	Information	[Instrument Response] 24 ImageBlock 700 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 14:07:07.644	Information	[Instrument Response] 24 ImageBlock 800 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 14:07:07.660	Information	Transferring image block at (0,168) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:07:07.675	Information	Transferring image block at (0,168) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:07:13.315	Information	[Instrument Response] 24 Progress 63% 1 minute, 50 seconds	GMT -5:00
12/15/23 14:07:13.596	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         126 V  	GMT -5:00
12/15/23 14:07:13.659	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:07:13.659	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:07:13.659	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:07:13.659	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/15/23 14:07:13.659	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 14:07:14.612	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/15/23 14:07:14.659	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:07:14.659	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 14:07:15.612	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:07:20.918	Information	[Instrument Response] 24 ImageBlock 700 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 14:07:20.918	Information	[Instrument Response] 24 ImageBlock 800 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 14:07:20.918	Information	Transferring image block at (0,182) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:07:20.949	Information	Transferring image block at (0,182) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:07:24.574	Information	[Instrument Response] 24 Progress 66% 1 minute, 40 seconds	GMT -5:00
12/15/23 14:07:26.670	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         151 V  	GMT -5:00
12/15/23 14:07:26.717	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         145 V  	GMT -5:00
12/15/23 14:07:26.717	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:07:26.717	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:07:26.717	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:07:27.670	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:07:27.733	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          85 V  	GMT -5:00
12/15/23 14:07:27.733	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:07:28.673	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:07:34.188	Information	[Instrument Response] 24 ImageBlock 700 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 14:07:34.188	Information	[Instrument Response] 24 ImageBlock 800 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 14:07:34.188	Information	Transferring image block at (0,196) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:07:34.219	Information	Transferring image block at (0,196) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:07:34.625	Information	[Instrument Response] 24 Progress 70% 1 minute, 30 seconds	GMT -5:00
12/15/23 14:07:40.736	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/15/23 14:07:40.799	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:07:40.799	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:07:40.799	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:07:41.742	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:07:44.273	Information	[Instrument Response] 24 Progress 73% 1 minute, 20 seconds	GMT -5:00
12/15/23 14:07:47.451	Information	[Instrument Response] 24 ImageBlock 700 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 14:07:47.451	Information	[Instrument Response] 24 ImageBlock 800 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 14:07:47.451	Information	Transferring image block at (0,210) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:07:47.482	Information	Transferring image block at (0,210) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:07:53.122	Information	[Instrument Response] 24 Progress 76% 1 minute, 10 seconds	GMT -5:00
12/15/23 14:07:53.819	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:07:53.873	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         104 V  	GMT -5:00
12/15/23 14:07:53.873	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:07:53.873	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:07:54.820	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:08:00.722	Information	[Instrument Response] 24 ImageBlock 700 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 14:08:00.722	Information	[Instrument Response] 24 ImageBlock 800 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 14:08:00.722	Information	Transferring image block at (0,224) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:08:00.753	Information	Transferring image block at (0,224) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:08:04.379	Information	[Instrument Response] 24 Progress 80% 1 minute	GMT -5:00
12/15/23 14:08:06.879	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         114 V  	GMT -5:00
12/15/23 14:08:06.925	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:08:06.925	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:08:06.925	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:08:07.888	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:08:13.987	Information	[Instrument Response] 24 ImageBlock 700 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 14:08:13.987	Information	[Instrument Response] 24 ImageBlock 800 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 14:08:14.002	Information	Transferring image block at (0,238) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:08:14.018	Information	Transferring image block at (0,238) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:08:14.424	Information	[Instrument Response] 24 Progress 84% 50 seconds	GMT -5:00
12/15/23 14:08:19.939	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:08:20.002	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         127 V  	GMT -5:00
12/15/23 14:08:20.002	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:08:20.002	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:08:20.002	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:08:20.002	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 14:08:20.955	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:08:21.001	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:08:21.001	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/15/23 14:08:21.954	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:08:24.485	Information	[Instrument Response] 24 Progress 87% 40 seconds	GMT -5:00
12/15/23 14:08:27.250	Information	[Instrument Response] 24 ImageBlock 700 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 14:08:27.266	Information	[Instrument Response] 24 ImageBlock 800 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 14:08:27.266	Information	Transferring image block at (0,252) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:08:27.281	Information	Transferring image block at (0,252) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:08:32.931	Information	[Instrument Response] 24 Progress 90% 30 seconds	GMT -5:00
12/15/23 14:08:33.009	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         145 V  	GMT -5:00
12/15/23 14:08:33.071	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/15/23 14:08:33.071	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:08:33.071	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:08:33.071	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:08:34.009	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/15/23 14:08:34.071	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:08:34.071	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:08:35.024	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:08:40.530	Information	[Instrument Response] 24 ImageBlock 700 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 14:08:40.530	Information	[Instrument Response] 24 ImageBlock 800 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 14:08:40.530	Information	Transferring image block at (0,266) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:08:40.561	Information	Transferring image block at (0,266) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:08:44.593	Information	[Instrument Response] 24 Progress 94% 20 seconds	GMT -5:00
12/15/23 14:08:47.092	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:08:47.139	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:08:47.139	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:08:47.139	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:08:48.092	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:08:53.800	Information	[Instrument Response] 24 ImageBlock 700 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 14:08:53.800	Information	[Instrument Response] 24 ImageBlock 800 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 14:08:53.800	Information	Transferring image block at (0,280) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:08:53.831	Information	Transferring image block at (0,280) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:08:54.633	Information	[Instrument Response] 24 Progress 98% 10 seconds	GMT -5:00
12/15/23 14:08:57.127	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         145 V  	GMT -5:00
12/15/23 14:08:57.189	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/15/23 14:08:57.189	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:08:57.189	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:08:57.189	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:08:58.142	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/15/23 14:08:58.189	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:08:58.189	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:08:59.142	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:09:02.267	Information	[Instrument Response] 24 Progress 100% 0 seconds	GMT -5:00
12/15/23 14:09:03.157	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:09:03.220	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:09:03.220	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:09:03.220	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:09:04.172	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:09:04.266	Information	[Instrument Response] 24 ImageBlock 700 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 14:09:04.282	Information	[Instrument Response] 24 ImageBlock 800 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 14:09:04.282	Information	[Instrument Response] 24 Status 0 COMPLETE SCAN 291.071 seconds	GMT -5:00
12/15/23 14:09:04.282	Information	Transferring image block at (0,294) with size 319x8 to the 700 channel.	GMT -5:00
12/15/23 14:09:04.282	Information	[Instrument Response] 24 Status 0 Averages (sec): setup=2.00005 line=0.402433 => total=246	GMT -5:00
12/15/23 14:09:04.313	Information	Transferring image block at (0,294) with size 319x8 to the 800 channel.	GMT -5:00
12/15/23 14:09:04.313	Information	[Instrument Response] 24 Status 0 Comparisons: setup=1.00997 line=0.950274	GMT -5:00
12/15/23 14:09:04.360	Information	[Instrument Response] 24 Status 0 Finalize motors	GMT -5:00
12/15/23 14:09:04.360	Information	[Instrument Response] 24 Progress 100% Scan complete	GMT -5:00
12/15/23 14:09:04.391	Information	[Instrument Response] 24 Result 0 OK	GMT -5:00
12/15/23 14:09:04.391	Information	Acquisition 0000012_01 finished with channels: 700 800.	GMT -5:00
12/15/23 14:09:04.641	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/15/23 14:09:04.641	Information	Scan completed successfully	GMT -5:00
12/15/23 14:09:05.161	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:09:05.224	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          60 V  	GMT -5:00
12/15/23 14:09:05.224	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 14:09:05.224	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 14:09:05.224	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 14:09:06.172	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          57 V  	GMT -5:00
12/15/23 14:09:06.232	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          36 V  	GMT -5:00
12/15/23 14:09:07.182	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          35 V  	GMT -5:00
12/15/23 14:09:07.242	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          22 V  	GMT -5:00
12/15/23 14:09:08.239	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          24 V  	GMT -5:00
12/15/23 14:09:08.286	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          15 V  	GMT -5:00
12/15/23 14:09:09.188	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          17 V  	GMT -5:00
12/15/23 14:09:09.234	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          10 V  	GMT -5:00
12/15/23 14:09:10.190	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          11 V  	GMT -5:00
12/15/23 14:09:10.237	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           7 V  	GMT -5:00
12/15/23 14:09:11.191	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           8 V  	GMT -5:00
12/15/23 14:09:11.253	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           4 V  	GMT -5:00
12/15/23 14:09:12.190	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/15/23 14:09:13.192	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/15/23 14:09:13.255	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/15/23 14:09:14.209	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/15/23 14:09:17.215	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 14:09:18.213	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/15/23 14:09:32.683	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/15/23 14:10:11.909	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 14:10:25.747	Information	Starting acquisition...	GMT -5:00
12/15/23 14:10:26.684	Information	[Instrument Request]  25 ScanAuto 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 14:10:26.747	Information	[Instrument Response] 25 Status 1 Executing command: ScanAuto 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Status 0 Computed scan parameters	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Status 0 Blocks=23	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Progress 0% 5 minutes	GMT -5:00
12/15/23 14:10:26.762	Information	[Instrument Response] 25 Progress 0% Initializing	GMT -5:00
12/15/23 14:10:26.778	Information	[Instrument Response] 25 Status 0 Preparation complete	GMT -5:00
12/15/23 14:10:29.704	Information	[Instrument Response] 25 Status 0 Motors initialized	GMT -5:00
12/15/23 14:10:29.751	Information	[Instrument Response] 25 Status 0 X-Motor configured	GMT -5:00
12/15/23 14:10:29.751	Information	[Instrument Response] 25 Status 0 DSP configured	GMT -5:00
12/15/23 14:10:29.798	Information	[Instrument Response] 25 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 14:10:29.798	Information	[Instrument Response] 25 Status 0 Cooling	GMT -5:00
12/15/23 14:10:29.798	Information	[Instrument Response] 25 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 14:10:29.798	Information	[Instrument Response] 25 Status 0 Create MergeImageChannel: 700 20 0 0 319 14 319 302 xofs=1 yofs=0	GMT -5:00
12/15/23 14:10:29.798	Information	[Instrument Response] 25 Status 0 Create MergeImageChannel: 800 20 0 0 319 14 319 302 xofs=0 yofs=0	GMT -5:00
12/15/23 14:10:29.798	Information	[Instrument Response] 25 Status 0 start=34 scan_lines=14 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 14:10:29.798	Information	[Instrument Response] 25 Status 0 start=34 scan_lines=28 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 14:10:29.798	Information	[Instrument Response] 25 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 14:10:30.507	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 14:10:30.554	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:10:33.745	Information	[Instrument Response] 25 Progress 0% 5 minutes	GMT -5:00
12/15/23 14:10:34.510	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:10:34.573	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:10:34.573	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:10:34.573	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:10:34.573	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:10:36.523	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:10:38.531	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      305 mA 	GMT -5:00
12/15/23 14:10:39.531	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:10:41.541	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 14:10:41.603	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:10:41.603	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:10:41.603	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:10:41.603	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:10:42.556	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 14:10:42.603	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:10:43.549	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:10:44.560	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:10:45.560	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:10:46.560	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:10:47.576	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      336 mA 	GMT -5:00
12/15/23 14:10:47.622	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:10:48.966	Information	[Instrument Response] 25 ImageBlock 700 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 14:10:48.981	Information	[Instrument Response] 25 ImageBlock 800 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 14:10:48.981	Information	Transferring image block at (0,0) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:10:49.013	Information	Transferring image block at (0,0) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:10:54.643	Information	[Instrument Response] 25 Progress 7% 4 minutes, 30 seconds	GMT -5:00
12/15/23 14:10:55.616	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 14:10:55.678	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:10:55.678	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:10:55.678	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:10:56.616	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:10:58.631	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:10:59.630	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:11:02.241	Information	[Instrument Response] 25 ImageBlock 700 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 14:11:02.241	Information	[Instrument Response] 25 ImageBlock 800 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 14:11:02.241	Information	Transferring image block at (0,14) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:11:02.272	Information	Transferring image block at (0,14) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:11:08.688	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:11:08.735	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          99 V  	GMT -5:00
12/15/23 14:11:08.735	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:11:08.735	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:11:09.688	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:11:10.688	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:11:12.703	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:11:15.516	Information	[Instrument Response] 25 ImageBlock 700 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 14:11:15.516	Information	[Instrument Response] 25 ImageBlock 800 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 14:11:15.516	Information	Transferring image block at (0,28) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:11:15.547	Information	Transferring image block at (0,28) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:11:21.749	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/15/23 14:11:21.812	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:11:21.812	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:11:21.812	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:11:22.753	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:11:24.768	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:11:25.190	Information	[Instrument Response] 25 Progress 17% 4 minutes	GMT -5:00
12/15/23 14:11:26.768	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:11:28.783	Information	[Instrument Response] 25 ImageBlock 700 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 14:11:28.783	Information	[Instrument Response] 25 ImageBlock 800 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 14:11:28.783	Information	Transferring image block at (0,42) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:11:28.814	Information	Transferring image block at (0,42) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:11:34.817	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:11:34.864	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/15/23 14:11:34.864	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:11:34.864	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:11:35.818	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:11:36.818	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:11:37.833	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:11:42.048	Information	[Instrument Response] 25 ImageBlock 700 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 14:11:42.048	Information	[Instrument Response] 25 ImageBlock 800 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 14:11:42.048	Information	Transferring image block at (0,56) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:11:42.079	Information	Transferring image block at (0,56) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:11:47.887	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:11:47.934	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:11:47.934	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:11:47.934	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:11:47.934	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:11:48.891	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 14:11:48.938	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:11:49.891	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:11:50.890	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:11:51.898	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:11:54.962	Information	[Instrument Response] 25 Progress 28% 3 minutes, 30 seconds	GMT -5:00
12/15/23 14:11:55.321	Information	[Instrument Response] 25 ImageBlock 700 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 14:11:55.321	Information	[Instrument Response] 25 ImageBlock 800 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 14:11:55.321	Information	Transferring image block at (0,70) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:11:55.352	Information	Transferring image block at (0,70) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:12:01.965	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:12:02.012	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/15/23 14:12:02.012	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:12:02.012	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:12:02.965	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:12:04.965	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:12:05.981	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:12:08.591	Information	[Instrument Response] 25 ImageBlock 700 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 14:12:08.591	Information	[Instrument Response] 25 ImageBlock 800 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 14:12:08.591	Information	Transferring image block at (0,84) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:12:08.622	Information	Transferring image block at (0,84) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:12:15.028	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          98 V  	GMT -5:00
12/15/23 14:12:15.075	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:12:15.075	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:12:15.075	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:12:16.028	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:12:17.028	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:12:19.036	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:12:21.849	Information	[Instrument Response] 25 ImageBlock 700 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 14:12:21.849	Information	[Instrument Response] 25 ImageBlock 800 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 14:12:21.864	Information	Transferring image block at (0,98) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:12:21.880	Information	Transferring image block at (0,98) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:12:25.114	Information	[Instrument Response] 25 Progress 38% 3 minutes	GMT -5:00
12/15/23 14:12:28.093	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:12:28.156	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 14:12:28.156	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:12:28.156	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:12:29.093	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:12:31.108	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:12:33.108	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:12:35.124	Information	[Instrument Response] 25 ImageBlock 700 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 14:12:35.124	Information	[Instrument Response] 25 ImageBlock 800 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 14:12:35.124	Information	Transferring image block at (0,112) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:12:35.156	Information	Transferring image block at (0,112) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:12:41.156	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/15/23 14:12:41.219	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:12:41.219	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:12:41.219	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:12:42.156	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:12:48.384	Information	[Instrument Response] 25 ImageBlock 700 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 14:12:48.384	Information	[Instrument Response] 25 ImageBlock 800 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 14:12:48.399	Information	Transferring image block at (0,126) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:12:48.415	Information	Transferring image block at (0,126) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:12:54.060	Information	[Instrument Response] 25 Progress 49% 2 minutes, 30 seconds	GMT -5:00
12/15/23 14:12:54.232	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 14:12:54.279	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:12:54.279	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:12:54.279	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:12:54.279	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:12:55.232	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 14:12:55.279	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:12:56.241	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:12:57.235	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:12:58.250	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:13:01.656	Information	[Instrument Response] 25 ImageBlock 700 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 14:13:01.656	Information	[Instrument Response] 25 ImageBlock 800 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 14:13:01.671	Information	Transferring image block at (0,140) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:13:01.687	Information	Transferring image block at (0,140) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:13:08.296	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 14:13:08.358	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:13:08.358	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:13:08.358	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:13:09.312	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:13:11.311	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:13:12.313	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:13:14.933	Information	[Instrument Response] 25 ImageBlock 700 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 14:13:14.933	Information	[Instrument Response] 25 ImageBlock 800 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 14:13:14.933	Information	Transferring image block at (0,154) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:13:14.964	Information	Transferring image block at (0,154) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:13:21.363	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:13:21.425	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          99 V  	GMT -5:00
12/15/23 14:13:21.425	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:13:21.425	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:13:22.378	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:13:23.378	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:13:25.018	Information	[Instrument Response] 25 Progress 59% 2 minutes	GMT -5:00
12/15/23 14:13:25.378	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:13:28.199	Information	[Instrument Response] 25 ImageBlock 700 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 14:13:28.199	Information	[Instrument Response] 25 ImageBlock 800 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 14:13:28.199	Information	Transferring image block at (0,168) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:13:28.230	Information	Transferring image block at (0,168) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:13:33.867	Information	[Instrument Response] 25 Progress 63% 1 minute, 50 seconds	GMT -5:00
12/15/23 14:13:34.445	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/15/23 14:13:34.492	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:13:34.492	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:13:34.492	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:13:35.444	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:13:37.448	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:13:39.463	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:13:41.466	Information	[Instrument Response] 25 ImageBlock 700 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 14:13:41.466	Information	[Instrument Response] 25 ImageBlock 800 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 14:13:41.466	Information	Transferring image block at (0,182) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:13:41.497	Information	Transferring image block at (0,182) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:13:45.123	Information	[Instrument Response] 25 Progress 66% 1 minute, 40 seconds	GMT -5:00
12/15/23 14:13:47.501	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:13:47.548	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/15/23 14:13:47.548	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:13:47.548	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:13:48.501	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:13:54.730	Information	[Instrument Response] 25 ImageBlock 700 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 14:13:54.730	Information	[Instrument Response] 25 ImageBlock 800 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 14:13:54.745	Information	Transferring image block at (0,196) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:13:54.761	Information	Transferring image block at (0,196) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:13:55.183	Information	[Instrument Response] 25 Progress 70% 1 minute, 30 seconds	GMT -5:00
12/15/23 14:14:00.567	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:14:00.629	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:14:00.629	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:14:00.629	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:14:00.629	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:14:01.567	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 14:14:01.629	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:14:02.582	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:14:03.582	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:14:04.582	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:14:04.816	Information	[Instrument Response] 25 Progress 73% 1 minute, 20 seconds	GMT -5:00
12/15/23 14:14:08.004	Information	[Instrument Response] 25 ImageBlock 700 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 14:14:08.004	Information	[Instrument Response] 25 ImageBlock 800 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 14:14:08.004	Information	Transferring image block at (0,210) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:14:08.035	Information	Transferring image block at (0,210) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:14:13.670	Information	[Instrument Response] 25 Progress 76% 1 minute, 10 seconds	GMT -5:00
12/15/23 14:14:14.645	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:14:14.692	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/15/23 14:14:14.692	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:14:14.692	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:14:15.645	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:14:17.654	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:14:18.654	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:14:21.263	Information	[Instrument Response] 25 ImageBlock 700 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 14:14:21.263	Information	[Instrument Response] 25 ImageBlock 800 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 14:14:21.279	Information	Transferring image block at (0,224) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:14:21.294	Information	Transferring image block at (0,224) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:14:24.936	Information	[Instrument Response] 25 Progress 80% 1 minute	GMT -5:00
12/15/23 14:14:27.717	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          98 V  	GMT -5:00
12/15/23 14:14:27.763	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:14:27.763	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:14:27.763	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:14:28.716	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:14:29.716	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:14:31.732	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:14:34.544	Information	[Instrument Response] 25 ImageBlock 700 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 14:14:34.544	Information	[Instrument Response] 25 ImageBlock 800 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 14:14:34.544	Information	Transferring image block at (0,238) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:14:34.575	Information	Transferring image block at (0,238) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:14:34.981	Information	[Instrument Response] 25 Progress 84% 50 seconds	GMT -5:00
12/15/23 14:14:40.777	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:14:40.840	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 14:14:40.840	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:14:40.840	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:14:41.777	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:14:43.792	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:14:45.026	Information	[Instrument Response] 25 Progress 87% 40 seconds	GMT -5:00
12/15/23 14:14:45.807	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:14:47.806	Information	[Instrument Response] 25 ImageBlock 700 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 14:14:47.806	Information	[Instrument Response] 25 ImageBlock 800 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 14:14:47.806	Information	Transferring image block at (0,252) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:14:47.837	Information	Transferring image block at (0,252) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:14:53.480	Information	[Instrument Response] 25 Progress 90% 30 seconds	GMT -5:00
12/15/23 14:14:53.839	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/15/23 14:14:53.902	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:14:53.902	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:14:53.902	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:14:54.839	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:14:55.855	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:14:56.854	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:15:01.075	Information	[Instrument Response] 25 ImageBlock 700 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 14:15:01.075	Information	[Instrument Response] 25 ImageBlock 800 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 14:15:01.075	Information	Transferring image block at (0,266) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:15:01.106	Information	Transferring image block at (0,266) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:15:05.131	Information	[Instrument Response] 25 Progress 94% 20 seconds	GMT -5:00
12/15/23 14:15:06.916	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 14:15:06.963	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:15:06.963	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:15:06.963	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:15:06.963	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:15:07.916	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 14:15:07.978	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:15:08.925	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:15:14.344	Information	[Instrument Response] 25 ImageBlock 700 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 14:15:14.344	Information	[Instrument Response] 25 ImageBlock 800 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 14:15:14.344	Information	Transferring image block at (0,280) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:15:14.375	Information	Transferring image block at (0,280) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:15:15.188	Information	[Instrument Response] 25 Progress 98% 10 seconds	GMT -5:00
12/15/23 14:15:17.965	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/15/23 14:15:18.012	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:15:18.012	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:15:18.012	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:15:18.965	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:15:22.818	Information	[Instrument Response] 25 Progress 100% 0 seconds	GMT -5:00
12/15/23 14:15:22.990	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 14:15:23.037	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:15:23.037	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:15:23.037	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:15:23.990	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 14:15:24.818	Information	[Instrument Response] 25 ImageBlock 700 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 14:15:24.818	Information	[Instrument Response] 25 ImageBlock 800 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 14:15:24.818	Information	[Instrument Response] 25 Status 0 COMPLETE SCAN 291.071 seconds	GMT -5:00
12/15/23 14:15:24.833	Information	Transferring image block at (0,294) with size 319x8 to the 700 channel.	GMT -5:00
12/15/23 14:15:24.833	Information	[Instrument Response] 25 Status 0 Averages (sec): setup=2.00005 line=0.402433 => total=246	GMT -5:00
12/15/23 14:15:24.849	Information	Transferring image block at (0,294) with size 319x8 to the 800 channel.	GMT -5:00
12/15/23 14:15:24.865	Information	[Instrument Response] 25 Status 0 Comparisons: setup=1.00997 line=0.950274	GMT -5:00
12/15/23 14:15:24.911	Information	[Instrument Response] 25 Status 0 Finalize motors	GMT -5:00
12/15/23 14:15:24.911	Information	[Instrument Response] 25 Progress 100% Scan complete	GMT -5:00
12/15/23 14:15:24.943	Information	[Instrument Response] 25 Result 0 OK	GMT -5:00
12/15/23 14:15:24.943	Information	Acquisition 0000013_01 finished with channels: 700 800.	GMT -5:00
12/15/23 14:15:25.146	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/15/23 14:15:25.146	Information	Scan completed successfully	GMT -5:00
12/15/23 14:15:25.255	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         142 V  	GMT -5:00
12/15/23 14:15:25.271	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 14:15:25.271	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 14:15:25.286	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 14:15:25.993	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          83 V  	GMT -5:00
12/15/23 14:15:26.055	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          52 V  	GMT -5:00
12/15/23 14:15:27.009	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          50 V  	GMT -5:00
12/15/23 14:15:27.069	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          32 V  	GMT -5:00
12/15/23 14:15:27.069	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      305 mA 	GMT -5:00
12/15/23 14:15:28.008	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          32 V  	GMT -5:00
12/15/23 14:15:28.068	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          20 V  	GMT -5:00
12/15/23 14:15:29.051	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          21 V  	GMT -5:00
12/15/23 14:15:29.105	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          13 V  	GMT -5:00
12/15/23 14:15:30.007	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          14 V  	GMT -5:00
12/15/23 14:15:30.069	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           9 V  	GMT -5:00
12/15/23 14:15:31.025	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          10 V  	GMT -5:00
12/15/23 14:15:31.072	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           7 V  	GMT -5:00
12/15/23 14:15:32.027	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           7 V  	GMT -5:00
12/15/23 14:15:32.073	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           4 V  	GMT -5:00
12/15/23 14:15:33.028	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/15/23 14:15:34.030	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/15/23 14:15:35.033	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/15/23 14:15:37.035	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/15/23 14:15:38.037	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 14:15:39.048	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/15/23 14:18:26.748	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/15/23 14:19:14.243	Information	Starting acquisition...	GMT -5:00
12/15/23 14:19:15.087	Information	[Instrument Request]  26 ScanAuto 169um medium 0.0 0.0 5.4 0.0 5.1 700 800 	GMT -5:00
12/15/23 14:19:19.961	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 1 Executing command: ScanAuto 169um medium 0 0 5.4 0 5.1 700 800	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 0 X mapping: [x0=0,x1=5.4] => [left=-60,right=2500] w=319	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 0 Computed scan parameters	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 0 Seconds/line=0.360775	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 0 linesPerBlock=28 flushLinesPerBlock=14	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 0 Blocks=23	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Status 0 Estimates (sec): setup=2.02 line=0.382421 => total=234	GMT -5:00
12/15/23 14:19:20.008	Information	[Instrument Response] 26 Progress 0% 5 minutes	GMT -5:00
12/15/23 14:19:20.023	Information	[Instrument Response] 26 Progress 0% Initializing	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 Preparation complete	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 Motors initialized	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 X-Motor configured	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 DSP configured	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 Digital / DSP configured	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 Cooling	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 Create MergeImageChannel: 700 20 0 0 319 14 319 302 xofs=1 yofs=0	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 Create MergeImageChannel: 800 20 0 0 319 14 319 302 xofs=0 yofs=0	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 start=34 scan_lines=14 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 start=34 scan_lines=28 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 26 Status 0 Move motors to initial positions	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    755 mV 	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:19:20.039	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    742 mV 	GMT -5:00
12/15/23 14:19:20.899	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:19:22.148	Information	[Instrument Response] 26 Progress 0% 5 minutes	GMT -5:00
12/15/23 14:19:22.917	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:19:22.964	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:19:22.964	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:19:22.964	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:19:22.964	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:19:24.919	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:19:25.919	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:19:26.934	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      305 mA 	GMT -5:00
12/15/23 14:19:29.938	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 14:19:30.001	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:19:30.001	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:19:30.001	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:19:30.001	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:19:30.952	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 14:19:30.999	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      336 mA 	GMT -5:00
12/15/23 14:19:30.999	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:19:31.952	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:19:32.951	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    755 mV 	GMT -5:00
12/15/23 14:19:33.961	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    742 mV 	GMT -5:00
12/15/23 14:19:37.370	Information	[Instrument Response] 26 ImageBlock 700 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 14:19:37.370	Information	[Instrument Response] 26 ImageBlock 800 0 0 319 14 319 302 8932	GMT -5:00
12/15/23 14:19:37.385	Information	Transferring image block at (0,0) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:19:37.401	Information	Transferring image block at (0,0) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:19:43.038	Information	[Instrument Response] 26 Progress 7% 4 minutes, 30 seconds	GMT -5:00
12/15/23 14:19:44.017	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 14:19:44.064	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:19:44.064	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:19:44.064	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:19:45.017	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:19:47.032	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:19:48.032	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:19:50.645	Information	[Instrument Response] 26 ImageBlock 700 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 14:19:50.645	Information	[Instrument Response] 26 ImageBlock 800 0 14 319 14 319 302 8932	GMT -5:00
12/15/23 14:19:50.645	Information	Transferring image block at (0,14) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:19:50.676	Information	Transferring image block at (0,14) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:19:57.088	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:19:57.135	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          99 V  	GMT -5:00
12/15/23 14:19:57.135	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:19:57.135	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:19:58.084	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:19:59.083	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:20:01.105	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:20:03.907	Information	[Instrument Response] 26 ImageBlock 700 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 14:20:03.907	Information	[Instrument Response] 26 ImageBlock 800 0 28 319 14 319 302 8932	GMT -5:00
12/15/23 14:20:03.907	Information	Transferring image block at (0,28) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:20:03.954	Information	Transferring image block at (0,28) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:20:10.156	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/15/23 14:20:10.203	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:20:10.203	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:20:10.203	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:20:11.156	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:20:13.165	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:20:13.603	Information	[Instrument Response] 26 Progress 17% 4 minutes	GMT -5:00
12/15/23 14:20:15.165	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:20:17.173	Information	[Instrument Response] 26 ImageBlock 700 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 14:20:17.173	Information	[Instrument Response] 26 ImageBlock 800 0 42 319 14 319 302 8932	GMT -5:00
12/15/23 14:20:17.189	Information	Transferring image block at (0,42) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:20:17.204	Information	Transferring image block at (0,42) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:20:23.219	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:20:23.266	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/15/23 14:20:23.266	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:20:23.266	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:20:24.219	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:20:30.448	Information	[Instrument Response] 26 ImageBlock 700 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 14:20:30.448	Information	[Instrument Response] 26 ImageBlock 800 0 56 319 14 319 302 8932	GMT -5:00
12/15/23 14:20:30.448	Information	Transferring image block at (0,56) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:20:30.479	Information	Transferring image block at (0,56) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:20:36.282	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:20:36.344	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:20:36.344	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:20:36.344	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/15/23 14:20:36.344	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:20:37.285	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 14:20:37.348	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:20:38.285	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:20:39.300	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:20:40.300	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:20:43.346	Information	[Instrument Response] 26 Progress 28% 3 minutes, 30 seconds	GMT -5:00
12/15/23 14:20:43.721	Information	[Instrument Response] 26 ImageBlock 700 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 14:20:43.721	Information	[Instrument Response] 26 ImageBlock 800 0 70 319 14 319 302 8932	GMT -5:00
12/15/23 14:20:43.721	Information	Transferring image block at (0,70) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:20:43.752	Information	Transferring image block at (0,70) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:20:50.355	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:20:50.418	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/15/23 14:20:50.418	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:20:50.418	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:20:51.356	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:20:53.371	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:20:54.371	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:20:56.983	Information	[Instrument Response] 26 ImageBlock 700 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 14:20:56.983	Information	[Instrument Response] 26 ImageBlock 800 0 84 319 14 319 302 8932	GMT -5:00
12/15/23 14:20:56.983	Information	Transferring image block at (0,84) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:20:57.014	Information	Transferring image block at (0,84) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:21:03.428	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          98 V  	GMT -5:00
12/15/23 14:21:03.475	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:21:03.475	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:21:03.475	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:21:04.428	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:21:05.428	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:21:07.440	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:21:10.256	Information	[Instrument Response] 26 ImageBlock 700 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 14:21:10.256	Information	[Instrument Response] 26 ImageBlock 800 0 98 319 14 319 302 8932	GMT -5:00
12/15/23 14:21:10.256	Information	Transferring image block at (0,98) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:21:10.287	Information	Transferring image block at (0,98) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:21:13.509	Information	[Instrument Response] 26 Progress 38% 3 minutes	GMT -5:00
12/15/23 14:21:16.493	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:21:16.555	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 14:21:16.555	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:21:16.555	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:21:17.493	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:21:19.509	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    755 mV 	GMT -5:00
12/15/23 14:21:19.556	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:21:20.509	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    742 mV 	GMT -5:00
12/15/23 14:21:21.509	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:21:23.524	Information	[Instrument Response] 26 ImageBlock 700 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 14:21:23.524	Information	[Instrument Response] 26 ImageBlock 800 0 112 319 14 319 302 8932	GMT -5:00
12/15/23 14:21:23.524	Information	Transferring image block at (0,112) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:21:23.555	Information	Transferring image block at (0,112) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:21:29.554	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/15/23 14:21:29.617	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:21:29.617	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:21:29.617	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:21:30.564	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:21:31.564	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:21:32.564	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:21:36.783	Information	[Instrument Response] 26 ImageBlock 700 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 14:21:36.783	Information	[Instrument Response] 26 ImageBlock 800 0 126 319 14 319 302 8932	GMT -5:00
12/15/23 14:21:36.798	Information	Transferring image block at (0,126) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:21:36.829	Information	Transferring image block at (0,126) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:21:42.468	Information	[Instrument Response] 26 Progress 49% 2 minutes, 30 seconds	GMT -5:00
12/15/23 14:21:42.624	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 14:21:42.687	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:21:42.687	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:21:42.687	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:21:42.687	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:21:43.633	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 14:21:43.680	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:21:44.627	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:21:45.642	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:21:46.635	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:21:50.050	Information	[Instrument Response] 26 ImageBlock 700 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 14:21:50.065	Information	[Instrument Response] 26 ImageBlock 800 0 140 319 14 319 302 8932	GMT -5:00
12/15/23 14:21:50.065	Information	Transferring image block at (0,140) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:21:50.097	Information	Transferring image block at (0,140) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:21:56.702	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 14:21:56.749	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:21:56.749	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:21:56.749	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:21:57.702	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:21:59.717	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:22:00.717	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:22:03.326	Information	[Instrument Response] 26 ImageBlock 700 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 14:22:03.326	Information	[Instrument Response] 26 ImageBlock 800 0 154 319 14 319 302 8932	GMT -5:00
12/15/23 14:22:03.326	Information	Transferring image block at (0,154) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:22:03.357	Information	Transferring image block at (0,154) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:22:09.772	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:22:09.819	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          99 V  	GMT -5:00
12/15/23 14:22:09.819	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:22:09.819	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:22:10.772	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:22:11.772	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:22:13.412	Information	[Instrument Response] 26 Progress 59% 2 minutes	GMT -5:00
12/15/23 14:22:13.787	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:22:16.599	Information	[Instrument Response] 26 ImageBlock 700 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 14:22:16.599	Information	[Instrument Response] 26 ImageBlock 800 0 168 319 14 319 302 8932	GMT -5:00
12/15/23 14:22:16.599	Information	Transferring image block at (0,168) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:22:16.630	Information	Transferring image block at (0,168) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:22:22.265	Information	[Instrument Response] 26 Progress 63% 1 minute, 50 seconds	GMT -5:00
12/15/23 14:22:22.829	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/15/23 14:22:22.892	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:22:22.892	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:22:22.892	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:22:23.845	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:22:25.845	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:22:27.857	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:22:29.857	Information	[Instrument Response] 26 ImageBlock 700 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 14:22:29.857	Information	[Instrument Response] 26 ImageBlock 800 0 182 319 14 319 302 8932	GMT -5:00
12/15/23 14:22:29.872	Information	Transferring image block at (0,182) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:22:29.888	Information	Transferring image block at (0,182) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:22:33.529	Information	[Instrument Response] 26 Progress 66% 1 minute, 40 seconds	GMT -5:00
12/15/23 14:22:35.904	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:22:35.951	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/15/23 14:22:35.951	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:22:35.951	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:22:36.904	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:22:37.914	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:22:38.911	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:22:43.130	Information	[Instrument Response] 26 ImageBlock 700 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 14:22:43.130	Information	[Instrument Response] 26 ImageBlock 800 0 196 319 14 319 302 8932	GMT -5:00
12/15/23 14:22:43.130	Information	Transferring image block at (0,196) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:22:43.161	Information	Transferring image block at (0,196) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:22:43.567	Information	[Instrument Response] 26 Progress 70% 1 minute, 30 seconds	GMT -5:00
12/15/23 14:22:48.961	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:22:49.024	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:22:49.024	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:22:49.024	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:22:49.024	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:22:49.024	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/15/23 14:22:49.977	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/15/23 14:22:50.024	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:22:50.024	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 14:22:50.977	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:22:53.227	Information	[Instrument Response] 26 Progress 73% 1 minute, 20 seconds	GMT -5:00
12/15/23 14:22:56.404	Information	[Instrument Response] 26 ImageBlock 700 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 14:22:56.404	Information	[Instrument Response] 26 ImageBlock 800 0 210 319 14 319 302 8932	GMT -5:00
12/15/23 14:22:56.404	Information	Transferring image block at (0,210) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:22:56.435	Information	Transferring image block at (0,210) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:23:02.066	Information	[Instrument Response] 26 Progress 76% 1 minute, 10 seconds	GMT -5:00
12/15/23 14:23:03.035	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:23:03.097	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/15/23 14:23:03.097	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:23:03.097	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:23:04.050	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:23:06.050	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    755 mV 	GMT -5:00
12/15/23 14:23:07.049	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    742 mV 	GMT -5:00
12/15/23 14:23:09.675	Information	[Instrument Response] 26 ImageBlock 700 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 14:23:09.675	Information	[Instrument Response] 26 ImageBlock 800 0 224 319 14 319 302 8932	GMT -5:00
12/15/23 14:23:09.675	Information	Transferring image block at (0,224) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:23:09.706	Information	Transferring image block at (0,224) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:23:13.331	Information	[Instrument Response] 26 Progress 80% 1 minute	GMT -5:00
12/15/23 14:23:16.114	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          99 V  	GMT -5:00
12/15/23 14:23:16.161	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/15/23 14:23:16.161	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:23:16.161	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:23:17.114	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/15/23 14:23:18.114	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:23:20.120	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:23:22.932	Information	[Instrument Response] 26 ImageBlock 700 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 14:23:22.932	Information	[Instrument Response] 26 ImageBlock 800 0 238 319 14 319 302 8932	GMT -5:00
12/15/23 14:23:22.947	Information	Transferring image block at (0,238) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:23:22.963	Information	Transferring image block at (0,238) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:23:23.385	Information	[Instrument Response] 26 Progress 84% 50 seconds	GMT -5:00
12/15/23 14:23:29.182	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:23:29.229	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/15/23 14:23:29.229	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:23:29.229	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:23:30.182	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:23:32.183	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    755 mV 	GMT -5:00
12/15/23 14:23:32.246	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:23:33.433	Information	[Instrument Response] 26 Progress 87% 40 seconds	GMT -5:00
12/15/23 14:23:34.198	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:23:36.213	Information	[Instrument Response] 26 ImageBlock 700 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 14:23:36.213	Information	[Instrument Response] 26 ImageBlock 800 0 252 319 14 319 302 8932	GMT -5:00
12/15/23 14:23:36.213	Information	Transferring image block at (0,252) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:23:36.229	Information	Transferring image block at (0,252) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:23:37.220	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    742 mV 	GMT -5:00
12/15/23 14:23:41.875	Information	[Instrument Response] 26 Progress 90% 30 seconds	GMT -5:00
12/15/23 14:23:42.234	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/15/23 14:23:42.297	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:23:42.297	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/15/23 14:23:42.297	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:23:43.251	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:23:44.251	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/15/23 14:23:45.252	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:23:49.467	Information	[Instrument Response] 26 ImageBlock 700 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 14:23:49.467	Information	[Instrument Response] 26 ImageBlock 800 0 266 319 14 319 302 8932	GMT -5:00
12/15/23 14:23:49.482	Information	Transferring image block at (0,266) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:23:49.498	Information	Transferring image block at (0,266) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:23:53.536	Information	[Instrument Response] 26 Progress 94% 20 seconds	GMT -5:00
12/15/23 14:23:55.317	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 14:23:55.364	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:23:55.364	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:23:55.364	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:23:55.364	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/15/23 14:23:55.364	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/15/23 14:23:56.317	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 14:23:56.364	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/15/23 14:23:56.364	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/15/23 14:23:57.317	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/15/23 14:24:02.738	Information	[Instrument Response] 26 ImageBlock 700 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 14:24:02.738	Information	[Instrument Response] 26 ImageBlock 800 0 280 319 14 319 302 8932	GMT -5:00
12/15/23 14:24:02.738	Information	Transferring image block at (0,280) with size 319x14 to the 700 channel.	GMT -5:00
12/15/23 14:24:02.769	Information	Transferring image block at (0,280) with size 319x14 to the 800 channel.	GMT -5:00
12/15/23 14:24:03.582	Information	[Instrument Response] 26 Progress 98% 10 seconds	GMT -5:00
12/15/23 14:24:06.371	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/15/23 14:24:06.419	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/15/23 14:24:06.419	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/15/23 14:24:06.419	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/15/23 14:24:07.371	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/15/23 14:24:11.218	Information	[Instrument Response] 26 Progress 100% 0 seconds	GMT -5:00
12/15/23 14:24:11.390	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/15/23 14:24:11.437	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         137 V  	GMT -5:00
12/15/23 14:24:11.437	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/15/23 14:24:11.437	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/15/23 14:24:12.385	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/15/23 14:24:13.228	Information	[Instrument Response] 26 ImageBlock 700 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 14:24:13.228	Information	[Instrument Response] 26 ImageBlock 800 0 294 319 8 319 302 5104	GMT -5:00
12/15/23 14:24:13.228	Information	[Instrument Response] 26 Status 0 COMPLETE SCAN 291.071 seconds	GMT -5:00
12/15/23 14:24:13.228	Information	Transferring image block at (0,294) with size 319x8 to the 700 channel.	GMT -5:00
12/15/23 14:24:13.228	Information	[Instrument Response] 26 Status 0 Averages (sec): setup=2.00005 line=0.402433 => total=246	GMT -5:00
12/15/23 14:24:13.244	Information	Transferring image block at (0,294) with size 319x8 to the 800 channel.	GMT -5:00
12/15/23 14:24:13.260	Information	[Instrument Response] 26 Status 0 Comparisons: setup=1.00997 line=0.950274	GMT -5:00
12/15/23 14:24:13.306	Information	[Instrument Response] 26 Status 0 Finalize motors	GMT -5:00
12/15/23 14:24:13.306	Information	[Instrument Response] 26 Progress 100% Scan complete	GMT -5:00
12/15/23 14:24:13.338	Information	[Instrument Response] 26 Result 0 OK	GMT -5:00
12/15/23 14:24:13.338	Information	Acquisition 0000014_01 finished with channels: 700 800.	GMT -5:00
12/15/23 14:24:13.603	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/15/23 14:24:13.603	Information	Scan completed successfully	GMT -5:00
12/15/23 14:24:13.713	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         142 V  	GMT -5:00
12/15/23 14:24:13.728	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/15/23 14:24:13.728	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/15/23 14:24:13.728	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/15/23 14:24:14.390	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          83 V  	GMT -5:00
12/15/23 14:24:14.454	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          52 V  	GMT -5:00
12/15/23 14:24:15.405	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          50 V  	GMT -5:00
12/15/23 14:24:15.460	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          32 V  	GMT -5:00
12/15/23 14:24:16.405	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          32 V  	GMT -5:00
12/15/23 14:24:16.464	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          20 V  	GMT -5:00
12/15/23 14:24:16.464	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      305 mA 	GMT -5:00
12/15/23 14:24:17.401	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          21 V  	GMT -5:00
12/15/23 14:24:17.463	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          13 V  	GMT -5:00
12/15/23 14:24:18.418	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          14 V  	GMT -5:00
12/15/23 14:24:18.572	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           9 V  	GMT -5:00
12/15/23 14:24:19.420	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          10 V  	GMT -5:00
12/15/23 14:24:19.467	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           7 V  	GMT -5:00
12/15/23 14:24:20.422	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           7 V  	GMT -5:00
12/15/23 14:24:20.469	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           4 V  	GMT -5:00
12/15/23 14:24:21.424	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/15/23 14:24:22.425	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/15/23 14:24:23.427	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/15/23 14:24:26.449	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/15/23 14:24:27.443	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/15/23 14:24:27.506	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    755 mV 	GMT -5:00
12/15/23 14:24:29.459	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/15/23 14:26:49.885	Information	Successfully saved C:\Users\wmao2\2023-12-15\Image_0000006_01.tif	GMT -5:00
12/15/23 14:26:57.352	Information	Successfully saved C:\Users\wmao2\2023-12-15\Image_0000007_01.tif	GMT -5:00
12/15/23 14:27:11.552	Information	Successfully saved C:\Users\wmao2\2023-12-15\Image_0000008_01.tif	GMT -5:00
12/15/23 14:27:21.160	Information	Successfully saved C:\Users\wmao2\2023-12-15\Image_0000009_01.tif	GMT -5:00
12/15/23 14:27:30.517	Information	Successfully saved C:\Users\wmao2\2023-12-15\Image_0000010_01.tif	GMT -5:00
12/15/23 14:27:39.048	Information	Successfully saved C:\Users\wmao2\2023-12-15\Image_0000011_01.tif	GMT -5:00
12/15/23 14:27:48.055	Information	Successfully saved C:\Users\wmao2\2023-12-15\Image_0000012_01.tif	GMT -5:00
12/15/23 14:27:55.929	Information	Successfully saved C:\Users\wmao2\2023-12-15\Image_0000013_01.tif	GMT -5:00
12/15/23 14:31:45.672	Information	[Instrument Request]  27 Stop 	GMT -5:00
12/15/23 14:31:45.672	Information	[Instrument Response] 27 Result 0 OK	GMT -5:00
12/15/23 14:31:46.234	Information	[Instrument Response] 0 Status 1 Closing connection	GMT -5:00
